a:2:{s:9:"oldRecord";a:1:{s:8:"bodytext";s:2779:"The SVT 20 can be readily equipped with a liquid/gas-temperature controlled measuring cell, MC-TFC 130, or with a peltier/gas-temperature controlled measuring cell, MC-TPC 180. Inside the measuring cell gas (preferably helium)&nbsp; can be used, as a heat transfer medium, to optimise the transfer of the required temperature regime to the measuring capillary. Every measuring cell is easily mountable/dismountable and includes the water-coolable high-dynamic measuring drive.
Measuring cell MC-TFC 130
<h2>MC-TFC 130</h2>
Temperature controlled measuring cell for installation onto the tilting stage of the SVT.<br />For measurements lower or higher than room temperature. For use with a liquid circulator, e.g. the liquid cryostat F25 ME. The cell temperature is set by the liquid circulator. In the measurement chamber gas (preferably helium) can be used as a heat transfer medium, to optimise the transfer of the required temperature regime to the measuring capillary.<br /><strong><em>Consisting of:</em></strong>
<ul><li>Vibration free, electronically commuted, high-dynamic DC-Drive.</li><li>Liquid/gas temperature controlled chamber with 2 observation windows made from optically clear glass with quick -clamp device for holding the capillary.</li><li>PT100 temperature measurement sensor.</li></ul>
<br /><em><strong>Technical data:</strong></em><br />
<ul><li>Temperature range -10 to 130 Ã°C.</li><li>Heating up and cooling down rate: dependant on the liquid circulator.</li></ul>
<link http://www.dataphysics.de/?action=zoom_view&id=505&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2009+by+Dataphysics _blank></link>PID controller TC 180/SVT with measuring cell MC-TPC 180<br />
<h2>MC-TPC 180</h2>
Temperature controlled measuring cell for installation onto the tilting stage of the SVT.<br />For measurements lower or higher than room temperature. For use with the PID Temperature controller, TC 180/MC-TPC and a liquid circulator, e.g. the liquid cryostat F25 ME. The temperature setting can be done at the TC 180/MC-TPC or at the touch panel of the SVT.<br />In the measurement chamber gas (preferably helium) can be used as a heat transfer medium.<br />Needs counter cooling with fluid circulator.<br />
<ul><li><em><strong>Consisting of:<br /></strong></em></li><li>Vibration free, electronically commuted, high-dynamic DC-Drive.</li><li>Peltier/gas temperature controlled chamber with 2 observation windows made from optically clear glass with quick clamp device for holding the capillary.</li><li>Two resistance thermometers Pt 100 as measuring and control sensors.</li></ul>
<br /><em><strong>Technical data:</strong></em><br />
<ul><li>Temperature range: -30 to 180 Ã°C; Ã±0.1 K.</li><li>Heat up and cool down rate: Ã±1 K/s.</li></ul>";}s:9:"newRecord";a:1:{s:8:"bodytext";s:2514:"The SVT 20 can be readily equipped with a liquid/gas-temperature controlled measuring cell, MC-TFC 130, or with a peltier/gas-temperature controlled measuring cell, MC-TPC 180. Inside the measuring cell gas (preferably helium)&nbsp; can be used, as a heat transfer medium, to optimise the transfer of the required temperature regime to the measuring capillary. Every measuring cell is easily mountable/dismountable and includes the water-coolable high-dynamic measuring drive.
Measuring cell MC-TFC 130
<h2>MC-TFC 130</h2>
Temperature controlled measuring cell for installation onto the tilting stage of the SVT.<br />For measurements lower or higher than room temperature. For use with a liquid circulator, e.g. the liquid cryostat F25 ME. The cell temperature is set by the liquid circulator. In the measurement chamber gas (preferably helium) can be used as a heat transfer medium, to optimise the transfer of the required temperature regime to the measuring capillary.<br /><b><i>Consisting of:</i></b>
<ul><li>Vibration free, electronically commuted, high-dynamic DC-Drive.</li><li>Liquid/gas temperature controlled chamber with 2 observation windows made from optically clear glass with quick -clamp device for holding the capillary.</li><li>PT100 temperature measurement sensor.</li></ul>
<br /><i><b>Technical data:</b></i>
<ul><li>Temperature range -10 to 130 Ã°C.</li><li>Heating up and cooling down rate: dependant on the liquid circulator.</li></ul>
<h2>MC-TPC 180</h2>
Temperature controlled measuring cell for installation onto the tilting stage of the SVT.<br />For measurements lower or higher than room temperature. For use with the PID Temperature controller, TC 180/MC-TPC and a liquid circulator, e.g. the liquid cryostat F25 ME. The temperature setting can be done at the TC 180/MC-TPC or at the touch panel of the SVT.<br />In the measurement chamber gas (preferably helium) can be used as a heat transfer medium.<br />Needs counter cooling with fluid circulator.
<ul><li><em><strong>Consisting of:<br /></strong></em></li><li>Vibration free, electronically commuted, high-dynamic DC-Drive.</li><li>Peltier/gas temperature controlled chamber with 2 observation windows made from optically clear glass with quick clamp device for holding the capillary.</li><li>Two resistance thermometers Pt 100 as measuring and control sensors.</li></ul>
<br /><i><b>Technical data:</b></i>
<ul><li>Temperature range: -30 to 180 Ã°C; Ã±0.1 K.</li><li>Heat up and cool down rate: Ã±1 K/s.</li></ul>";}}