a:2:{s:9:"oldRecord";a:1:{s:8:"bodytext";s:5785:"For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning (Ã±0.05 mm) of the dosing needles. Available in four variants, each with the option of further enhancement to the &quot;plus&quot; version with increased positioning accuracy ( Ã±0.005 mm):
<ul><li>E-MD/2 and E-MD/2plus for max. two dosing needles.</li><li>E-MD/3 and E-MD/3plus for max. three dosing needles.</li><li>E-MD/4 and E-MD/4plus for max. four dosing needles.</li><li>E-MD/6 and E-MD/6plus for max. six dosing needles.</li></ul>
<h2>Electronic multiple direct dosing system TD-DE/3</h2>
For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning of up to three direct dosing modules. Combinable with maximum three electronic syringe modules ES-D/M.
<h2>SIngle-Direct dosing systems SD-Dx</h2>
Paints, varnishes, glues and latex dispersions are typical applications for the&nbsp;SD-DMand&nbsp;SD-DE&nbsp;direct dosing systems. Both direct dosing systems can be used to dose either pendant or sessile liquid drops (on a substrate) through disposable syringes and needles without the need for tubing. The SD-DE is an attachment to the E-MD electronic multiple dosing system, whereas the SD-DM is mounted as a stand-alone system directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and a connection cable
<h2>Double-Direct dosing system DD-DM</h2>
Paints, varnishes, glues and latex dispersions are typical applications for the&nbsp;DD-DMdouble direct dosing system. Thedouble direct dosing system can be used to dose two different sessile liquid drops (on a substrate), for the determination of the surface free energy, through disposable syringes and needles without the need for tubing. The DD-DM is mounted directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of two syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and two connection cables.
<h2>Electronic syringe module ES</h2>
Software controlled (automatable) dosing volume (min. 50 nl) and dosing rate (0.06&nbsp;Ãµl/s...26.4 Ãµl/s). Also in a vairant for the direct dosing systems SD-DM, SD-DE and TD-DE available. Easy expandable with the syringe heating device SHD.<br /><link http://www.dataphysics.de/?action=zoom_view&id=765&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>Nanoliter dosing system on an OCA 40 Micro consisting of PCU Nanodrop, 3 DUN/E-MD, and 3 NDV INKA
<h2>Nanoliter dosing system</h2>
The software controlled nanoliter dosing system -comprising the pressure control unitPCU Nanodrop&nbsp;and up to four nanodrop dosing valves&nbsp;NDV xx- enables the automatic measurement of wetting properties of microscopic samples using droplets in the nanoliter range (1...500 nl). The dosing valves were mounted with the holderDUN/SD-DM&nbsp;for a direct dosing system or&nbsp;DUN/E-MD&nbsp;for the electronic multiple dosing system E-MD/x plus.
<h2>Picoliter dosing system</h2>
The innovative picoliter dosing system comprises a dosing unit, picodrop DUP/xx and a Picodrop dispenser head, PDH/xx. It enables the analysis of surfaces in sub-microscopic dimensions by dispensing single droplets of 15 to 400 pl precisely to the pre-defined position.<br /><link http://www.dataphysics.de/?action=zoom_view&id=848&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link><br />Picoliter droplet on an human hair
<h2>Refill and rinse system with liquid pump cleaner RRS-LPC 3/1</h2>
For the automated, software controlled, refill and rinse of the dosing syringes and for the suction cleaning of a sample onto which liquid has been previously dosed. To be used with the following optional components:
<ul><li>electronic multiple dosing system,&nbsp;E-MD/4.</li><li>3 electronic syringe modules,&nbsp;ES.</li><li>suction cleaning system,&nbsp;LPC OCA.</li><li>Vacuum pump,&nbsp;MVP.</li></ul>
<h2>Dosing syringes DS xx</h2>
Subsequent a selection of the most important dosing syringes:
<ul><li>DS 100/GT&nbsp;gas tight 100 Ãµl syringe.</li><li>DS 500/GT&nbsp;gas tight 500 Ãµl syringe.</li><li>DS 1000/GT&nbsp;gas tight 1000 Ãµl syringe.</li><li>DS-D 1000&nbsp;1000 Ãµl disposable syringe.</li><li>DS-D 1000 SF&nbsp;1000 Ãµl disposable syringe, silicone free.</li></ul>
<h2>Dosing needles SN xx</h2>
Our stock needle options comprises of more than 40 different possibilities, made of stainless steel, hydrophobic coated stainless steel and PTFE arranged in groups according to the standard application of the needle:
<ul><li>SNS xx&nbsp;dosing needles for sessile drop measurements.</li><li>SNP xx&nbsp;dosing needles for pendant drop measurements.</li><li>SNC xx&nbsp;upwards bended dosing needles for captive bubble measurements.</li><li>SNS-D xx&nbsp;disposable dosing needles for sessile drop measurements.</li><li>SNP-D xx&nbsp;disposable dosing needles for pendant drop measurements.</li></ul>
<h2>Dosing tubes DT xx</h2>
If using a standard dosing system (SD, MD/4, or E-MD/x) the dosing needle is connected, to the syringe, by a stainless steel tLuer Lock adapter,&nbsp;LL A. (The Luer Lock adaptor also serves as a means of securing the position of the needle relative to the dosing system)&nbsp; A dosing tube&nbsp;DT xx&nbsp;(made of PTFE) connects the dosing syringe,DS xx, to the needle.&nbsp;<br />HY-Kit
<h2>Hydrophobisation kit for dosing needles SNS/SNP HY-Kit</h2>
Special hydrophobic coating kit to prevent the wetting of the outer surface of the dosing needles.";}s:9:"newRecord";a:1:{s:8:"bodytext";s:5342:"For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning (Ã±0.05 mm) of the dosing needles. Available in four variants, each with the option of further enhancement to the &quot;plus&quot; version with increased positioning accuracy ( Ã±0.005 mm):
<ul><li>E-MD/2 and E-MD/2plus for max. two dosing needles.</li><li>E-MD/3 and E-MD/3plus for max. three dosing needles.</li><li>E-MD/4 and E-MD/4plus for max. four dosing needles.</li><li>E-MD/6 and E-MD/6plus for max. six dosing needles.</li></ul>
<h2>Electronic multiple direct dosing system TD-DE/3</h2>
For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning of up to three direct dosing modules. Combinable with maximum three electronic syringe modules ES-D/M.
<h2>SIngle-Direct dosing systems SD-Dx</h2>
Paints, varnishes, glues and latex dispersions are typical applications for the&nbsp;SD-DMand&nbsp;SD-DE&nbsp;direct dosing systems. Both direct dosing systems can be used to dose either pendant or sessile liquid drops (on a substrate) through disposable syringes and needles without the need for tubing. The SD-DE is an attachment to the E-MD electronic multiple dosing system, whereas the SD-DM is mounted as a stand-alone system directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and a connection cable
<h2>Double-Direct dosing system DD-DM</h2>
Paints, varnishes, glues and latex dispersions are typical applications for the&nbsp;DD-DMdouble direct dosing system. Thedouble direct dosing system can be used to dose two different sessile liquid drops (on a substrate), for the determination of the surface free energy, through disposable syringes and needles without the need for tubing. The DD-DM is mounted directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of two syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and two connection cables.
<h2>Electronic syringe module ES</h2>
Software controlled (automatable) dosing volume (min. 50 nl) and dosing rate (0.06&nbsp;Ãµl/s...26.4 Ãµl/s). Also in a vairant for the direct dosing systems SD-DM, SD-DE and TD-DE available. Easy expandable with the syringe heating device SHD.
<h2>Nanoliter dosing system</h2>
The software controlled nanoliter dosing system -comprising the pressure control unitPCU Nanodrop&nbsp;and up to four nanodrop dosing valves&nbsp;NDV xx- enables the automatic measurement of wetting properties of microscopic samples using droplets in the nanoliter range (1...500 nl). The dosing valves were mounted with the holderDUN/SD-DM&nbsp;for a direct dosing system or&nbsp;DUN/E-MD&nbsp;for the electronic multiple dosing system E-MD/x plus.
<h2>Picoliter dosing system</h2>
The innovative picoliter dosing system comprises a dosing unit, picodrop DUP/xx and a Picodrop dispenser head, PDH/xx. It enables the analysis of surfaces in sub-microscopic dimensions by dispensing single droplets of 15 to 400 pl precisely to the pre-defined position.
<h2>Refill and rinse system with liquid pump cleaner RRS-LPC 3/1</h2>
For the automated, software controlled, refill and rinse of the dosing syringes and for the suction cleaning of a sample onto which liquid has been previously dosed. To be used with the following optional components:
<ul><li>electronic multiple dosing system,&nbsp;E-MD/4.</li><li>3 electronic syringe modules,&nbsp;ES.</li><li>suction cleaning system,&nbsp;LPC OCA.</li><li>Vacuum pump,&nbsp;MVP.</li></ul>
<h2>Dosing syringes DS xx</h2>
Subsequent a selection of the most important dosing syringes:
<ul><li>DS 100/GT&nbsp;gas tight 100 Ãµl syringe.</li><li>DS 500/GT&nbsp;gas tight 500 Ãµl syringe.</li><li>DS 1000/GT&nbsp;gas tight 1000 Ãµl syringe.</li><li>DS-D 1000&nbsp;1000 Ãµl disposable syringe.</li><li>DS-D 1000 SF&nbsp;1000 Ãµl disposable syringe, silicone free.</li></ul>
<h2>Dosing needles SN xx</h2>
Our stock needle options comprises of more than 40 different possibilities, made of stainless steel, hydrophobic coated stainless steel and PTFE arranged in groups according to the standard application of the needle:
<ul><li>SNS xx&nbsp;dosing needles for sessile drop measurements.</li><li>SNP xx&nbsp;dosing needles for pendant drop measurements.</li><li>SNC xx&nbsp;upwards bended dosing needles for captive bubble measurements.</li><li>SNS-D xx&nbsp;disposable dosing needles for sessile drop measurements.</li><li>SNP-D xx&nbsp;disposable dosing needles for pendant drop measurements.</li></ul>
<h2>Dosing tubes DT xx</h2>
If using a standard dosing system (SD, MD/4, or E-MD/x) the dosing needle is connected, to the syringe, by a stainless steel tLuer Lock adapter,&nbsp;LL A. (The Luer Lock adaptor also serves as a means of securing the position of the needle relative to the dosing system)&nbsp; A dosing tube&nbsp;DT xx&nbsp;(made of PTFE) connects the dosing syringe,DS xx, to the needle.&nbsp;<br />HY-Kit
<h2>Hydrophobisation kit for dosing needles SNS/SNP HY-Kit</h2>
Special hydrophobic coating kit to prevent the wetting of the outer surface of the dosing needles.";}}