HGC 30
The humidity generator HGC 30 is designed for the automated regulation of the relative humidity of ambient clean air, nitrogen, argon, and other inert gases for a wide range of applications.&nbsp; It is easily connectable to any temperature controlled measuring chamber (OCA, TGA, TMA, DSC etc.). The ambient air or the external supplied gas first flows through the integrated desiccant reservoir. An electronic valve, controlled by the humidity and temperature sensor â??directly mounted in the measuring chamberâ?? diverts some air/gas through the heated water reservoir before it flows through the heated connection tube. Main features
 &nbsp;Stand alone generation of dry air/gas with and without an external pressurized gas supply. Heated water reservoir. Heated connection gas tube (transfer line) to thermal chamber to avoid condensation. Easy control via touch-screen or optional software. Serial RS 485 or USB interface. Calculation of dew point in the measuring chamber. Automated drying and regeneration of desiccant by a built-in heating system in two alternatively used desiccant reservoirs (total stored desiccant amount about 1.0 kg). External gas supply with integrated pressure regulator. Integrated flow rate controller (0.07 ... 3.5 Nl/min; calibrated for N2 and Ar).
 Data Sheet HGC
