a:2:{s:9:"oldRecord";a:2:{s:8:"bodytext";s:3098:"<h2>Spinning Drop Video Tensiometer.</h2>
The spinning drop video tensioÃ­meter, SVT 20N, is a special purpose optical instrument for measuring of conventional to extremely low interfacial tensions.&nbsp;<br />The control/data analysis software is designed for intuative use and fast access to all control elements.&nbsp;<br />Combined with the software packages SVTS 20, SVTS 21, and SVTS 22 the SVT 20N provides the following functions:
<ul><li>Automatic control of the electronic tilting base, for positioning of the rotating droplet along the rotary axis of the capillary</li><li>Automatic variation of the rotation speed</li><li>Measurement and analysis of the static, time, and temperature dependent interfacial tension (1Ã·10<sup>-6&nbsp;</sup>to 2Ã·10<sup>3</sup>&nbsp;mN/m) according to the spinning&nbsp; drop method and oscillating spinning&nbsp; drop method.</li><li>Measurement and analysis of the 2D and 3D extensional relaxation of visco-elastic liquids and membrane encapsulated liquids. drops by variation of the rotational speed.</li><li>Determination of the yield point of liquid crystalline.</li><li>Determination of the dispersive and polar parts of liquids on the basis of interfacial tension, with error limits.</li><li>Calculation of the geometry of rotating drops related to the rotary axis.</li></ul>
<br /><b>Features of the basic instrument configuration:</b>
<ul><li>High dynamic, electronically commutated direct current motor with software controlled speed control (0 to 20?000 rev/min; Ã± 0.001 rev/min)&nbsp; to realise spped steps (Ã± 2?000 rev/min/s) and sinusoidal speed variations (0.01 to 200 Hz) for relaxation measurements.</li><li>High-performance 6-fold power zoom lens with an integrated continuous fine focus and adjustable observation angle.</li><li>Video measuring system with USB-camera (max. 146 images/s sample rate).</li><li>USB connection to PC (or Notebook).</li><li>LED lighting with software controlled adjustable intensity and strobe frequency.<br />Integrated touch screen for controlling basic functions.</li><li>Automatic, software controlled position and tilt adjustment.</li><li>Automatic calibration function for correcting optical distortions caused by the capillaries (cylindrical lens effects).</li></ul>
<br /><b>Optional SVT accessories:</b>
<ul><li>Easy mountable liquid/gas-temperature controlled measuring cell MC TFC 130; temperature range&nbsp; -10 to 130 Ã°C.</li><li>Easy mountable peltier/gas-temperature controlled measuring cell MC TPC 180; temperature range -30 to 180 Ã°C; Ã±0.1 K resolution with a heat up and cool down rate of Ã±1 K/s.</li><li>Fast-exchangeable special high-temperature glass capillaries (up to 180 Ã°C) FEC&nbsp;622/4 0-HT.</li><li>Cost-effective (disposable) fast-exchange capillaries made of polycarbonate (up to 60 Ã°C) FEC&nbsp;780/400-P.</li><li>Automatic dosing system ES-D/S.</li><li>Calibration standard DCS-SVT.</li><li>The available software modules for the SVT 20N are:</li><p>- SVTS 20&nbsp; - Interfacial Tension.</p><p>- SVTS 21 - Oscillation.</p><p>- SVTS 22 - Geometry.</p></ul>";s:15:"l18n_diffsource";s:2661:"a:22:{s:5:"CType";s:4:"text";s:6:"colPos";s:1:"1";s:16:"sys_language_uid";s:1:"0";s:6:"header";s:24:"Temperierbare Messzellen";s:13:"header_layout";s:1:"0";s:15:"header_position";s:0:"";s:4:"date";s:1:"0";s:11:"header_link";s:0:"";s:8:"bodytext";s:2092:"Auch das SVT 20N folgt dem flexiblen, modularen DataPhysics â??Baukasten-Prinzipâ??. Es kann wahlweise mit einer flÃ¼ssigkeits-/gastemperierbaren Messzelle oder einer Peltier-/gastemperierbaren Messzelle ausgerÃ¼stet werden. In der Messzelle dient Gas (vorzugsweise Helium) als WÃ¤rmetrÃ¤ger zur Messkapillare.
Jede Messzelle ist einfach montierbar/demontierbar und enthÃ¤lt ausser der temperierbaren Messkammer auch den wasserkÃ¼hlbaren, hochdynamischen Messantrieb.
<h2>MC-TFC 130</h2>
FlÃ¼ssigkeits-/gastemperierbare Messzelle fÃ¼r das Spinning Drop Video Tensiometer SVT 20N, zum AnschluÃ? an einen KÃ¤lte-UmwÃ¤lzthermostaten FÃ¼r Messungen unter oder Ã¼ber Raumtemperatur, z.B. fÃ¼r die Aufnahme von Temperaturkoeffizienten
Bestehend aus:
<ul><li>flÃ¼ssigkeitskÃ¼hlbarem, elektronisch kommutiertem, hochdynamischem Messantrieb</li><li>Schnellspannvorrichtung fÃ¼r die Messkapillare</li><li>flÃ¼ssigkeits-/gastemperierbarer Kammer mit 2 Sichtfenstern aus optischem Spezialglas</li><li>Anschluss fÃ¼r Temperiergas</li><li>Widerstandsthermometer Pt 100 als MessfÃ¼hler</li></ul>
<b>Technische Daten:</b>
<ul><li>Temperaturbereich: -10...130Ã° C</li><li>Aufheiz- und AbkÃ¼hlrate: abhÃ¤ngig vom FlÃ¼ssigkeitsthermostat</li></ul>
<h2>MC-TPC 180</h2>
Peltier-/gastemperierbare Messzelle fÃ¼r das Spinning Drop Video Tensiometer SVT 20N FÃ¼r Messungen unter oder Ã¼ber Raumtemperatur, z.B. fÃ¼r die Aufnahme von Temperaturkoeffizienten Manuelle oder softwaregesteuerte Temperaturvorgabe Ã¼ber den PID-TemperaturÃ­regler TC 180/SVT Das eingebaute Peltierelement benÃ¶tigt eine FlÃ¼ssigkeitsgegenkÃ¼hlung.
Bestehend aus:
<ul><li>lÃ¼ssigkeitskÃ¼hlbarem, elektronisch kommutiertem, hochdynamischem Messantrieb</li><li>Schnellspannvorrichtung fÃ¼r die Messkapillare</li><li>flÃ¼ssigkeits-/gastemperierbarer Kammer mit 2 Sichtfenstern aus optischem Spezialglas</li><li>Anschluss fÃ¼r Temperiergas</li><li>Widerstandsthermometer Pt 100 als MessfÃ¼hler</li></ul>
<b>Technische Daten:</b>
<ul><li>Temperaturbereich: -30...180Ã° C; Ã±0,1 K</li><li>Aufheiz- und AbkÃ¼hlrate: Ã± 1 K/s</li></ul>";s:11:"rte_enabled";s:1:"0";s:6:"layout";s:1:"0";s:11:"spaceBefore";s:1:"0";s:10:"spaceAfter";s:1:"0";s:13:"section_frame";s:1:"0";s:6:"hidden";s:1:"0";s:12:"sectionIndex";s:1:"1";s:9:"linkToTop";s:1:"0";s:9:"starttime";N;s:7:"endtime";N;s:8:"fe_group";s:0:"";s:10:"categories";s:1:"0";s:11:"l18n_parent";s:1:"0";}";}s:9:"newRecord";a:2:{s:8:"bodytext";s:3186:"<h2>Spinning Drop Video Tensiometer.</h2>
The spinning drop video tensioÃ­meter, SVT 20N, is a special purpose optical instrument for measuring of conventional to extremely low interfacial tensions.&nbsp;<br />The control/data analysis software is designed for intuative use and fast access to all control elements.&nbsp;<br />Combined with the software packages SVTS 20, SVTS 21, and SVTS 22 the SVT 20N provides the following functions:
<ul><li>Automatic control of the electronic tilting base, for positioning of the rotating droplet along the rotary axis of the capillary</li><li>Automatic variation of the rotation speed</li><li>Measurement and analysis of the static, time, and temperature dependent interfacial tension (1Ã·10<sup>-6&nbsp;</sup>to 2Ã·10<sup>3</sup>&nbsp;mN/m) according to the spinning&nbsp; drop method and oscillating spinning&nbsp; drop method.</li><li>Measurement and analysis of the 2D and 3D extensional relaxation of visco-elastic liquids and membrane encapsulated liquids. drops by variation of the rotational speed.</li><li>Determination of the yield point of liquid crystalline.</li><li>Determination of the dispersive and polar parts of liquids on the basis of interfacial tension, with error limits.</li><li>Calculation of the geometry of rotating drops related to the rotary axis.</li></ul>
<br /><b>Features of the basic instrument configuration:</b>
<ul><li>High dynamic, electronically commutated direct current motor with software controlled speed control (0 to 20?000 rev/min; Ã± 0.001 rev/min)&nbsp; to realise spped steps (Ã± 2?000 rev/min/s) and sinusoidal speed variations (0.01 to 200 Hz) for relaxation measurements.</li><li>High-performance 6-fold power zoom lens with an integrated continuous fine focus and adjustable observation angle.</li><li>Video measuring system with USB-camera (max. 146 images/s sample rate).</li><li>USB connection to PC (or Notebook).</li><li>LED lighting with software controlled adjustable intensity and strobe frequency.<br />Integrated touch screen for controlling basic functions.</li><li>Automatic, software controlled position and tilt adjustment.</li><li>Automatic calibration function for correcting optical distortions caused by the capillaries (cylindrical lens effects).</li></ul>
<br /><b>Optional SVT accessories:</b>
<ul><li>Easy mountable liquid/gas-temperature controlled measuring cell MC TFC 130; temperature range&nbsp; -10 to 130 Ã°C.</li><li>Easy mountable peltier/gas-temperature controlled measuring cell MC TPC 180; temperature range -30 to 180 Ã°C; Ã±0.1 K resolution with a heat up and cool down rate of Ã±1 K/s.</li><li>Fast-exchangeable special high-temperature glass capillaries (up to 180 Ã°C) FEC&nbsp;622/4 0-HT.</li><li>Cost-effective (disposable) fast-exchange capillaries made of polycarbonate (up to 60 Ã°C) FEC&nbsp;780/400-P.</li><li>Automatic dosing system ES-D/S.</li><li>Calibration standard DCS-SVT.</li><li>The available software modules for the SVT 20N are:<ul><li>SVTS 20&nbsp; - Interfacial Tension.</li><li>SVTS 21 - Oscillation.</li><li>SVTS 22 - Geometry</li></ul></li></ul>
<link file:370 - download "Initiates file download">Data Sheet SVT 20 N</link>";s:15:"l18n_diffsource";s:3229:"a:22:{s:5:"CType";s:4:"text";s:6:"colPos";s:1:"1";s:16:"sys_language_uid";s:1:"0";s:6:"header";s:8:"SVT 20 N";s:13:"header_layout";s:1:"0";s:15:"header_position";s:0:"";s:4:"date";s:1:"0";s:11:"header_link";s:0:"";s:8:"bodytext";s:2677:"<h2>Spinning Drop Video Tensiometer</h2>
Das Spinning Drop Video TensioÃ­meter SVT 20N ist ein optisches SpezialgerÃ¤t fÃ¼r die Messung extrem niedriger GrenzflÃ¤chenspannungen.&nbsp;<br />FÃ¼r die Analyse der Tensidwirkung im Rahmen der Emulsionsentwicklung oder der tertiÃ¤ren Ã?lfÃ¶rderung bietet diese Messtechnik einzigartige MÃ¶glichkeiten. Die vielfÃ¤ltigen Funktionen des SVT&nbsp;20N in Kombination mit den Softwaremodulen SVTS&nbsp;20, SVTS 21 und SVTS 22 sind:
<ul><li>automatische Steuerung der elektronischen KippÃ­vorrichtung zur Positionierung des rotierenden Tropfens entlang der Drehachse der Kapillare</li><li>automatische Variation der Rotationsgeschwindigkeit</li><li>Messung und Auswertung der statischen, zeit- und temperaturabhÃ¤ngigen GrenzflÃ¤chenspannung (1Ã·10<sup>-6</sup>&nbsp;â?¦ 2Ã·10<sup>3</sup>&nbsp;mN/m) nach der Spinning&nbsp; Drop-Methode und der Oscillating Spinning&nbsp; Drop-Methode</li><li>Messung und Auswertung der 2D- und 3D-Dehnrelaxation von viskoelastischen FlÃ¼ssigkeiten und FlÃ¼ssigkeitstropfen, die in Membranen eingeschlossen sind</li><li>Bestimmung der FlieÃ?grenze von flÃ¼ssigkristallinen Materialien</li><li>Berechnung des dispersen und polaren Anteils von FlÃ¼ssigkeiten anhand gemessener OberflÃ¤chen- und GrenzflÃ¤chenspannungen inklusive Fehlergrenzen</li><li>Berechnung der Geometrie rotierender Tropfen in Bezug auf die optische Achse des Systems</li></ul>
&nbsp;<br />Das SVT 20 N besteht aus dem GrundgerÃ¤t mit folgender technischer Ausstattung:
<ul><li>hochdynamischer, elektronisch kommutierter Gleichstrommotor mit softwaregesteuerter Drehzahlregelung (0 â?¦ 20?000 U/min; Ã± 0,001 U/min)&nbsp; zur Realisierung von DrehzahlsprÃ¼ngen (Ã± 2?000 U/min/s) und sinusfÃ¶rmigen Drehzahlvariationen (0,01 â?¦ 200 Hz) fÃ¼r Relaxationsmessungen</li><li>Hochleistungsobjektiv mit 6fach-Zoom, integriertem stufenlosem Feinfokus und einstellbarem Beobachtungswinkel</li><li>CCIR-Videomesssystem (768 x 576 Pixel) mit einer maximalen Bildaufnahmerate von 146 Bildern/s</li><li>Anschluss an PC oder Notebook Ã¼ber USB</li><li>prÃ¤zises, schnell und einfach zu bedienendes Kapillarwechselsystem</li><li>Messzelle mit automatischer,&nbsp; softwaregesteuerter Positions- und Neigungsanpassung</li><li>hysteresefreie LED-Beleuchtung mit softwaregesteuerter IntensitÃ¤t und Stroboskopfrequenz zur Tropfenbeobachtung in verschiedenen Drehlagen</li><li>integrierter Touchscreen zur Steuerung der Basisfunktionen</li><li>automatische Kalibrierfunktion zur Korrektur der optischen Verzerrung durch den Zylinderlinsen-Effekt der Kapillare</li></ul>
<link file:369 - download "Initiates file download">Datenblatt SVT 20 N</link>";s:11:"rte_enabled";s:1:"0";s:6:"layout";s:1:"0";s:11:"spaceBefore";s:1:"0";s:10:"spaceAfter";s:1:"0";s:13:"section_frame";s:1:"0";s:6:"hidden";s:1:"0";s:12:"sectionIndex";s:1:"1";s:9:"linkToTop";s:1:"0";s:9:"starttime";N;s:7:"endtime";N;s:8:"fe_group";s:0:"";s:10:"categories";s:1:"0";s:11:"l18n_parent";s:1:"0";}";}}