The<b>&nbsp;dynamic contact angle measuring instrument&nbsp;</b>and&nbsp;<b>tensiometer</b>&nbsp;DCAT&nbsp;11 is the standard instrument for the weight based measurement of contact angle, surface and interfacial tension, critical micelle formation concentration, density, sedimentation and penetration rate, plus penetration resistance.<br /><b>Features of the basic instrument configuration:</b>
<ul><li>High-precision electro dynamic compensation weighing system (weighing range: 10 Ãµg&nbsp; to 210 g) with automatic calibration.</li><li>Software controlled height positioning (travel: 74 mm; resolution: 0.1 Ãµm) of the sample receptacles with variable speed (0.74 Ãµm/s to 500 mm/min).</li><li>Automatic electronic coupling lock for the balance.</li><li>Integrated measurement and control electronics with connections for two Pt 100 temperature sensors (-60 to +450 Ã°C; Ã± 0.01 K resolution; precision 1/3 DIN IEC 751, Class B).</li><li>Display for weighing data, temperature and other information.</li><li>Illuminated sample chamber with inert gas or vapour inlet.</li><li>USB interface with connection cable to PC/notebook.</li></ul>
<br /><b>All DCAT accessories can be used with the DCAT 11 except the electrical temperature control unit TEC 250/DCAT.</b>
<link file:329 - download "Initiates file download">Data Sheet DCAT 11</link>