DCAT accessories range
For this product series, we offer a variety of components and an extensive range of accessories.
 DCAT accessories
DCAT 11EC
The&nbsp;dynamic contact angle measuring instrument&nbsp;and&nbsp;tensiometer&nbsp;DCAT&nbsp;11EC is the budget-priced instrument for the weight-based measurement of contact angle, surface and interfacial tension, critical micelle formation concentration, density, sedimentation and penetration rate, plus penetration resistance. Features of the basic instrument configuration:
 High precision electro dynamic compensation weighing system (weighing&nbsp; range: 100 µg to 210 g) with automatic calibration. Software controlled height positioning (travel: 74 mm; resolution: 1 µm) of the sample receptacles with variable speed (2 µm/s to 60 mm/min) Automatic electronic coupling lock for the balance. Integrated measurement and control electronics with connections for two Pt 100 temperature sensors (-60 to +450 °C; ± 0.01 K resolution; precision 1/3 DIN IEC 751, Class B). Display for weighing data, temperature and other information. Illuminated sample chamber with inert gas or vapor inlet. USB interface with connection cable to PC/notebook.
 All DCAT accessories can be used with the DCAT 11EC except the electrical temperature control unit TEC 250/DCAT.
 Data Sheet DCAT 11 EC
