INSERT INTO `tx_kesearch_index` (`uid`, `pid`, `tstamp`, `crdate`, `cruser_id`, `starttime`, `endtime`, `fe_group`, `targetpid`, `content`, `params`, `type`, `tags`, `abstract`, `sortdate`, `orig_uid`, `orig_pid`, `title`, `language`, `directory`, `hash`) VALUES
(175, 5, 1431355495, 1428385888, 0, 0, 0, '', '210', 'Optical standards for verification of precise performance\n DataPhysics provide several sets of optical standards for the validation/verification of total instrument (hardware and software) performance. Each glass slide features a standard sessile, pendant drop or lamella contour for image capture and subsequent analysis. The comparison of the calculated contact angle or the surface/interfacial tension mea­sure­ment with that stated on the slide is a crucial factor in the evaluation of instrument performance and reliability. All slides are manufactured with the highest precision, as per ISO 9001, in a controlled photolithographic process to serve as reliable references for the precision of measured data.\r\n OCAS basic set&nbsp;three optical drop shape standards for the validation of the contact angle and the surface and interfacial tension measurement according to the height/width method, the Young-Laplace method, and the pendant drop method. OCAS/HW&nbsp;­four optical contact angle standards according to the height/width method (15° ± 0.09°, 30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°). OCAS/HW-LOW&nbsp; three optical contact angle standards according to the height/width method (15° ± 0.09°, 10° ± 0.09°, 5° ± 0.09°). OCAS/YL&nbsp;three optical contact angle standards according to&nbsp; the Young/Laplace method (30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°). OCAS/YL-HIGH&nbsp;four optical contact angle standards according to&nbsp; the Young/Laplace method (160° ± 0.04°, 165° ± 0.04°, 170° ± 0.04°, 175° ± 0.04°). OCAS/PD&nbsp;four optical Pendant Drop standards (20, 35, 50 and 72.8 mN/m; ± 0.02 mN/m). OCAS/LA&nbsp; two optical lamella standards (20 and 35 mN/m; ± 0.02 mN/m) and one Vertical-Rod standard (CA = 30° ± 0,09° and SFT 50 ± 0,02 mN/m).\r\nKeyPad for OCA 35/40 Micro KP 40\r\n For controlling, outside of the software, the position of the motorized X-, Y-, Z-axes, the tilting angle of a tilting base unit, TBU 90E, the angle of an electronic turn table, ETT, and of the focus and observation angle (only OCA 40) of the optical system. The X,Y,Z co-ordinates of a sample table or the angle/position of a tilting unit/turn table, adjusted by the keypad, can be transferred into the SCA 20 software in order to facilitate the automated return to an optimised position during a subsequent automated measurement routine.\r\nTop view video system module for OCA TV-VS\r\n All OCA systems can be equipped with the top view video system TV-VS, to provide an alternative view of the positioning of the dispensed drop on the surface. The USB camera, with up to 87 images per second, together with the high performance 6x parfocal zoom lens and the adjustable observation angle enables a detailed view to your sample.\r\nOscillating drop generator ODG 20\r\n For the measurement of surface elasticities and for relaxational studies at phase boundaries, DataPhysics has developed the oscillating drop generator, ODG 20, and the innovative oscillating drop generator ODG 20P with a high resolution pressure sensor. This extension to the pendant drop contour analysis capability of the OCA series â based on a piezoelectric transducer â excites oscillating drops with a wide range of frequencies and amplitudes. Periodic variations in the drop volume, or alternatively in the drop shape, with constant volume, can be performed. The excitation frequency ranges up to 50 Hz, depending on the density and viscosity of the fluid. The amplitude of the axis-symmetric oscillations varies from micrometers to millimeters. The oscillating drops can be analyzed by conventional drop contour analysis or by the new approach of pressure measurement with the high-performance pressure sensor (ODG 20P).\r\nElectro wetting platform EWP 100\r\n The way in which a defined electrical field influences the wetting and adsorption behaviour of modern, engineered, materials is becoming relevant to an increasing number of applications. The EWP 100 facilitates the analysis of pendant drops and sessile drops, incident with solid surfaces, under a user definable electrical field.\r\nLiquid injection system LIA-MD\r\n The LIA-MD facilitates the study of the effect, on a drops surface, of adding a second liquid directly into that first drop. This special accessory enables the rapid addition of a second liquid, usually a surface-active or chemically reactive phase, into a drop already visible through the OCA optics and displayed on the monitor. The relaxation (back to interfacial equilibrium with the surrounding medium) of the original drop can be observed and evaluated via the SCA software. The overall response time, together with the rates of change of surface and interfacial tensions, during the relaxation process, are indicative of the reactivity of the liquids under study. Developments of synthetic eye and lung fluids, for medical purposes, have already benefitted from studies of this type and more innovative applications are expected in the field of protein and surfactant engineering.\nSpecial accessories\nIn close cooperation with customers, research institutes, and universities we have developed a wide range of application specific accessories to complement the OCA series of instruments. In doing so we are continuously broadening the range of surface chemistry applications within both the laboratory and process fields.\n', '', 'page', '', '', 1429621602, 210, 186, 'Accessoires spéciaux', 3, '', ''),
(176, 5, 1431355495, 1428385888, 0, 0, 0, '', '210', 'Optical standards for verification of precise performance \nDataPhysics provide several sets of optical standards for the validation/verification of total instrument (hardware and software) performance. Each glass slide features a standard sessile, pendant drop or lamella contour for image capture and subsequent analysis. The comparison of the calculated contact angle or the surface/interfacial tension mea­sure­ment with that stated on the slide is a crucial factor in the evaluation of instrument performance and reliability. All slides are manufactured with the highest precision, as per ISO 9001, in a controlled photolithographic process to serve as reliable references for the precision of measured data.\r\n OCAS basic set&nbsp;three optical drop shape standards for the validation of the contact angle and the surface and interfacial tension measurement according to the height/width method, the Young-Laplace method, and the pendant drop method. OCAS/HW&nbsp;­four optical contact angle standards according to the height/width method (15° ± 0.09°, 30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°). OCAS/HW-LOW&nbsp; three optical contact angle standards according to the height/width method (15° ± 0.09°, 10° ± 0.09°, 5° ± 0.09°). OCAS/YL&nbsp;three optical contact angle standards according to&nbsp; the Young/Laplace method (30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°). OCAS/YL-HIGH&nbsp;four optical contact angle standards according to&nbsp; the Young/Laplace method (160° ± 0.04°, 165° ± 0.04°, 170° ± 0.04°, 175° ± 0.04°). OCAS/PD&nbsp;four optical Pendant Drop standards (20, 35, 50 and 72.8 mN/m; ± 0.02 mN/m). OCAS/LA&nbsp; two optical lamella standards (20 and 35 mN/m; ± 0.02 mN/m) and one Vertical-Rod standard (CA = 30° ± 0,09° and SFT 50 ± 0,02 mN/m).\r\nKeyPad for OCA 35/40 Micro KP 40\r\nFor controlling, outside of the software, the position of the motorized X-, Y-, Z-axes, the tilting angle of a tilting base unit, TBU 90E, the angle of an electronic turn table, ETT, and of the focus and observation angle (only OCA 40) of the optical system. The X,Y,Z co-ordinates of a sample table or the angle/position of a tilting unit/turn table, adjusted by the keypad, can be transferred into the SCA 20 software in order to facilitate the automated return to an optimised position during a subsequent automated measurement routine.\r\nTop view video system module for OCA TV-VS\r\nAll OCA systems can be equipped with the top view video system TV-VS, to provide an alternative view of the positioning of the dispensed drop on the surface. The USB camera, with up to 87 images per second, together with the high performance 6x parfocal zoom lens and the adjustable observation angle enables a detailed view to your sample.\r\nOscillating drop generator ODG 20\r\nFor the measurement of surface elasticities and for relaxational studies at phase boundaries, DataPhysics has developed the oscillating drop generator, ODG 20, and the innovative oscillating drop generator ODG 20P with a high resolution pressure sensor. This extension to the pendant drop contour analysis capability of the OCA series â based on a piezoelectric transducer â excites oscillating drops with a wide range of frequencies and amplitudes. Periodic variations in the drop volume, or alternatively in the drop shape, with constant volume, can be performed. The excitation frequency ranges up to 50 Hz, depending on the density and viscosity of the fluid. The amplitude of the axis-symmetric oscillations varies from micrometers to millimeters. The oscillating drops can be analyzed by conventional drop contour analysis or by the new approach of pressure measurement with the high-performance pressure sensor (ODG 20P).\r\nElectro wetting platform EWP 100\r\nThe way in which a defined electrical field influences the wetting and adsorption behaviour of modern, engineered, materials is becoming relevant to an increasing number of applications. The EWP 100 facilitates the analysis of pendant drops and sessile drops, incident with solid surfaces, under a user definable electrical field.\r\nLiquid injection system LIA-MD\r\nThe LIA-MD facilitates the study of the effect, on a drops surface, of adding a second liquid directly into that first drop. This special accessory enables the rapid addition of a second liquid, usually a surface-active or chemically reactive phase, into a drop already visible through the OCA optics and displayed on the monitor. The relaxation (back to interfacial equilibrium with the surrounding medium) of the original drop can be observed and evaluated via the SCA software. The overall response time, together with the rates of change of surface and interfacial tensions, during the relaxation process, are indicative of the reactivity of the liquids under study. Developments of synthetic eye and lung fluids, for medical purposes, have already benefitted from studies of this type and more innovative applications are expected in the field of protein and surfactant engineering.\nSpecial accessories\nIn close cooperation with customers, research institutes, and universities we have developed a wide range of application specific accessories to complement the OCA series of instruments. In doing so we are continuously broadening the range of surface chemistry applications within both the laboratory and process fields.\n', '', 'page', '', '', 1429621602, 210, 186, 'Special accessories', 2, '', ''),
(177, 5, 1431355495, 1428385888, 0, 0, 0, '', '211', 'Applications du processus et contrôle de la production\n For this product series, we offer a variety of components and an extensive range of accessories.\r\n  read more\nPCA 100M\nVideo-based portable contact angle measuring instrument.\r\nWhether it be for large or small substrate areas, the PCA 100M is the ideal, handheld, portable instrument for controlling the qualities of coatings, examining the surface energies of pre-treated polymer films in the production process, or testing the qualities of cleaned or adhesion promoted glass surfaces, prior to the next processing stage. The compact design and the âplug and playâ connectivity, to any notebook or on-site available PC, without the need for a special frame grabber, enables the measurement of surface properties almost everywhere â including use in cleanrooms (US FED STD 209E cleanroom standard class 10).\r\nFeatures of the basic configuration:\r\n High-performance telecentric lens with integrated aperture and adjustable observation angle. Video measuring system (max. 50 images/s sample rate). Software controlled continually adjustable back lighting, without hysteresis. Integrated multiple dosing system, with either two dosing needles (PCA 100M/2) or four dosing needles (PCA 100M/4), for software controlled automatic needle selection and positioning. Integrated automatic refill system with flushing and cleaning functions for dosing up to 5000 drops per filling and per liquid receptacle. Power supply with automatic voltage adjustment.\n', '', 'page', '', '', 1429621722, 211, 187, 'Instruments de mesure d''angle de contact portable', 3, '', ''),
(178, 5, 1431355495, 1428385888, 0, 0, 0, '', '211', 'Process applications and production control\nFor this product series, we offer a variety of components and an extensive range of accessories.\r\n read more\nPCA 100M\nVideo-based portable contact angle measuring instrument.\r\nWhether it be for large or small substrate areas, the PCA 100M is the ideal, handheld, portable instrument for controlling the qualities of coatings, examining the surface energies of pre-treated polymer films in the production process, or testing the qualities of cleaned or adhesion promoted glass surfaces, prior to the next processing stage. The compact design and the âplug and playâ connectivity, to any notebook or on-site available PC, without the need for a special frame grabber, enables the measurement of surface properties almost everywhere â including use in cleanrooms (US FED STD 209E cleanroom standard class 10).\r\nFeatures of the basic configuration:\r\n High-performance telecentric lens with integrated aperture and adjustable observation angle. Video measuring system (max. 50 images/s sample rate). Software controlled continually adjustable back lighting, without hysteresis. Integrated multiple dosing system, with either two dosing needles (PCA 100M/2) or four dosing needles (PCA 100M/4), for software controlled automatic needle selection and positioning. Integrated automatic refill system with flushing and cleaning functions for dosing up to 5000 drops per filling and per liquid receptacle. Power supply with automatic voltage adjustment.\n', '', 'page', '', '', 1429621722, 211, 187, 'Portable contact angle measuring instruments', 2, '', ''),
(179, 5, 1431355495, 1428385888, 0, 0, 0, '', '211', 'Prozessanwendungen und Produktionskontrollen\nZu dieser Produktreihe bieten wir eine Vielzahl an Komponenten und ein umfangreiches Zubehörprogramm an.\r\n Ãbersicht\nPCA 100M\nPortables, videogestütztes Kontaktwinkelmessgerät\r\nOb für groÃe oder kleine Flächen, das PCA 100M ist das tragba­re Gerät, um die Qualität von Beschichtungen zu kontrollieren, die Oberflächenenergie von vorbehandelten Polymerschichten zu ermitteln oder die Eigenschaften von gereinigten oder haftungsoptimierten Oberflächen vor dem nächsten Pro­duk­tions­schritt zu testen.\r\nDie platzsparende Konstruktion und die âplug &amp; playâ-Verbindung zu jedem Notebook oder vor Ort verfügbarem PC ermöglicht die Messung von Oberflächeneigenschaften fast überall â sogar in Reinräumen.\r\nDas PCA 100M ermöglicht die Reduktion subjektiver Faktoren und des Zeitauf­wandes für Kontaktwinkelmessungen in der Qualitäts- und Produktionskontrolle.\r\nFür die softwaregesteuerte Messung und Auswertung:\r\n des statischen und dynamischen Kontaktwinkels nach der Sessile Drop-Methode des Benetzungsverhaltens von Festkörperoberflächen der freien Oberflächenenergie von Festkörpern und deren Komponenten der Oberflächenspannung aus der Tropfenkonturanalyse\r\nDas PCA 100M besteht aus dem Grundgerät mit folgender technischer Ausstattung:\r\n telezentrisches Hochleistungsobjektiv mit integrierter Blende und anpassbarem Betrachtungswinkel hysteresefreie Hintergrundbeleuchtung mit softwaregesteuerter Intensitäts­anpassung Videomesssystem (maximale Bildaufnahmerate 52 Bilder/s) integriertes Mehrfachdosiersystem, wahlweise mit zwei (PCA 100M/2) oder vier Dosierkanülen (PCA 100M/4) softwaregesteuerte automatisierte Kanülenauswahl und -positionierung integriertes automatisches Nachfüllsystem mit Spül- und Reinigungsfunktion für die Dosierung von bis zu 5000 Tropfen pro Flüssigkeitsbehälter Netzteil mit automatischer Spannungsanpassung\n', '', 'page', '', '', 1429621722, 211, 187, 'Portable Kontaktwinkelmessgeräte', 0, '', ''),
(180, 5, 1431355495, 1428385888, 0, 0, 0, '', '214', 'Applications du processus et contrôle de la production\n For this product series, we offer a variety of components and an extensive range of accessories.\r\n  read more\nPCA 100R\nRobotically guided contact angle measuring instrument for process control.\r\nIn combination with the single and multi axis cartesian robots of the RCC series, the PCA 100R contact angle measuring instrument is a specialised system for measuring wetting properties on substrates of very large surface areas, on site, during the production process. The fully automatic instrument enables the removes operator variability and other subjective factors and of the time necessary for the contact angle measurement. Some important applications for the robotically guided PCA 100R are:\r\n determination of the surface cleanliness of wafers and glass substrates for flat screens before the next manufacturing step. examining the surface energies of pre-treated polymer films in the production process. development of high-performance composites. surface finishing and the cleaning of textiles.\r\n The PCA 100R and the robots of the RCC series are designed for the usage in cleanrooms according the US FED STD 209E cleanroom standard class 10.\r\n\r\nFeatures of the basic instrument configuration:\r\n measuring head with integrated optics, illumination and multiple dosing unit. high-performance telecentric lens with integrated aperture and adjustable observation angle. video-based measuring system with high-resolution CCD camera and high-performance digitizing adapter. software controlled continuous adjustable intensity without hysteresis for a homogeneous back lighting. automatic needle selection and positioning.\r\n The consequent expansion of our accessories range allows us to adapt or to refine the robotically guided contact angle measuring instrument for your application. If we aroused your interest, please give us the chance find or develop a&nbsp;robotically guided contact angle measuring instrument ideal for your process. One of our strengths is the customisation of our products to your needs âplease&nbsp; call directly&nbsp;or fill out the&nbsp; contact form.\n', '', 'page', '', '', 1429621778, 214, 187, 'Instruments de mesure avec des systèmes de robot', 3, '', ''),
(181, 5, 1431355495, 1428385888, 0, 0, 0, '', '214', 'Process applications and production control\n For this product series, we offer a variety of components and an extensive range of accessories.\r\n  read more\nPCA 100R\nRobotically guided contact angle measuring instrument for process control.\r\nIn combination with the single and multi axis cartesian robots of the RCC series, the PCA 100R contact angle measuring instrument is a specialised system for measuring wetting properties on substrates of very large surface areas, on site, during the production process. The fully automatic instrument enables the removes operator variability and other subjective factors and of the time necessary for the contact angle measurement. Some important applications for the robotically guided PCA 100R are:\r\n determination of the surface cleanliness of wafers and glass substrates for flat screens before the next manufacturing step. examining the surface energies of pre-treated polymer films in the production process. development of high-performance composites. surface finishing and the cleaning of textiles.\r\n The PCA 100R and the robots of the RCC series are designed for the usage in cleanrooms according the US FED STD 209E cleanroom standard class 10.\r\n\r\nFeatures of the basic instrument configuration:\r\n measuring head with integrated optics, illumination and multiple dosing unit. high-performance telecentric lens with integrated aperture and adjustable observation angle. video-based measuring system with high-resolution CCD camera and high-performance digitizing adapter. software controlled continuous adjustable intensity without hysteresis for a homogeneous back lighting. automatic needle selection and positioning.\r\n The consequent expansion of our accessories range allows us to adapt or to refine the robotically guided contact angle measuring instrument for your application. If we aroused your interest, please give us the chance find or develop a&nbsp;robotically guided contact angle measuring instrument ideal for your process. One of our strengths is the customisation of our products to your needs âplease&nbsp; call directly&nbsp;or fill out the&nbsp; contact form.\n', '', 'page', '', '', 1429621778, 214, 187, 'Robott-assisted contact angle measuring instruments', 2, '', ''),
(182, 5, 1431355495, 1428385888, 0, 0, 0, '', '214', 'Prozessanwendungen und Produktionskontrollen\nZu dieser Produktreihe bieten wir eine Vielzahl an Komponenten und ein umfangreiches Zubehörprogramm an.\r\n Ãbersicht\nPCA 100R\nRobotergeführtes Kontaktwinkelmessgerät für die Prozesskontrolle\r\nIn Verbindung mit den Einzel- und Mehrachsen-Robotern der RCC- Serie ist das PCA 100R-Kontaktwinkelmessgerät ein spezialisiertes System für die Messung von Benetzungseigenschaften auf sehr groÃen Trägermaterialien während des Produktionsprozess vor Ort. Das vollautomatisch gesteuerte Gerät ermöglicht eine Reduzierung des subjektiven Faktors und der für die Kontaktwinkelmessungen benötigten Zeit.\r\nEinige wichtige Anwendungen für das PCA 100R-Kontakt­winkel­messgerät sind:\r\n Ermittlung der Oberflächenreinheit von Wafern und Glasträgern für Flachbildschirme vor dem nächsten Produktionsschritt Prüfung der Oberflächenenergie von vorbehandelten Polymerschichten im Produktionsprozess Entwicklung von Hochleistungsverbundstoffen Oberflächenveredelung und Reinigung von Textilien Das PCA 100R und die Roboter der RCC-Serie sind für eine Nutzung in Reinräumen nach der &quot;US FED STD 209E Reinraumklasse 10&quot; ausgelegt.\r\nDas PCA 100R besteht aus dem Grundgerät mit folgender technischer Ausstattung:\r\n Messkopf mit integrierter Optik, Beleuchtung und Mehrfachdosiersystem lichtstarkes, telezentrisches Messobjektiv mit integrierter Blende Videomesssystem mit hochauflösender CCD-Kamera Hochleistungs-Digitalisierkarte (Framegrabber) für den Einbau in ein PC-System Beleuchtungselektronik mit steuerbarer Lichtintensität für eine homogene Hintergrundbeleuchtung Mess- und Steuerungselektronik für die Auswahl und Positionierung der Dosiernadeln Netzteil mit automatischer Spannungsanpassung\r\nDie konsequente Weiterführung des Baukastenprinzips ermöglicht es uns, das für Ihren Anwendungsfall am besten geeignete robotergestützte Kontaktwinkelmessgerät innerhalb kürzester Zeit anzupassen oder weiterzuentwickeln.\r\nHaben wir Ihr Interesse geweckt, so geben Sie uns die Chance, ein für Ihre Bedürfnisse ideales und preiswertes Robotersystem zu finden oder zu entwickeln. Bitte sprechen Sie uns direkt an oder füllen Sie das Kontaktformular aus.\n', '', 'page', '', '', 1429621778, 214, 187, 'Robotergeführte Kontaktwinkelmessgeräte', 0, '', ''),
(183, 5, 1431355495, 1428385888, 0, 0, 0, '', '217', 'DCAT accessories range\nFor this product series, we offer a variety of components and an extensive range of accessories.\r\n DCAT accessories\nDCAT 11EC\nThe&nbsp;dynamic contact angle measuring instrument&nbsp;and&nbsp;tensiometer&nbsp;DCAT&nbsp;11EC is the budget-priced instrument for the weight-based measurement of contact angle, surface and interfacial tension, critical micelle formation concentration, density, sedimentation and penetration rate, plus penetration resistance. Features of the basic instrument configuration:\r\n High precision electro dynamic compensation weighing system (weighing&nbsp; range: 100 µg to 210 g) with automatic calibration. Software controlled height positioning (travel: 74 mm; resolution: 1 µm) of the sample receptacles with variable speed (2 µm/s to 60 mm/min) Automatic electronic coupling lock for the balance. Integrated measurement and control electronics with connections for two Pt 100 temperature sensors (-60 to +450 °C; ± 0.01 K resolution; precision 1/3 DIN IEC 751, Class B). Display for weighing data, temperature and other information. Illuminated sample chamber with inert gas or vapor inlet. USB interface with connection cable to PC/notebook.\r\n All DCAT accessories can be used with the DCAT 11EC except the electrical temperature control unit TEC 250/DCAT.\r\n Data Sheet DCAT 11 EC\n', '', 'page', '', '', 1429691680, 217, 189, 'DCAT 11EC', 2, '', ''),
(184, 5, 1431355495, 1428385888, 0, 0, 0, '', '217', 'Komponenten und Zubehör\nZu dieser Produktreihe bieten wir eine Vielzahl an Komponenten und ein umfangreiches Zubehörprogramm an.\r\n DCAT Zubehör\nDCAT 11EC\nDas dynamische Kontaktwinkelmessgerät und Tensiometer DCAT 11EC ist das preiswerte Gerät für die gewichtsbasierte Messung von Kontaktwinkeln, Ober- und Grenzflächenspannung, kritischer Mizellbildungskonzentration (CMC), Dichte, Sedimentations- und Penetrationsgeschwindigkeit, Penetrationskraft sowie des Penetrationswiderstands.\r\nDas DCAT 11EC besteht aus dem Grundgerät mit folgender technischer Ausstattung:\r\n hochpräzises elektrodynamisches Kompensations-Wägesystem (Wägebereich: 100 µg â¦ 210 g) mit automatischer Kalibrierfunktion softwaregesteuerte Höhenpositionierung (Verfahrweg: 74 mm; Auflösung: 1 µm) der ProbengefäÃe mit variabler Geschwindigkeit (2 µm/s â¦ 60 mm/min) automatische elektronisch gesteuerte Waagenarretierung integrierte Mess- und Steuerungselektronik mit Anschlüssen für zwei Pt100 Temperatursensoren (-60 â¦ +450 °C; 0,01 K Auflösung; Genauigkeit 1/3 DIN IEC 751, Klasse B) Display für Messwerte, Temperatur und Geräteinformationen beleuchteter Probenraum mit Inertgas- oder Dampfeinlass USB-Schnittstelle mit Verbindungskabel zum PC\r\nDas DCAT 11EC kann mit allem Zubehör aus dem DCAT-Baukasten erweitert werden, auÃer der elektrischen Thermostatisiereinrichtung TEC 250/DCAT.\r\n Datenblatt DCAT 11EC\n', '', 'page', '', '', 1429691680, 217, 189, 'DCAT 11EC', 0, '', ''),
(185, 5, 1431355495, 1428385888, 0, 0, 0, '', '218', 'DCAT 11\nThe&nbsp;dynamic contact angle measuring instrument&nbsp;and&nbsp;tensiometer&nbsp;DCAT&nbsp;11 is the standard instrument for the weight based measurement of contact angle, surface and interfacial tension, critical micelle formation concentration, density, sedimentation and penetration rate, plus penetration resistance. Features of the basic instrument configuration:\r\n High-precision electro dynamic compensation weighing system (weighing range: 10 µg&nbsp; to 210 g) with automatic calibration. Software controlled height positioning (travel: 74 mm; resolution: 0.1 µm) of the sample receptacles with variable speed (0.74 µm/s to 500 mm/min). Automatic electronic coupling lock for the balance. Integrated measurement and control electronics with connections for two Pt 100 temperature sensors (-60 to +450 °C; ± 0.01 K resolution; precision 1/3 DIN IEC 751, Class B). Display for weighing data, temperature and other information. Illuminated sample chamber with inert gas or vapour inlet. USB interface with connection cable to PC/notebook.\r\n All DCAT accessories can be used with the DCAT 11 except the electrical temperature control unit TEC 250/DCAT.\r\n Data Sheet DCAT 11\n', '', 'page', '', '', 1429691680, 218, 189, 'DCAT 11', 2, '', ''),
(186, 5, 1431355495, 1428385888, 0, 0, 0, '', '218', 'DCAT 11\nDas dynamische Kontaktwinkelmessgerät und Tensiometer DCAT 11 ist das Standardgerät für die gewichtsbasierte Messung von Kontaktwinkeln, Ober- und Grenzflächenspannung, kritischer Mizellbildungskonzentration (CMC), Dichte, Sedimentations- und Penetrationsgeschwindigkeit, Penetrationskraft sowie des Penetrationswiderstands.\r\nDas DCAT 11 besteht aus dem Grundgerät mit folgender technischer Ausstattung:\r\n hochpräzises elektrodynamisches Kompensations-Wägesystem (Wägebereich: 10 µg â¦ 210 g) mit automatischer Kalibrierfunktion softwaregesteuerte, motorische Höhenpositionierung (Verfahrweg: 74 mm; Auflösung: 0,1 µm) der ProbengefäÃe mit variabler Geschwindigkeit (0,74 µm/s â¦ 500 mm/min) automatische elektronisch gesteuerte Waagenarretierung integrierte Mess- und Steuerungselektronik mit Anschlüssen für zwei Pt100 Temperatursensoren (-60 â¦ +450 °C; 0,01 K Auflösung; Genauigkeit 1/3 DIN IEC 751, Klasse B) Display für Messwerte, Temperatur und Geräteinformationen beleuchteter Probenraum mit Inertgas- oder Dampfeinlass USB-Schnittstelle mit Verbindungskabel zum PC\r\nDas DCAT 11 kann mit allem Zubehör aus dem DCAT-Baukasten erweitert werden, auÃer der elektrischen Thermostatisiereinrichtung TEC 250/DCAT.\r\n Datenblatt DCAT 11\n', '', 'page', '', '', 1429691680, 218, 189, 'DCAT 11', 0, '', ''),
(187, 5, 1431355495, 1428385888, 0, 0, 0, '', '219', 'DCAT 21\nThe&nbsp;dynamic contact angle measuring instrument&nbsp;and&nbsp;tensiometer&nbsp;DCAT&nbsp;21 is the all-purpose instrument for the weight-based measurement of contact angle, surface and interfacial tension, critical micelle formation concentration, density, sedimentation and penetration rate, plus penetration resistance. Besides this it allows the use of the optional electrical temperature control unit TEC 250/DCAT,&nbsp; due to the 30 mm longer travel for the sample stage than the DCAT 11 and DCAT 11EC. Features of the basic instrument configuration:\r\n High-precision electro dynamic compensation weighing system (weighing&nbsp; range: 10 µg to 210 g) with automatic calibration. Software controlled height positioning (travel: 105 mm; resolution: 0.1 µm) of the sample receptacles with variable speed (0.74 µm/s to 500 mm/min). Automatic electronic coupling lock for the balance. Integrated measurement and control electronics with connections for two Pt 100 temperature sensors (-60 to +450 °C; ± 0.01 K resolution; precision 1/3 DIN IEC 751, Class B). Display for weighing data, temperature and other information. Illuminated sample chamber with inert gas or vapour inlet. USB interface with connection cable to PC/notebook.\r\n All DCAT accessories can be used with the DCAT 21, including the electrical temperature control unit TEC 250/DCAT.\r\n Data Sheet DCAT 21\n', '', 'page', '', '', 1430731731, 219, 189, 'DCAT 21', 2, '', ''),
(188, 5, 1431355495, 1428385888, 0, 0, 0, '', '219', 'DCAT 21\nDas dynamische Kontaktwinkelmessgerät und Tensiometer DCAT 21 ist das Standardgerät für die gewichtsbasierte Messung von Kontaktwinkeln, Ober- und Grenzflächenspannung, kritischer Mizellbildungskonzentration (CMC), Dichte, Sedimentations- und Penetrationsgeschwindigkeit, Penetrationskraft sowie des Penetrationswiderstands.\r\nDas DCAT 21 besteht aus dem Grundgerät mit folgender technischer Ausstattung:\r\n hochpräzises elektrodynamisches Kompensations-Wägesystem (Wägebereich: 10 µg â¦ 210 g) mit automatischer Kalibrierfunktion softwaregesteuerte Höhenpositionierung (Verfahrweg: 105 mm; Auflösung: 0,1 µm) der ProbengefäÃe mit variabler Geschwindigkeit (0,74 µm/s â¦ 500 mm/min) automatische elektronisch gesteuerte Waagenarretierung integrierte Mess- und Steuerungselektronik mit Anschlüssen für zwei Pt100 Temperatursensoren (-60 â¦ +450 °C; 0,01 K Auflösung; Genauigkeit 1/3 DIN IEC 751, Klasse B) Display für Messwerte, Temperatur und Geräteinformationen beleuchteter Probenraum mit Inertgas- oder Dampfeinlass USB-Schnittstelle mit Verbindungskabel zum PC\r\nDas DCAT 21 kann mit allem Zubehör aus dem DCAT-Baukasten erweitert werden, auÃer der elektrischen Thermostatisiereinrichtung TEC 250/DCAT.\r\n Datenblatt DCAT 21\n', '', 'page', '', '', 1430731731, 219, 189, 'DCAT 21', 0, '', ''),
(189, 5, 1431355495, 1428385888, 0, 0, 0, '', '220', 'Temperature control units\nEach of the temperature control units, available to the DCAT tensiometers, are easy to install and are available for different sample vessel sizes and temperature ranges. TV 70&nbsp;Liquid temperature control unit\r\nTV 50 / 70 / 100\r\nLiquid temperature control unit with a built-in Pt 100 temperature measuring sensor. For temperature range of -10 to 130 °C, controlled by an external liquid circulator. Available in the following standard sizes:\r\n TV 50&nbsp;for sample vessels with 50 mm diameter. TV 70&nbsp;for sample vessels with 70 mm diameter. TV 100&nbsp;for sample vessels with 100 mm diameter.\r\n\r\nTV 70NM\r\nNon-magnetic (hard gold-plated brass) liquid temperature control unit, including a micro-electronic stirrer (Super flat removable magnetic stirrer without any motor), sealed and chemical resistant and a built-in Pt 100 temperature measuring sensor. For temperature range of -10 to 130 °C, controlled by an external liquid circulator.\r\n\r\nTEC 250/DCAT\r\nElectrical temperature control unit to use with the dynamic contact angle measuring device/tensiometer DCAT 21. For measurements at temperatures up to 250 °C under protective atmosphere, e.g. for measuring of resins, hot melt adhesives and solder. Manual or software controlled temperature setting over temperature controller TC 250. Diameter of sample container: 50 mm\n', '', 'page', '', '', 1429691903, 220, 190, 'Temperature control units', 2, '', ''),
(190, 5, 1431355495, 1428385888, 0, 0, 0, '', '220', ' Thermostatiserungseinrichtungen\nIn Kombination mit den DCAT-Systemen stehen den Anwendern mehrere Thermostatisiereinrichtungen zur Verfügung.\r\nTV 50 / 70 / 100\r\nEinfach austauschbare Flüssigkeits-Thermostatisiereinrichtung mit integriertem Magnetrührer und Pt100-Temperatursensor, für Temperaturen von - 10â¦130 °C zum Anschluss an einen Kälte-Umwälzthermostat.\r\nVerfügbar in folgenden StandardgröÃen:\r\n TV 50 für ProbengefäÃe mit 50 mm Durchmesser TV 70 für ProbengefäÃe mit 70 mm Durchmesser TV 100 für ProbengefäÃe mit 100 mm Durchmesser\r\nTV 70NM\r\nNichtmagnetische (MS, hartvergoldet) Flüssigkeits-Thermostatisiereinrichtung für ProbengefäÃe mit Durchmessern von 70 mm, mit herausnehmbarem mikro­elektronischen Rührsystem und eingebautem Pt100-Temperatursensor. Für Temperaturen von - 10â¦130 °C zum Anschluss an einen Kälte-Umwälzthermostat.\r\nTEC 250/DCAT\r\nElektrische Thermostatisiereinrichtung für den Einsatz mit dem dynamischen KontaktwinkelmeÃgerät und Tensiometer DCAT 21. Zur Messung von Flüssigkeiten bei Temperaturen bis 250 °C unter Schutzgas-Atmosphäre, z.B. für die Messung an Harzen, Schmelzklebern und Loten. Manuelle oder softwaregesteuerte Temperaturvorgabe über PID-Regler TC 250. Durchmesser ProbengefäÃ 50 mm\n', '', 'page', '', '', 1429691903, 220, 190, 'Thermostatisiereinrichtungen', 0, '', ''),
(191, 5, 1431355495, 1428385888, 0, 0, 0, '', '221', 'Liquid handling components\nFor the easy handling and programmed addition of liquids DataPhysics can design and deliver a near infinite number of possibilities. A selection of the most demanded you can find here:\r\nLDU 2/2&nbsp;Automatic liquid dosing and refill unit\r\nAutomatic liquid dosing and refill unit LDU\r\nMotor-driven and software controlled, connectable to the dynamic contact angle measuring instrument and tensiometer of the DCAT series. For the precise dosing of liquids into the sample vessel of the DCAT, e.g. for concentration increments, as well as for the automatic, software controlled refilling or rinsing of the dosing syringes. Available in four standard variants, adaptable to your requirements:\r\n LDU 1/0&nbsp;with one electronic syringe module ES 11. LDU 2/0&nbsp;with two electronic syringe module ES 11. LDU 1/1&nbsp;with one electronic syringe module ES 11 and one refill and rinse system RRS 11. LDU 2/2&nbsp;with two electronic syringe modules ES 11 and two refill and rinse systems RRS 11.\r\n The electronic syringe module comprising a gas tight 500 µl dosing syringe&nbsp;DS 500/GT&nbsp;and a PTFE dosing tube with special dosing needle&nbsp;DT 500/04. The refill and rinse system RRS 11 is consisting of an electromagnetic valve, a PTFE dosing tube&nbsp;DT 200/04, two PTFE dosing tubes DT 500/16&nbsp;and a manual three-way valve&nbsp;TWV-LDU.\r\nDS 500/GT\r\nGas tight 500 µl dosing syringe.\r\nTWV-LDU\r\nThree-way valve for LDU. Facilitates the bubble-free filling of the complete dosing system.\r\nDT 500/04\r\nDosing tube (Valve/Syringe to DCAT) made of PTFE with 500 mm length and 0.4 mm inner diameter.&nbsp; With Luer-Lock connector to the syringe DS or to the electromagnetic valve and a special dosing canulla made of stainless steel.\r\nDT 200/04\r\nDosing tube (Syringe to valve) made of PTFE with 200 mm length and 0.4 mm inner diameter.\r\nDT 500/16\r\nDosing tube (Bottle to valve) made of PTFE with 500 mm length and 1.6 mm inner diameter. GS 70 with CP 70 and GS 50 with CP 50\r\nGS 100 / GS 70 / GS 50\r\nSample vessels made of borosilicate glass with 100 mm, 70 mm, and 50 mm outer diameter.\r\nCP 100 / CP 70 / CP 50\r\nCover paltes made of POM for the sample vessels GS 100, GS 70, and GS 50. GTR 70\r\nGTR 70\r\nSample vessel for measurements under special atmospheres (e.g. inert gas) made of borosilicate glass.&nbsp; Cover plates made of stainless steel, with built-in Pt 100 temperature sensor and two quick coupling tube connectors for gas inlet and outlet. Outer diameter 70 mm, height 85 mm.\n', '', 'page', '', '', 1429701997, 221, 190, 'Liquid handling', 2, '', ''),
(192, 5, 1431355495, 1428385888, 0, 0, 0, '', '221', 'Liquid-Handling-Komponenten\nFür die Dosierung und das Handling von Flüssigkeiten für die Verwendung mit den DCAT-Messgeräten finden Sie hier alles, was Sie brauchen - und wenn doch nicht, lassen Sie es uns wissen (entweder direkt oder füllen Sie das&nbsp; Kontaktformular&nbsp;aus).\r\nElektrisches Dosier- und Refill-System LDU\r\nMotorisch betrieben und softwaregesteuert, für den AnschluÃ an die dynamischen KontaktwinkelmeÃgeräte und Tensiometer DCAT 11EC, DCAT 11 und DCAT 21. Für die hochgenaue Dosierung von Flüssigkeiten in die ProbengefäÃe des DCAT, z.B. zur Herstellung von Konzentrationsreihen, sowie zum automatischen, softwaregesteuertem Wiederbefüllen und Reinigen der Dosierspritzen.\r\nVerfügbar in vier Varianten, anpassbar an Ihren Bedarf:\r\n LDU 1/0&nbsp;mit einem elektronischen Dosiermodul ES 11 LDU 2/0&nbsp;mit zwei elektronischen Dosiermodulen ES 11 LDU 1/1&nbsp;mit einem elektronischen Dosiermodul ES 11 und einem Füll- und Spülsystem RRS 11 LDU 2/2&nbsp;mit zwei elektronischen Dosiermodulen ES 11 und zwei Füll- und Spülsystemen RRS 11\r\nDie elektronischen Dosiermodule beinhalten eine gasdichte 500 µl Dosierspritze DS 500/GT und einen PTFE-Schlauch mit Dosierspitze DT 500/04. Das Füll- und Spülsystem RRS 11 besteht aus einem elektronischen Mischventil, einem PTFE-Schlauch DT 200/04, zwei PTFE-Schläuchen DT 500/16 und einem manuellen Dreiwege-Ventil TWV-LDU.\r\nDS 500/GT\r\nGasdichte 500 µl-Präzisionsspritze\r\nTWV-LDU\r\nDreiwegehahn für LDU Erleichtert die blasenfreie Befüllung des Dosiersystems\r\nDT 500/04\r\nDosierschlauch (Ventil/Spritze - DCAT) aus PTFE mit 500 mm Länge und 0,4 mm Innendurchmesser mit AnschluÃ für Spritze&nbsp; DS oder Mischventil und einer Edelstahl-Dosierkanüle\r\nDT 500/16\r\nDosierschlauch (Reservoir - Ventil) aus PTFE mit 500 mm Länge und 1,6 mm Innendurchmesser\r\nGS 70 / GS 50\r\nProbengefäÃ aus Borosilikatglas mit 70 mm und 50 mm Durchmesser\r\nCP 70 / CP 50\r\nAbdeckplatten aus POM für ProbengefäÃ GS 70 und GS 50\r\nGTR 70\r\nProbengefäÃ für Messungen unter Schutzgasatmosphäre aus Borosilikatglas für die Verwendung in der Flüssigkeits-Thermostatisiereinrichtung TV 70. Mit zwei Abdeckplatten aus Edelstahl, zwei Schnellverschluss-Schlauchanschlüssen und einem eingebauten Pt 100-Temperatursensor. Aussendurchmesser 70 mm, Höhe 85 mm.\n', '', 'page', '', '', 1429701997, 221, 190, 'Liquid Handling', 0, '', ''),
(193, 5, 1431355495, 1428385888, 0, 0, 0, '', '222', 'Sample holders and measuring bodies\nA great variety of sample geometries can be adapted on the tensiometers of the DCAT series. Whether for films, plates, fibres, fibre bundles, woven and non-woven threads, pigments, powders, or for magnetic materials, you will find a suitable solution here.\r\nRG 11\r\nDu Noüy ring according to DIN 53914.\r\n Ring height: 25 mm. Ring diameter: 18.7 mm. Wire thickness: 0.37 mm.\r\n\r\nRG 10\r\nDu Noüy ring according to DIN 53914 for small sample volumes (down to&nbsp; 0.2 ml).\r\n Ring height: 6 mm. Ring diameter: 10 mm. Wire thickness: 0.3 mm.\r\nPT 9\r\nWilhelmy plate according to DIN 53914.&nbsp;\r\n Length: 10 mm. Width: 9.95 mm. Thickness: 0.2 mm.\r\n\r\nPT 11\r\n&nbsp;Wilhelmy-Platte aus Platin-Iridium nach DIN 53914.\r\n Length: 10 mm. Width: 19.9 mm. Thickness: 0.2 mm.\r\nPT 10\r\nCylindrical plate made of&nbsp; Platinum-Iridium for small sample volumes (&gt;0.2 ml).\r\n Inner diameter: 6.2 mm. Outer diameter: 6.6 mm. Cylinder height: 10 mm.\r\nDIS 11\r\nDensity determination set consisting of:\r\n Special holder made of Platinum-Iridium. Immersion body made of silicon.\r\nPP 11\r\nPenetrometer probe for measuring yield forces on soft gels, pastes etc.\r\nSC 11\r\nSedimentation cone for use in particle size and sedimentation rate analysis.\r\nPUR 11\r\nAccessories kit for measurements of powder samples for the measurement of adsorption and wetting behaviour of powder samples Consisting of:\r\n Special adapter for Sample vessel made of glass 1 Sample vessel made of glass for the measurement of powders\r\n\r\nGT 11\r\nSet of five sample vessels made of glass&nbsp;\r\nFP 11\r\nFilter paper for PUR 11\r\nFH 11\r\nSpecial sample holder to determine the wetting behaviour of adsorbing materials like powders, pigments, fibres and fibre bundles, textiles and non-vowen\r\nFP 12\r\nFilter paper for FH 11\r\nFO 11\r\nSpecial sample holder for films.\r\nPSH 11\r\nSpecial sample holder for solids in plate form.\r\nFH 12\r\nSample holder for the pick up of single fibres down to 10 µm diameter.\r\nFH 13\r\nSample holder for the pick up multiple single fibres down to 3 µm diameter.\r\nGlue FH 12/13\r\nLiquid glue for the replenishing of the adhesive surface of the FH 12 /FH 13.\r\n 1 bottle 20ml.\n', '', 'page', '', '', 1429702024, 222, 190, 'Sample holder / Test pieces', 2, '', ''),
(194, 5, 1431355495, 1428385888, 0, 0, 0, '', '222', 'Probenhalter und Messkörper\nAn die Tensiometer von DataPhysics kann eine Vielzahl unterschiedlicher Probengeometrien adaptiert werden. Ob für Folien, Platten, Fasern, Pulver oder für magnetische und gewebeartige Materialien - hier findet sich für die vielfältigen Bedürfnisse unserer Anwender immer etwas Passendes. Und falls nicht, fragen Sie uns (entweder direkt oder füllen Sie das&nbsp; Kontaktformular&nbsp;aus) nach Sondervarianten der Geräte und des Zubehörs. Meist können wir nach Kundenwunsch eine Anpassung durchführen.\r\nRG 11\r\nDu Noüy-Ring aus Platin-Iridium nach DIN 53914\r\n Ringhöhe: 25 mm Ringdurchmesser: 18,7 mm Drahtdicke: 0,37 mm\r\nRG 10\r\nDu Noüy-Ring aus Platin-Iridium für 0,2 ml Probenvolumen\r\n Ringhöhe: 6 mm Ringdurchmesser: 10 mm Drahtdicke: 0,3 mm\r\nPT 9\r\nWilhelmy-Platte aus Platin-Iridium nach DIN 53914\r\n Länge: 10 mm Breite: 9,95 mm Dicke: 0,2 mm\r\nPT 11\r\nWilhelmy-Platte aus Platin-Iridium nach DIN 53914\r\n Länge: 10 mm Breite: 19,9 mm Dicke: 0,2 mm\r\nPT 10\r\nZylindrische Platte aus Platin-Iridium für 0,2 ml Probenvolumen\r\n Innendurchmesser: 6,2 mm AuÃendurcmesser: 6,6 mm Zylinderhöhe: 10 mm\r\nDIS 11\r\nDichtebestimmungsset bestehend aus:\r\n Spezialhalter aus Platin-Iridium (80/20) Tauchkörper aus Silizium\r\nPP 11\r\nPrüfkörper für Penetrationsversuche zur Messung von FlieÃgrenzen in Gelen, Pasten usw.\r\nSC 11\r\nSedimentationskegel für die Analyse von PartikelgröÃen und Sedimentationsraten\r\nPUR 11\r\nZubehörsatz für die Messung der Adsorptions- und Benetzungseigenschaften von pulverförmigen Proben Bestehend aus:\r\n Spezialadapter für ProbengefäÃ aus Glas 1 ProbengefäÃ aus Glas für Pulvermessungen\r\nGT 11\r\nSatz mit fünf ProbengefäÃen aus Glas\r\nFH 11\r\nSpezial-Probenhalter zur Untersuchung des Benetzungsverhaltens von benetzbaren Materialien wie Faserbündel, Textilproben und Non-Woven\r\nGT 11\r\nSatz mit fünf ProbengefäÃen aus Glas\r\nFP 12\r\nFilterpapier für FH 11\r\nFO 11\r\nSpezial-Probenhalter für Folien\r\nPSH 11\r\nSpezial-Probenhalter für Festkörper in Plattenform\r\nFH 12\r\nProbenhalter für Einzelfasern zur Aufnahme von Einzelfasern mit einem Durchmesser von minimal 10 µm\r\nFH 13\r\nProbenhalter für Einzelfasern zur Aufnahme von mehreren Einzelfasern bis zu 3 µm Durchmesser.\r\nGlue FH 12/13\r\nFlüssiger Klebstoff für die Erneuerung der Klebefläche des FH 12 sowie FH 13\r\n 1 Flasche a 20ml \n', '', 'page', '', '', 1429702024, 222, 190, 'Probenhalter | Messkörper', 0, '', ''),
(195, 5, 1431355495, 1428385888, 0, 0, 0, '', '223', 'Software-Module\nSoftware für effizientes Arbeiten\r\nDie intuitiv bedienbare SCAT-Software unterstützt Sie in der Benutzung des DCAT durch die einfache Definition von Messabläufen und beim Erfassen und Auswerten von Daten. Die SCAT-Software kann als modulares Programm für die Betriebssysteme Windows 7®,&nbsp; Windows Vista®&nbsp;sowohl im 32- als auch im 64-bit Modus betrieben werden, unter Windows XP®&nbsp;/SP3 nur im 32-bit Modus. Die verfügbaren Module sind:SCAT 31\r\nSCAT 31\r\n Messung der statischen, zeit- und temperatur­abhängigen Ober- und Grenzflächenspannungen nach der Du Noüy-Ring-Methode und der Wilhelmy-Platten-Methode automatische Ring-Korrektur nach Zuidema &amp; Waters, Mason &amp; Huh und Harkins &amp; Jordan Gas-, Flüssigkeits- und Festkörperdatenbank\r\nSCAT 32\r\n kraftbasierte Bestimmung des dynamischen Kontaktwinkels prismatischer oder zylindrischer Festkörper (z. B. Platten, Filme, Stäbe und einzelne Fasern) sowie der benetzten Länge nach der Wilhelmy-Methode Adsorptionsmessung von Pulvern und Faserbündeln; Berechnung des gemittelten Kontaktwinkels nach der modifizierten und erweiterten Washburn-Methode Analyse der freien Oberflächenenergie sowie deren Komponenten (z. B. disperse, polare und Wasserstoffbrücken- bzw. Säure-/Base-Anteile) nach neun verschiedenen Theorien Berechnung und Darstellung von âWetting Envelopeâ- und Adhäsions/Kontaktwinkel-Diagrammen\r\nSCAT 33\r\n automatisierte Bestimmung der Kritischen Mizellbildungskonzentration (CMC) von Tensiden Berechnung der minimalen Oberflächenspannung aus Synergieeffekten von Tensidgemischen Berechnung des Platzbedarfs pro Molekül an der Grenzfläche Berechnung der freien Adsorptionsenergie nach Gibbs Berechnung des Oberflächenexzesses automatische Steuerung der Dosiereinheiten LDU x/x für additive und subtraktive Dosierung\r\nSCAT 34\r\n Dichtebestimmung von Flüssigkeiten\r\nSCAT 35\r\n Bestimmung der Sedimentationsgeschwindigkeit Messung der Penetrationskraft auf elastischen Gelen, Leim etc. Messung des Penetrationswiderstands und der Penetrationsgeschwindigkeit\n', '', 'page', '', '', 1429691903, 223, 190, 'Software-Module', 0, '', '');
INSERT INTO `tx_kesearch_index` (`uid`, `pid`, `tstamp`, `crdate`, `cruser_id`, `starttime`, `endtime`, `fe_group`, `targetpid`, `content`, `params`, `type`, `tags`, `abstract`, `sortdate`, `orig_uid`, `orig_pid`, `title`, `language`, `directory`, `hash`) VALUES
(196, 5, 1431355495, 1428385888, 0, 0, 0, '', '223', 'Software-Module\nIntuitive software to ensure the highest quality of data. \r\nThe SCAT software is designed as a modular program, available in various discrete usable modules. Every software module includes the control of the DCAT and its accessories, the measurement, the analysis, and the presentation of the results.Under Windows 7® / Vista® it works in 32- and in 64-bit mode; under Windows XP® only in 32-bit mode. The available software modules are:SCAT 31\r\nSCAT 31\r\n Measurement of the static, time and temperature dependent surface and interfacial tensions according to the Du Noüy-ring method, the Wilhelmy plate and the wire hoop method. Automatic ring corrections according to Zuidema &amp; Waters, Mason &amp; Huh and Harkins &amp; Jordan. Gas, liquids and solids data base.\r\nSCAT 32\r\n Force-based measurement of the dynamic contact angle of prismatic and cylindrical solids (e.g. plates, films, rods and single fibers) as well as the wetted length accor­ding to the Wilhelmy method. Adsorption measurement on powders and fibre bundles with the determination of the average contact angle according to the modified and the extended Washburn method. Analysis of the surface free energy of solids as well as their components (e.g. dispersive, polar and hydrogen bond parts, acid and base portions, respectively) according to nine different theories. Calculation of work of adhesion.\r\nSCAT 33\r\n Fully automatic determination of the critical micelle formation concentration (CMC) of surfactants in forward, reversed and extended mode. Calculation of the minimum surface tension in case of synergistic effects of surfactant mixtures. Calculation of the head space required by molecules on the surface. Calculation of the free adsorption energy according to Gibbs. Calculation of the surface excess. Automatic control of the dosing devices LDU x/x for additive and subtractive dosing.\r\nSCAT 34\r\n Determination of the density of liquids.\r\nSCAT 35\r\n Determination of the sedimentation rate. Measuring of the yield forces on soft gels, pastes etc. Measuring of the penetration resistance and penetration rate.\n', '', 'page', '', '', 1429691903, 223, 190, 'Software modules SCAT', 2, '', ''),
(197, 5, 1431355495, 1428385888, 0, 0, 0, '', '224', 'Contract measurements & more\nIn the DataPhysics''&nbsp;application centre for surface chemistry'' we are able to perform research and development work for our customers, on a commercial/confidential basis. Our experts will work with you to find the most effective, surface science based, solution for your specific problem, in conjunction with the most modern instrumentation available.\r\n\r\nExamples of the areas to which our application centre staff will apply their knowledge:\r\n1. Investigation of wetting properties and adhesion of solids and liquids:\r\n Paints and varnishes. Surfactants and Polymer solutions. Powders and fibres. Fabrics. Polymers films. Adhesives. Cosmetics. Coatings. Glass and ceramics. Microstructures.\r\n 2. Investigations of surface properties in a temperature range from -30°C up to 1800°C. 3. High speed measurement, in conjunction with the ''movie'' function at frame rates of up to 2200 images/s, for the investigations of extremely fast absorption and spreading processes. 4. The provision of detailed reports of investigations, including description of measurements and discussion of results.  5. Drafting of detailed measurement procedures, for transfer to customers own laboratory/instrumentation.&nbsp; 6. Complete support in arriving at the optimum test conditions for special applications.&nbsp; 7. Quick support on application related software questions.\n', '', 'page', '', '', 1429352452, 224, 12, 'Mesures contractuelles', 3, '', ''),
(198, 5, 1431355495, 1428385888, 0, 0, 0, '', '224', 'Contract measurements & more\nIn the DataPhysics''&nbsp;application centre for surface chemistry'' we are able to perform research and development work for our customers, on a commercial/confidential basis. Our experts will work with you to find the most effective, surface science based, solution for your specific problem, in conjunction with the most modern instrumentation available.\r\n\r\nExamples of the areas to which our application centre staff will apply their knowledge:\r\n 1. Investigation of wetting properties and adhesion of solids and liquids:\r\n Paints and varnishes. Surfactants and Polymer solutions. Powders and fibres. Fabrics. Polymers films. Adhesives. Cosmetics. Coatings. Glass and ceramics. Microstructures.\r\n 2. Investigations of surface properties in a temperature range from -30°C up to 1800°C. 3. High speed measurement, in conjunction with the ''movie'' function at frame rates of up to 2200 images/s, for the investigations of extremely fast absorption and spreading processes. 4. The provision of detailed reports of investigations, including description of measurements and discussion of results.  5. Drafting of detailed measurement procedures, for transfer to customers own laboratory/instrumentation.&nbsp; 6. Complete support in arriving at the optimum test conditions for special applications.&nbsp; 7. Quick support on application related software questions.\n', '', 'page', '', '', 1429352452, 224, 12, 'Contract Measurements', 2, '', ''),
(199, 5, 1431355495, 1428385888, 0, 0, 0, '', '225', 'Surface phenomena\nWhy are some insects able to skate on the surface of water while others would sink?&nbsp; Why in some cases, does the water sprinkled on a glass surface collect into drops and in other cases spread like a thin film?&nbsp;\r\nWhy does water climb up a thin tube?\r\nWhy can you make bubbles with soapy water and not with tap water?\r\nTo answer these questions it is important to appreciate and quantify the energy balance at the respective surface or interface.\nApplication notes\nOur relatively limited application note database contains discussions of some of the most common practical uses of surface science instrumentation. For more details on specific applications you may also refer to our&nbsp; literature support.\r\n\r\nApplication notes regarding our OCA-series:\r\n 1 - The pendant drop method  2 - Absorption measurements  3 - Interfacial tension between a printing ink and fountain water  4 - Determination of surface free energies of polymers  5 - Surface tension of molten polymers and hot-melts  6 - Determination of the surface free energy by Wu  13 - Use of the Wetting Envelope  14 - Determination of the maximum work of adhesion of certain liquids&nbsp;  17 - Calculation of a new reference liquid on a known surface  21 - Contact Angle Measurements on single fibres  22 - Dynamic Contact Angle Measurements on curved surfaces\r\n\r\nApplication notes regarding our DCAT-series:\r\n 10 - Determination of surface free energy of powders  11 - Determination of Critical Micelle Concentration (CMC)  13 - Use of the Wetting Envelope  14 - Determination of the maximum work of adhesion of certain liquids&nbsp;  15 - Determination of the dynamic contact angle on single fibres  24 - Protein Adsorption\r\n\r\nApplication notes regarding our SVT20N:\r\n 20 - Validation of the Spinning Drop Tensiometer SVT20N  23 - Determination of low interfacial tensions\r\n \n', '', 'page', '', '', 1429536992, 225, 12, 'Rapports dâapplications', 3, '', ''),
(200, 5, 1431355495, 1428385888, 0, 0, 0, '', '225', 'Surface phenomena\nWhy are some insects able to skate on the surface of water while others would sink?&nbsp; Why in some cases, does the water sprinkled on a glass surface collect into drops and in other cases spread like a thin film?&nbsp;\r\nWhy does water climb up a thin tube?\r\nWhy can you make bubbles with soapy water and not with tap water?\r\nTo answer these questions it is important to appreciate and quantify the energy balance at the respective surface or interface.\nApplication notes\nOur relatively limited application note database contains discussions of some of the most common practical uses of surface science instrumentation. For more details on specific applications you may also refer to our&nbsp; literature support.\r\n\r\nApplication notes regarding our OCA-series:\r\n 1 - The pendant drop method  2 - Absorption measurements  3 - Interfacial tension between a printing ink and fountain water  4 - Determination of surface free energies of polymers  5 - Surface tension of molten polymers and hot-melts  6 - Determination of the surface free energy by Wu  13 - Use of the Wetting Envelope  14 - Determination of the maximum work of adhesion of certain liquids&nbsp;  17 - Calculation of a new reference liquid on a known surface  21 - Contact Angle Measurements on single fibres  22 - Dynamic Contact Angle Measurements on curved surfaces\r\n\r\nApplication notes regarding our DCAT-series:\r\n 10 - Determination of surface free energy of powders  11 - Determination of Critical Micelle Concentration (CMC)  13 - Use of the Wetting Envelope  14 - Determination of the maximum work of adhesion of certain liquids&nbsp;  15 - Determination of the dynamic contact angle on single fibres  24 - Protein Adsorption\r\n\r\nApplication notes regarding our SVT20N:\r\n 20 - Validation of the Spinning Drop Tensiometer SVT20N  23 - Determination of low interfacial tensions\r\n\r\n\n', '', 'page', '', '', 1429536992, 225, 12, 'Application notes', 2, '', ''),
(201, 5, 1431355495, 1428385888, 0, 0, 0, '', '226', '\nTraining and Seminars\r\nPlease&nbsp; contact us&nbsp;to discuss the provison of training, in order that we might optimise the impact that your purchase of DataPhysics instrumentation will have on your business. Our application centre will be happy to respond to you.\n\nTraining, Seminars, Workshops\r\nWe offer, application specific, training and seminars either within your familiar work environment or at our own facility in Germany. Yourself and your team will be trained in both practical and theoretical aspects of the relevant science, using the relevant instrumentation and developed methods. Application specific training, on your instrument/at your location, or our application laboratory guarantees the most effective support on all relevant aspects of your specific applications. Additionally we provide telephone, e-mail and remote access support on the development and application of measurement procedures and the quickest possible support for all, application related, software questions.\n', '', 'page', '', '', 1430731565, 226, 12, 'Formation | séminaires | ateliers', 3, '', ''),
(202, 5, 1431355495, 1428385888, 0, 0, 0, '', '226', '\nTraining and Seminars\r\nPlease&nbsp; contact us&nbsp;to discuss the provison of training, in order that we might optimise the impact that your purchase of DataPhysics instrumentation will have on your business. Our application centre will be happy to respond to you.\n\nTraining, Seminars, Workshops\r\nWe offer, application specific, training and seminars either within your familiar work environment or at our own facility in Germany. Yourself and your team will be trained in both practical and theoretical aspects of the relevant science, using the relevant instrumentation and developed methods. Application specific training, on your instrument/at your location, or our application laboratory guarantees the most effective support on all relevant aspects of your specific applications. Additionally we provide telephone, e-mail and remote access support on the development and application of measurement procedures and the quickest possible support for all, application related, software questions.\n', '', 'page', '', '', 1430731565, 226, 12, 'Training | Seminars | Workshops', 2, '', ''),
(203, 5, 1431355495, 1428385888, 0, 0, 0, '', '227', '\n Imagination is more important than knowledge. For knowledge is limited to all we now know and understand, while imagination embraces the entire world, and all there ever will be to know and understand. Albert Einstein Sometimes the step(s) to an increase in ''knowledge'' are aided by the application of a little ''imagination''. DataPhysics promote and encourage knowledge AND imagination in all aspects of surface science.&nbsp; All software packages include a gas, liquids and solids database containing relevant information on more than 170 materials.&nbsp;\n\nDatabase information\r\n Data relating to the surface tension of liquids and the surface free energies of solids together with references to application related publications can be provided on&nbsp; request. A short list of the available data can also be found under&nbsp; www.surface-tension.de. We will be pleased draw upon the detail contained within our database in order to respond to your detailed questions. We have access to much collated information on specific measuring regimes, such as high or low temperature, high pressure or vacuum. We will happily help you locate the relevant references.\n', '', 'page', '', '', 1429623612, 227, 12, 'Banques de données', 3, '', ''),
(204, 5, 1431355495, 1428385888, 0, 0, 0, '', '227', '\nImagination is more important than knowledge. For knowledge is limited to all we now know and understand, while imagination embraces the entire world, and all there ever will be to know and understand.\r\nAlbert Einstein\r\nSometimes the step(s) to an increase in ''knowledge'' are aided by the application of a little ''imagination''. DataPhysics promote and encourage knowledge AND imagination in all aspects of surface science.&nbsp;\r\nAll software packages include a gas, liquids and solids database containing relevant information on more than 170 materials.&nbsp;\n\nDatabase information\r\nData relating to the surface tension of liquids and the surface free energies of solids together with references to application related publications can be provided on&nbsp; request.\r\nA short list of the available data can also be found under&nbsp; www.surface-tension.de. We will be pleased draw upon the detail contained within our database in order to respond to your detailed questions. We have access to much collated information on specific measuring regimes, such as high or low temperature, high pressure or vacuum. We will happily help you locate the relevant references.\n', '', 'page', '', '', 1429623612, 227, 12, 'Database', 2, '', ''),
(205, 5, 1431355495, 1428385888, 0, 0, 0, '', '228', '\nWe may know, or know, who knows\r\nAn individual cannot be expected to have all relevant application information at his/her fingertips. Finding a reliable source of complementary knowledge may by key to the success of your projects. DataPhysics and its carefully selected sales/support partners will happily share the knowledge gained during decades of experience in this varied field of study.\n\nLiterature Support\r\nWe administer an extensive database of electronic publications in which surface chemistry takes a prominent role. We will happily share application specific information to aid your journey through this continually developing field. If you would like our help in tracking down published data&nbsp; Please contact us.\r\nLiterature list\r\nBelow here is a selection of popular literature in which colloid and surface science feature prominently:\r\nBooks\r\n E. Matijevic (Ed.):&nbsp;Surface and Colloid Science, Vol.1-15, Wiley Interscience &amp; Plenum Publ. Co., 1969-1993, ISBN (Vol. 1) 0-4715-7630-1 ... ISBN (Vol. 15) 0-3064-4150-0, ISSN 0081-9573 K.S. Birdi (Ed.):&nbsp;Handbook of Surface and Colloid Chemistry, CRC Press, Boca Raton, New York, 1997, ISBN 0-8493-945-7; 2nd Ed., 2002, ISBN 0-8493-1079-2 A.W. Adamson, A.P. Gast:&nbsp;Physical Chemistry of Surfaces, 6th Ed., John Wiley &amp; Sons, New York, 1997, ISBN 0-417-14873-3 D.J. Shaw:&nbsp;Introduction to Colloid and Surface Chemistry, 3rd Ed., Butterworths, London 1980, ISBN 0-408-71049-7 M.J. Schwuger:&nbsp;Lehrbuch der Grenzflächenchemie, Georg Thieme Verlag, Stuttgart, New York, 1996, ISBN 3-13-137501-9 G. Brezesinski, H.-J. Mögel:&nbsp;Grenzflächen und Kolloide, Spektrum Verlag, Heidelberg, Berlin, Oxford, 1993, ISBN 3-86025-016-7 H.-D. Dörfler:&nbsp;Grenzflächen- und Kolloidchemie, VCH Verlag, Weinheim, 1994, ISBN 3-527-29072-9 H.-D. Dörfler:&nbsp;Grenzflächen und kolloid-disperse Systeme - Physik und Chemie, Springer-Verlag, Berlin, Heidelberg, 2002, ISBN 3-540-42547-0 P.C. Hiemenz, R. Rajagopalan:&nbsp;Principles of Colloid and Surface Chemistry, 3rd Ed., Marcel Dekker, New York, 1997, ISBN 0-8247-9397-8 D. Myers:&nbsp;Surfaces, Interfaces, and Colloids, 2nd Ed., Wiley-VCH, New York, 1999, ISBN 0-471-33060-4 M.J. Jaycock, G.D. Parfitt:&nbsp;Chemistry of Interfaces, Ellis Horwood, Chichester, 1981, ISBN 0-470-27013-6 F. MacRitchie:&nbsp;Chemistry at Interfaces, Academic Press, San Diego, 1990, ISBN 0-12-464785-5 D.F. Evans, H. Wennerström:&nbsp;The Colloidal Domain - Where Physics, Chemistry, Biology, and Technology meet, VCH, New York, 1994, ISBN 1-56081-525-6 J. Lyklema:&nbsp;Fundamentals of Interface and Colloid Science: Solid-Liquid Interfaces, Academic Press, New York, 1995, ISBN 0-1246-0524-9 R.J. Hunter:&nbsp;Introduction to Modern Colloid Science, Oxford Univ. Press, Oxford, 1993, ISBN 0-19-855387-0 J.N. Israelachvili:&nbsp;Intermolecular and Surface Forces, 2nd Ed., Academic Press, New York, 1991, ISBN 0-1237-5181-0 C.J. van Oss:&nbsp;Interfacial Forces in Aqueous Media, Marcel Dekker, New York, 1994, ISBN 0-8274-9168-1 J.S. Rowlinson, B. Widom:&nbsp;Molecular Theory of Capillarity, Clarendon Press, Oxford, 1989, ISBN 0-19-855642-X R. Finn:&nbsp;Equilibrium Capillary Surfaces, Springer Verlag, New York, 1986, ISBN 0-387-96174-7 A.I. Rusanov, F.C. Goodrich (Eds.):&nbsp;The Modern Theory of Capillarity, Akademie Verlag, Berlin, 1981 S. Hartland, R.W. Hartley:&nbsp;Axissymmetric Fluid-Liquid Interfaces, Elsevier, Amsterdam, 1976, ISBN 0-4444-1396-0 K.L. Mittal (Ed): Contact Angle,&nbsp;Wettability and Adhesion, VSP, Utrecht, 1993, ISBN 90-6764-157-X J.C. Berg:&nbsp;Wettability, Marcel Dekker, New York, 1993, ISBN 0-8247-9046-0 J. F. Padday (Ed.):&nbsp;Wetting, Spreading and Adhesion, Academic Press, New York, 1978, ISBN 0-1254-4350-1 M.E. Schrader, G. Loeb (Eds):&nbsp;Modern Approaches to Wettability, Plenum Press, New York, 1992, ISBN 0-3064-3985-9 A.I. Rusanov, V.A. Prokhorov:&nbsp;Interfacial Tensiometry, Elsevier, Amsterdam, 1996, ISBN 0-444-92128-7 K.L. Mittal (Ed):&nbsp;Adsorption at Interfaces, 5th Ed., Wiley Interscience, New York, 1990, ISBN 0-8412-0249-4 S.S. Dukhin, G. Kretschmar, R. Miller:&nbsp;Dynamics of Adsorption at Liquid Interfaces, Elsevier, Amsterdam 1995, ISBN 0-444-88117-4 C.A. Miller, P. Neogi:&nbsp;Interfacial Phenomena, Marcel Dekker, New York, 1985, ISBN 0-8274-7490-6 C. Isenberg:&nbsp;The Science of Soap Films and Soap Bubbles, Dover Publ., New York, 1992, ISBN 0-486-26960-4 D. Möbius, R. Miller:&nbsp;Drops and Bubbles in Interfacial Research, Elsevier, Amsterdam, 1998, ISBN 0-444-82894-X N.R. Morrow (Ed):&nbsp;Interfacial Phenomena in Petroleum Recovery, Marcel Dekker, New York, 1990, ISBN 0-8274-8385-9 C.A. Croxton:&nbsp;Fluid Interfacial Phenomena, John Wiley, Chichester, 1986, ISBN 0-4719-0757-X D.K. Chattoraj, K.S. Birdi:&nbsp;Adsorption and the Gibbs Surface Excess, Plenum Press, New York, 1984, ISBN 0-306-41334-5 K. Tsujii:&nbsp;Surface Activity - Principles, Phenomena, and Applications, Academic Press, San Diego, 1998, ISBN 0-12-702280-5 H.-F. Eicke:&nbsp;Interfacial Phenomena in Apolar Media, Marcel Dekker, New York, 1987, ISBN 0-8247-7506-6 T.F. Tadros:&nbsp;Surfactants, Academic Press, London, 1984, ISBN 0-12-682180-1 D. Myers:&nbsp;Surfactant Science and Technology, VCH Publishers, New York, 1988, ISBN 0-89573-339-0 K. Kosswig, H. Stache (Eds.):&nbsp;Die Tenside, C. Hanser Verlag, München, Wien, 1993, ISBN 3-446-16201-1 N.M. van Os, J.R. Haak, L.A.M. Rupert:&nbsp;Physio-chemical Properties of Selected Anionic, Cationic and Nonionic Surfactants, Elsevier, Amsterdam, 1993, ISBN 0-444089691-0 M.J. Rosen:&nbsp;Surfactants and Interfacial Phenomena, 2nd Ed., John Wiley, New York, 1989, ISBN 0-471-83651-6 M.J. Rosen (Ed.):&nbsp;Structure performance relationships in surfactants, ACS, Washington, DC, 1984, ISBN 0-8412-0839-5 K.L. Mittal (Ed.):&nbsp;Surfactants in Solution, Marcel Dekker, New York, 1996, ISBN 0-8247-9792-2 P.M. Holland, D.N. Rubingh:&nbsp;Mixed Surfactant Systems, ACS, Washington D.C., 1992, ISBN 0-8412-2468-4 A.I. Rusanov:&nbsp;Micellization in Surfactant Solution, Gordon &amp; Breach, New York, 1999, ISBN 90-5702-297-4 K. Holmberg, B. Jönsson, B. Lindman:&nbsp;Surfactants and Polymers in Aqueous Solution, 2nd. Ed., John Wiley &amp; Sons, Ltd, Chichester, 2003, ISBN 0-471-49883 J.D. Andrade (Ed.):&nbsp;Surface and Interfacial Aspects of Biomedical Polymers, Vol.1: Surface chemistry and physics, Vol. 2: Protein adsorption, Plenum Press, New York, 1985, ISBN 0-306-41741-3 &amp; 0-306-41742-1 M. Bender (Ed.):&nbsp;Interfacial Phenomena in Biological Systems, Marcel Dekker, New York, 1991, ISBN 0-8247-8436-7 D. Möbius, R. Miller:&nbsp;Proteins at Interfaces, Elsevier, Amsterdam, 1998, ISBN 0-4448-2944-X S. Magdassi:&nbsp;Surface Activity of Proteins: Chemical and Physicochemical Modifications, Dekker, New York, 1996, ISBN 0-8247-9532-6 M.N. Jones:&nbsp;Micelles, Monolayers, and Biomembranes, Wiley-Liss, New York, 1994, ISBN 0-471-56139-8 V.N. Kazakov, O.V. Sinyachenko, V.B. Fainerman, U.Pison, R. Miller:&nbsp;Dynamic Surface Tensiometry in Medicine, Elsevier, Amsterdam, 2000, ISBN 0-4445-0411-7 I. Noda, D.N. Rubingh:&nbsp;Polymer Blends, Solutions, and Interfaces, Elsevier, Amsterdam, 1992, ISBN 0-4448-9397-0 S.Wu:&nbsp;Polymer Interface and Adhesion, Marcel Dekker, New York, 1982, ISBN 0-8247-1533-0 K.L. Mittal (ed.):&nbsp;Polymer Surface Modification: Relevance to Adhesion, VSP, Utrecht, 1996, ISBN 90-6764-201-0 M.R. Piggott (Ed.):&nbsp;Interfaces in Composites, Elsevier, Amsterdam, 1991, ISBN 1-85166-667-2 K.L. Mittal, H.R. Anderson (Eds):&nbsp;Acid-Base Interactions: Relevance to Adhesion Science and Technology, VSP, Utrecht 1991, ISBN 90-6764-135-9 L.H. Lee (Ed):&nbsp;Fundamentals of Adhesion, Plenum Press, New York, 1990, ISBN 0-3064-3470-9 W.C. Wake:&nbsp;Adhesion and Formulation of Adhesives, Elsevier, New York, 1982, ISBN 0-8533-4134-1 I. Skeist:&nbsp;Handbook of Adhesives, 3rd Ed., Chapman &amp; Hall, New York, 1990, ISBN 0-412-09681-1 A.J. Kinloch:&nbsp;Adhesion and Adhesives, Chapman &amp; Hall, London, 1987, ISBN 0-412-27440-X A.F.M. Barton:&nbsp;Handbook of Solubility Parameters and Other Cohesion Parameters, CRC Press, Boca Raton, New York, 1991, ISBN 0-8493-0176-9 C.M. Hansen:&nbsp;Hansen Solubility Parameters, CRC Press, Boca Raton, 2000, ISBN 0-8493-1525-5 H.J. Jakobasch:&nbsp;Oberflächenchemie faserbildender Polymere, Akademie Verlag, Berlin, 1984 I.B. Ivanov:&nbsp;Thin Liquid Films, Marcel Dekker, New York, Basel, 1988, ISBN 0-8247-7763-8 M.K. Sharma:&nbsp;Surface Phenomena and Additives in Water-Based Coatings and Printing Technology, Plenum Press, New York, 1992, ISBN 0-3064-4103-9 R.J. Stokes, D.F. Evans:&nbsp;Fundamentals of Interfacial Engineering, VCH, New York, 1996, ISBN 0-471-18647-3 K.L. Mittal:&nbsp;Surface and Colloid Science in Computer Technolgy, Plenum Press, New York, 1987, ISBN 0-306-42602-1 P.A. Kralchevsky, K. Nagayama:&nbsp;Particles at Fluid Interfaces and Membranes, Elsevier, Amsterdam, 2001, ISBN 0-4445-0234-2 D.A. Edwards, H. Brenner, D.T. Wasan:&nbsp;Interfacial Transport Processes and Rheology, Butterworth-Heinemann, Boston, 1991, ISBN 0-7506-9185-9 A.W. Neumann, J.K. Spelt:&nbsp;Applied Surface Thermodynamics, Marcel Dekker, New York, 1996, ISBN 0-8274-9096-0 S.A. Safran:&nbsp;Statistical Thermodynamics of Surfaces, Interfaces, and Membranes, Addison-Wesley, Reading MA, 1994, ISBN 0-201-62633-0 S. Chen, R. Rajagopalan:&nbsp;Statistical Thermodynamics of Micelles and Microemulsions, Springer Verlag, New York, 1990, ISBN 3-5409-7106-8 M. Borówko:&nbsp;Computational Methods in Surface and Colloid Science, Marcel Dekker, New York, 2000, ISBN 0-8247-0323-5 P. Becher:&nbsp;Dictionary of Colloid and Surface Science, Marcel Dekker, New York, 1989, ISBN 0-8274-8326-3 L.E. Schramm:&nbsp;Dictionary of Colloid and Interface Science, 2nd Ed., Wiley-Interscience, New York, 2001, ISBN 0-471-39406-8\r\n&nbsp;\r\nProfessional Journals on surface chemistry:\r\n Journal of Colloid and Interface Science, Academic Press, New York, ISSN 0021-9797 Langmuir, American Chemical Society, Washington D.C., ISSN 0743-7463 Colloids and Surfaces A - Physicochemical and engineering aspects, Elsevier, Amsterdam, ISSN 0927-7757 Colloids and Surfaces B - Biointerfaces, Elsevier, Amsterdam, ISSN 0927-7765 Advances in Colloid and Interface Science, Elsevier, Amsterdam, ISSN 0001-8686 Journal of Adhesion Science Technology, VNU Science Press, ISSN 0169-4243 International Journal of Adhesion and Adhesives, Elsevier, Amsterdam, ISSN 0143-2691 Journal of Dispersion Science Technology, Marcel Dekker, New York, ISSN 0193-2691 Colloid Journal&nbsp;(engl. transl. of Kolloidnyi Zhurnal) Maik Nauka, Moscow/Interperiodica Publ., ISSN 0190-4337 Powder Technology, Elsevier, Amsterdam, ISSN 0032-5910 SIA Surface and Interface Analysis, Wiley, Chichester, ISSN 0142-2421 Trends in Colloid and Interfacial Science, Verlag Th. Steinkopff, Darmstadt, ISSN 0340-255X Colloid &amp; Polymer Science, Verlag Th. Steinkopff, Darmstadt, ISSN 0303-402X Tenside, Surfactants, Detergents, Hanser Verlag, München, ISSN 0040-3490 SÃFW-Journal, Verlag für Chemische Industrie, Augsburg, ISSN 0173-5500\n', '', 'page', '', '', 1429352452, 228, 12, 'Support littéraire', 3, '', ''),
(206, 5, 1431355495, 1428385888, 0, 0, 0, '', '228', ' (copy 3)\nWe may know, or know, who knows\r\nAn individual cannot be expected to have all relevant application information at his/her fingertips. Finding a reliable source of complementary knowledge may by key to the success of your projects. DataPhysics and its carefully selected sales/support partners will happily share the knowledge gained during decades of experience in this varied field of study.\n\nLiterature Support\r\nWe administer an extensive database of electronic publications in which surface chemistry takes a prominent role. We will happily share application specific information to aid your journey through this continually developing field. If you would like our help in tracking down published data&nbsp; Please contact us.\r\nLiterature list\r\nBelow here is a selection of popular literature in which colloid and surface science feature prominently:\r\nBooks\r\n E. Matijevic (Ed.):&nbsp;Surface and Colloid Science, Vol.1-15, Wiley Interscience &amp; Plenum Publ. Co., 1969-1993, ISBN (Vol. 1) 0-4715-7630-1 ... ISBN (Vol. 15) 0-3064-4150-0, ISSN 0081-9573 K.S. Birdi (Ed.):&nbsp;Handbook of Surface and Colloid Chemistry, CRC Press, Boca Raton, New York, 1997, ISBN 0-8493-945-7; 2nd Ed., 2002, ISBN 0-8493-1079-2 A.W. Adamson, A.P. Gast:&nbsp;Physical Chemistry of Surfaces, 6th Ed., John Wiley &amp; Sons, New York, 1997, ISBN 0-417-14873-3 D.J. Shaw:&nbsp;Introduction to Colloid and Surface Chemistry, 3rd Ed., Butterworths, London 1980, ISBN 0-408-71049-7 M.J. Schwuger:&nbsp;Lehrbuch der Grenzflächenchemie, Georg Thieme Verlag, Stuttgart, New York, 1996, ISBN 3-13-137501-9 G. Brezesinski, H.-J. Mögel:&nbsp;Grenzflächen und Kolloide, Spektrum Verlag, Heidelberg, Berlin, Oxford, 1993, ISBN 3-86025-016-7 H.-D. Dörfler:&nbsp;Grenzflächen- und Kolloidchemie, VCH Verlag, Weinheim, 1994, ISBN 3-527-29072-9 H.-D. Dörfler:&nbsp;Grenzflächen und kolloid-disperse Systeme - Physik und Chemie, Springer-Verlag, Berlin, Heidelberg, 2002, ISBN 3-540-42547-0 P.C. Hiemenz, R. Rajagopalan:&nbsp;Principles of Colloid and Surface Chemistry, 3rd Ed., Marcel Dekker, New York, 1997, ISBN 0-8247-9397-8 D. Myers:&nbsp;Surfaces, Interfaces, and Colloids, 2nd Ed., Wiley-VCH, New York, 1999, ISBN 0-471-33060-4 M.J. Jaycock, G.D. Parfitt:&nbsp;Chemistry of Interfaces, Ellis Horwood, Chichester, 1981, ISBN 0-470-27013-6 F. MacRitchie:&nbsp;Chemistry at Interfaces, Academic Press, San Diego, 1990, ISBN 0-12-464785-5 D.F. Evans, H. Wennerström:&nbsp;The Colloidal Domain - Where Physics, Chemistry, Biology, and Technology meet, VCH, New York, 1994, ISBN 1-56081-525-6 J. Lyklema:&nbsp;Fundamentals of Interface and Colloid Science: Solid-Liquid Interfaces, Academic Press, New York, 1995, ISBN 0-1246-0524-9 R.J. Hunter:&nbsp;Introduction to Modern Colloid Science, Oxford Univ. Press, Oxford, 1993, ISBN 0-19-855387-0 J.N. Israelachvili:&nbsp;Intermolecular and Surface Forces, 2nd Ed., Academic Press, New York, 1991, ISBN 0-1237-5181-0 C.J. van Oss:&nbsp;Interfacial Forces in Aqueous Media, Marcel Dekker, New York, 1994, ISBN 0-8274-9168-1 J.S. Rowlinson, B. Widom:&nbsp;Molecular Theory of Capillarity, Clarendon Press, Oxford, 1989, ISBN 0-19-855642-X R. Finn:&nbsp;Equilibrium Capillary Surfaces, Springer Verlag, New York, 1986, ISBN 0-387-96174-7 A.I. Rusanov, F.C. Goodrich (Eds.):&nbsp;The Modern Theory of Capillarity, Akademie Verlag, Berlin, 1981 S. Hartland, R.W. Hartley:&nbsp;Axissymmetric Fluid-Liquid Interfaces, Elsevier, Amsterdam, 1976, ISBN 0-4444-1396-0 K.L. Mittal (Ed): Contact Angle,&nbsp;Wettability and Adhesion, VSP, Utrecht, 1993, ISBN 90-6764-157-X J.C. Berg:&nbsp;Wettability, Marcel Dekker, New York, 1993, ISBN 0-8247-9046-0 J. F. Padday (Ed.):&nbsp;Wetting, Spreading and Adhesion, Academic Press, New York, 1978, ISBN 0-1254-4350-1 M.E. Schrader, G. Loeb (Eds):&nbsp;Modern Approaches to Wettability, Plenum Press, New York, 1992, ISBN 0-3064-3985-9 A.I. Rusanov, V.A. Prokhorov:&nbsp;Interfacial Tensiometry, Elsevier, Amsterdam, 1996, ISBN 0-444-92128-7 K.L. Mittal (Ed):&nbsp;Adsorption at Interfaces, 5th Ed., Wiley Interscience, New York, 1990, ISBN 0-8412-0249-4 S.S. Dukhin, G. Kretschmar, R. Miller:&nbsp;Dynamics of Adsorption at Liquid Interfaces, Elsevier, Amsterdam 1995, ISBN 0-444-88117-4 C.A. Miller, P. Neogi:&nbsp;Interfacial Phenomena, Marcel Dekker, New York, 1985, ISBN 0-8274-7490-6 C. Isenberg:&nbsp;The Science of Soap Films and Soap Bubbles, Dover Publ., New York, 1992, ISBN 0-486-26960-4 D. Möbius, R. Miller:&nbsp;Drops and Bubbles in Interfacial Research, Elsevier, Amsterdam, 1998, ISBN 0-444-82894-X N.R. Morrow (Ed):&nbsp;Interfacial Phenomena in Petroleum Recovery, Marcel Dekker, New York, 1990, ISBN 0-8274-8385-9 C.A. Croxton:&nbsp;Fluid Interfacial Phenomena, John Wiley, Chichester, 1986, ISBN 0-4719-0757-X D.K. Chattoraj, K.S. Birdi:&nbsp;Adsorption and the Gibbs Surface Excess, Plenum Press, New York, 1984, ISBN 0-306-41334-5 K. Tsujii:&nbsp;Surface Activity - Principles, Phenomena, and Applications, Academic Press, San Diego, 1998, ISBN 0-12-702280-5 H.-F. Eicke:&nbsp;Interfacial Phenomena in Apolar Media, Marcel Dekker, New York, 1987, ISBN 0-8247-7506-6 T.F. Tadros:&nbsp;Surfactants, Academic Press, London, 1984, ISBN 0-12-682180-1 D. Myers:&nbsp;Surfactant Science and Technology, VCH Publishers, New York, 1988, ISBN 0-89573-339-0 K. Kosswig, H. Stache (Eds.):&nbsp;Die Tenside, C. Hanser Verlag, München, Wien, 1993, ISBN 3-446-16201-1 N.M. van Os, J.R. Haak, L.A.M. Rupert:&nbsp;Physio-chemical Properties of Selected Anionic, Cationic and Nonionic Surfactants, Elsevier, Amsterdam, 1993, ISBN 0-444089691-0 M.J. Rosen:&nbsp;Surfactants and Interfacial Phenomena, 2nd Ed., John Wiley, New York, 1989, ISBN 0-471-83651-6 M.J. Rosen (Ed.):&nbsp;Structure performance relationships in surfactants, ACS, Washington, DC, 1984, ISBN 0-8412-0839-5 K.L. Mittal (Ed.):&nbsp;Surfactants in Solution, Marcel Dekker, New York, 1996, ISBN 0-8247-9792-2 P.M. Holland, D.N. Rubingh:&nbsp;Mixed Surfactant Systems, ACS, Washington D.C., 1992, ISBN 0-8412-2468-4 A.I. Rusanov:&nbsp;Micellization in Surfactant Solution, Gordon &amp; Breach, New York, 1999, ISBN 90-5702-297-4 K. Holmberg, B. Jönsson, B. Lindman:&nbsp;Surfactants and Polymers in Aqueous Solution, 2nd. Ed., John Wiley &amp; Sons, Ltd, Chichester, 2003, ISBN 0-471-49883 J.D. Andrade (Ed.):&nbsp;Surface and Interfacial Aspects of Biomedical Polymers, Vol.1: Surface chemistry and physics, Vol. 2: Protein adsorption, Plenum Press, New York, 1985, ISBN 0-306-41741-3 &amp; 0-306-41742-1 M. Bender (Ed.):&nbsp;Interfacial Phenomena in Biological Systems, Marcel Dekker, New York, 1991, ISBN 0-8247-8436-7 D. Möbius, R. Miller:&nbsp;Proteins at Interfaces, Elsevier, Amsterdam, 1998, ISBN 0-4448-2944-X S. Magdassi:&nbsp;Surface Activity of Proteins: Chemical and Physicochemical Modifications, Dekker, New York, 1996, ISBN 0-8247-9532-6 M.N. Jones:&nbsp;Micelles, Monolayers, and Biomembranes, Wiley-Liss, New York, 1994, ISBN 0-471-56139-8 V.N. Kazakov, O.V. Sinyachenko, V.B. Fainerman, U.Pison, R. Miller:&nbsp;Dynamic Surface Tensiometry in Medicine, Elsevier, Amsterdam, 2000, ISBN 0-4445-0411-7 I. Noda, D.N. Rubingh:&nbsp;Polymer Blends, Solutions, and Interfaces, Elsevier, Amsterdam, 1992, ISBN 0-4448-9397-0 S.Wu:&nbsp;Polymer Interface and Adhesion, Marcel Dekker, New York, 1982, ISBN 0-8247-1533-0 K.L. Mittal (ed.):&nbsp;Polymer Surface Modification: Relevance to Adhesion, VSP, Utrecht, 1996, ISBN 90-6764-201-0 M.R. Piggott (Ed.):&nbsp;Interfaces in Composites, Elsevier, Amsterdam, 1991, ISBN 1-85166-667-2 K.L. Mittal, H.R. Anderson (Eds):&nbsp;Acid-Base Interactions: Relevance to Adhesion Science and Technology, VSP, Utrecht 1991, ISBN 90-6764-135-9 L.H. Lee (Ed):&nbsp;Fundamentals of Adhesion, Plenum Press, New York, 1990, ISBN 0-3064-3470-9 W.C. Wake:&nbsp;Adhesion and Formulation of Adhesives, Elsevier, New York, 1982, ISBN 0-8533-4134-1 I. Skeist:&nbsp;Handbook of Adhesives, 3rd Ed., Chapman &amp; Hall, New York, 1990, ISBN 0-412-09681-1 A.J. Kinloch:&nbsp;Adhesion and Adhesives, Chapman &amp; Hall, London, 1987, ISBN 0-412-27440-X A.F.M. Barton:&nbsp;Handbook of Solubility Parameters and Other Cohesion Parameters, CRC Press, Boca Raton, New York, 1991, ISBN 0-8493-0176-9 C.M. Hansen:&nbsp;Hansen Solubility Parameters, CRC Press, Boca Raton, 2000, ISBN 0-8493-1525-5 H.J. Jakobasch:&nbsp;Oberflächenchemie faserbildender Polymere, Akademie Verlag, Berlin, 1984 I.B. Ivanov:&nbsp;Thin Liquid Films, Marcel Dekker, New York, Basel, 1988, ISBN 0-8247-7763-8 M.K. Sharma:&nbsp;Surface Phenomena and Additives in Water-Based Coatings and Printing Technology, Plenum Press, New York, 1992, ISBN 0-3064-4103-9 R.J. Stokes, D.F. Evans:&nbsp;Fundamentals of Interfacial Engineering, VCH, New York, 1996, ISBN 0-471-18647-3 K.L. Mittal:&nbsp;Surface and Colloid Science in Computer Technolgy, Plenum Press, New York, 1987, ISBN 0-306-42602-1 P.A. Kralchevsky, K. Nagayama:&nbsp;Particles at Fluid Interfaces and Membranes, Elsevier, Amsterdam, 2001, ISBN 0-4445-0234-2 D.A. Edwards, H. Brenner, D.T. Wasan:&nbsp;Interfacial Transport Processes and Rheology, Butterworth-Heinemann, Boston, 1991, ISBN 0-7506-9185-9 A.W. Neumann, J.K. Spelt:&nbsp;Applied Surface Thermodynamics, Marcel Dekker, New York, 1996, ISBN 0-8274-9096-0 S.A. Safran:&nbsp;Statistical Thermodynamics of Surfaces, Interfaces, and Membranes, Addison-Wesley, Reading MA, 1994, ISBN 0-201-62633-0 S. Chen, R. Rajagopalan:&nbsp;Statistical Thermodynamics of Micelles and Microemulsions, Springer Verlag, New York, 1990, ISBN 3-5409-7106-8 M. Borówko:&nbsp;Computational Methods in Surface and Colloid Science, Marcel Dekker, New York, 2000, ISBN 0-8247-0323-5 P. Becher:&nbsp;Dictionary of Colloid and Surface Science, Marcel Dekker, New York, 1989, ISBN 0-8274-8326-3 L.E. Schramm:&nbsp;Dictionary of Colloid and Interface Science, 2nd Ed., Wiley-Interscience, New York, 2001, ISBN 0-471-39406-8\r\n&nbsp;\r\nProfessional Journals on surface chemistry:\r\n Journal of Colloid and Interface Science, Academic Press, New York, ISSN 0021-9797 Langmuir, American Chemical Society, Washington D.C., ISSN 0743-7463 Colloids and Surfaces A - Physicochemical and engineering aspects, Elsevier, Amsterdam, ISSN 0927-7757 Colloids and Surfaces B - Biointerfaces, Elsevier, Amsterdam, ISSN 0927-7765 Advances in Colloid and Interface Science, Elsevier, Amsterdam, ISSN 0001-8686 Journal of Adhesion Science Technology, VNU Science Press, ISSN 0169-4243 International Journal of Adhesion and Adhesives, Elsevier, Amsterdam, ISSN 0143-2691 Journal of Dispersion Science Technology, Marcel Dekker, New York, ISSN 0193-2691 Colloid Journal&nbsp;(engl. transl. of Kolloidnyi Zhurnal) Maik Nauka, Moscow/Interperiodica Publ., ISSN 0190-4337 Powder Technology, Elsevier, Amsterdam, ISSN 0032-5910 SIA Surface and Interface Analysis, Wiley, Chichester, ISSN 0142-2421 Trends in Colloid and Interfacial Science, Verlag Th. Steinkopff, Darmstadt, ISSN 0340-255X Colloid &amp; Polymer Science, Verlag Th. Steinkopff, Darmstadt, ISSN 0303-402X Tenside, Surfactants, Detergents, Hanser Verlag, München, ISSN 0040-3490 SÃFW-Journal, Verlag für Chemische Industrie, Augsburg, ISSN 0173-5500\n', '', 'page', '', '', 1429352452, 228, 12, 'Literature-Support', 2, '', '');
INSERT INTO `tx_kesearch_index` (`uid`, `pid`, `tstamp`, `crdate`, `cruser_id`, `starttime`, `endtime`, `fe_group`, `targetpid`, `content`, `params`, `type`, `tags`, `abstract`, `sortdate`, `orig_uid`, `orig_pid`, `title`, `language`, `directory`, `hash`) VALUES
(207, 5, 1431355495, 1428385888, 0, 0, 0, '', '229', 'Glossar\nUm Ihnen den Einstieg in die Terminologie der Grenzflächenchemie zu erleichtern, finden Sie auf dieser Seite kurze Erläuterungen der gängigen Begriffe.\n\n       \r\n        \r\n          A\r\n            B\r\n            C\r\n            D\r\n            E\r\n            F\r\n            G\r\n            H\r\n            I\r\n            J\r\n            K\r\n            L\r\n            M\r\n            N\r\n            O\r\n            P\r\n            Q\r\n            R\r\n            S\r\n            T\r\n            U\r\n            V\r\n            W\r\n            X\r\n            Y\r\n            Z\r\n            \r\n        \r\n     \r\n        \r\n        A\r\n         \r\n        Absorption  \r\n        Unter Absorption versteht man das Eindringen von Teilchen einer  Phase in oder durch eine  Oberfläche einer anderen Phase.\r\n         Absorptionskinetik  \r\n          Zeitliches Verhalten der Absorption, das durch Oberflächenbeschaffenheit, Temperatur,  Porosität und Dicke des Materials bestimmt wird.\r\n          \r\n          Adhäsion\r\n          \r\n          Die chemische oder mechanische Verbindung (Haftung) eines Stoffes zu einem anderen Stoff.          \r\n         Adhäsionsenergie (Adhäsionsarbeit)\r\n          \r\n           \r\n          Bezeichnung die intermolekulare Anziehungsenergie zwischen zwei miteinander in Berührung stehenden  Phasen (Stoffen) pro Einheitsfläche.          \r\n         Adsorbat Die auf einem Untergrund angereicherte Substanz.          \r\n         Adsorbent  \r\n          Material, auf der eine Adsorption stattfindet; die angereicherte Oberfläche.          \r\n         Adsorption\r\n          \r\n           \r\n          Unter Adsorption versteht man  die Anreicherung bzw. Verdichtung von Teilchen an einer Oberfläche bzw. Grenzfläche.          \r\n         Adsorptionswärme  \r\n          Ist die freigesetzte Wärme, die bei einer Adsorption entsteht; wird meistens bei Gas / Festkörper-Verbindungen gemessen.          \r\n         Aerosol  \r\n          Dispersion von flüssigen oder festen Teilchen in einer gasförmigen Umgebung.          \r\n         Agglomerat Räumliche Ansammlung von Partikeln, die chemisch oder physikalisch nicht dauerhaft miteinander verbunden sind.          \r\n         Aktivierungsenergie  \r\n          Mindestenergie, die einem System zugefügt werden muÃ, um einen ProzeÃ in Gang zu setzen, meistens bei chemischen Reaktionen. \r\n\r\n        \r\n \r\n    \r\n        B\r\n         \r\n        Basislinie  \r\n        Bezeichnet die (orthogonale) Projektionslinie der Kontaktfläche zwischen einem liegenden Tropfen und dem Festkörper (Untergrund).\r\n\r\n\r\n         Benetzung\r\n          \r\n           \r\n          Phänomen des haftenden Kontakts von Flüssigkeiten an der  Oberfläche von Festkörpern und Flüssigkeiten.\r\n          \r\n          \r\n         Benetzungsverhalten (Benetzbarkeit)\r\n          \r\n           \r\n          Bezeichnet die Fähigkeit von Flüssigkeiten, sich auf einer Oberfläche auszubreiten; je besser die Benetzbarkeit, umso kleiner ist der bei der Benetzung auftretende Kontaktwinkel. Oberflächen werden auch als (unvollständig) benetzbar bezeichnet, wenn der Kontaktwinkel &lt;90° beträgt.\r\n          \r\n          \r\n         Bipolar\r\n          \r\n           \r\n          Als bipolar - im Kontext der Grenzflächenchemie - bezeichnet man eine zweifach funktionelle Substanz mit einer Oberfläche, die als Elektronenspender und als Elektronenempfänger fungieren kann.          \r\n         Blasendruck Methode \r\n        Methode zur Bestimmung der dynamischen Oberflächenspannung von Flüssigkeiten; mittels einer in eine Flüssigkeit senkrecht eingetauchte Kapillare wird eine Gasblase in die Flüssigkeit gedrückt und die Oberflächenspannung anhand des meÃbaren maximalen Blasendrucks ermittelt. \r\n        \r\n        \r\n        \r\n        C\r\n         Chemische Adsorption  \r\n          Siehe Chemisorption\r\n         Chemisorption\r\n          \r\n           \r\n          Ist eine Form der Adsorption, die durch eine chemische Reaktion zwischen dem angelagerten Stoff und der angereicherten  Oberfläche  entsteht          \r\n         Circle-Fit  \r\n          Auch Höhen-Breiten-Methode; Methode zur Berechnung des gemessenen  Kontaktwinkels von Tropfen oder Blasen. Der Kontaktwinkel wird bestimmt durch das geometrische Verhältnis aus Höhe und Breite des Tropfens. Diese Methode ist einfach zu handhaben , jedoch gibt es zur präziseren Auswertung andere Methoden, wenn der Kontaktwinkel gröÃer als 20° ist (z.B. Ellipsenmethode, Young-Laplace-Fit)          \r\n         Cluster  \r\n          Räumliche Ansammlung von Molekülen oder Atomen (Subeinheiten), die einen bestimmten Ordnungs- bzw. Symmetriezustand aufweisen und mit ihrer Umgebung im Stoffaustausch stehen. In Flüssigkeiten und Gasen weisen Cluster meist einen höheren Fluktuationsgrad der Subeinheiten an der Oberfläche auf.          \r\n         Coadsorption  \r\n          Ist die gleichzeitige Adsorption mehrerer Stoffe auf einer Oberfläche.\r\n        \r\n        \r\n        \r\n        D\r\n         Desorption  \r\n          Ist das Gegenteil einer Adsorption, also die Abtrennung angelagerter Teilchen von einer Oberfläche.\r\n         Detergens  \r\n          Synthetisches Waschhilfsmittel; oberflächenaktive Substanz zur Reinigung von Oberflächen; meist zur Entfernung von Schmutzpartikeln und chemischen Kontaminationen.          \r\n         Diffusion Konzentrationsabhängige Migration, vorrangig als thermisch bedingte Wanderungsbewegung, von Kolloidpartikeln, Molekülen oder Ionen.          \r\n         Diffusionskoeffizient  \r\n          MaÃ für die Stärke der Diffusion von Partikeln; abhängig von PartikelgröÃe, Viskosität und Temperatur.          \r\n         Dipol  \r\n          chem: polares Molekül; phys.: elektrisch oder auch magnetisch geladenes Teilchen mit räumlich separierter positiver und negativer Ladungsverteilung.          \r\n         Dipol-Dipol-Kräfte  \r\n          Elektromagnetische Wechselwirkungskräfte, die zwischen Dipolen bestehen.          \r\n         Dipolmoment  \r\n          GröÃe zur Beschreibung der Stärke und potentiellen Wirkung eines Dipols durch einen Vektor, der Ladung und Richtung des Dipols angibt.          \r\n         Disperse Anteile Die Anteile in der Oberflächenenergie oder Grenz- bzw. Oberflächenspannung, die ihre Ursache in dispersen Wechselwirkung an der Phasengrenzfläche finden.          \r\n         Disperse Wechselwirkung  \r\n          Anziehung zwischen induzierten Dipolen.          \r\n         Dispersion\r\n          \r\n           \r\n          Verteilung von festen Partikeln in einer Flüssigkeit.\r\n         Dreiphasenpunkt\r\n          \r\n             \r\n        (orthogonale) Projektion der geschlossenen Randlinie, an der sich Festkörper, Flüssigkeit und Gas treffen.          \r\n         Du Noüy-Ring Methode (nach P. Lecomte Du Noüy, 1919)  \r\n          Methode zur Bestimmung der statischen Oberflächen- und Grenzflächenspannung von Flüssigkeiten mittels eines Rings (meist aus einer Platinlegierung), der in die zu messende Flüssigkeit eingetaucht und wieder herausgezogen wird; Messung der maximalen Kraft, die notwendig ist, um die Flüssigkeitslamelle, die sich am Ring bildet, hinauf zu ziehen; wird häufig zur Messung der Kritischen Mizellbildungskonzentration (CMC) verwandt.          \r\n         Dynamische Grenzflächenspannung\r\n          \r\n           \r\n          Ist die Grenzflächenspannung auÃerhalb des (vollständigen) thermodynamischen Gleichgewichts.\r\n         Dynamischer Kontaktwinkel\r\n          \r\n           \r\n          Fortschreite- oder Rückzugswinkel einer Flüssigkeitsfront an einem Festkörper; wird vorrangig zur Bestimmung der Freien Oberflächenenergie gemessen.          \r\n         Dynamische Oberflächenspannung  \r\n          Ist die Oberflächenspannung auÃerhalb des (vollständigen) thermodynamischen Gleichgewichts.\r\n          Dynamische Wilhelmy-Methode (nach L. Wilhelmy, 1863)  \r\n          Methode zur Bestimmung des Fortschreite- und Rückzugwinkels von prismatischen und zylindrischen Probekörpern mit bekannter Geometrie, die in eine Flüssigkeit eingetaucht und wieder herausgezogen werden.\r\n        \r\n        \r\n        \r\n        E\r\n         Einzelfaserkontaktwinkel Methode\r\n\r\n         \r\n        Methode zur Bestimmung des  Fortschreite- und Rückzugwinkels  an Fasern und zur Bestimmung des benetzten Umfangs / Durchmessers von Fasern.\r\n         Elastizität\r\n          \r\n           \r\n          Verhältnis von angreifenden Kräften und der reversiblen Verformung des Körpervolumens bzw. der Oberfläche.          \r\n         Ellipsenmethode  \r\n          Näherungsmethode zur Berechnung des Kontaktwinkels aus der Kontur eine Flüssigkeitstropfens unter Annahme einer elliptischen Gestalt (meist verwendet in einem Kontaktwinkelbereich zwischen 20° und 110°).          \r\n         Emulsion\r\n          \r\n           \r\n          Dispersion von zwei nur bedingt miteinander mischbaren flüssigen Phasen, z. B. Wasser / Ãl.          \r\n         Enthalpie\r\n          \r\n             \r\n        Die Enthalpie H eines physikalischen System setzt sich zusammen aus der inneren Energie des Systems U und der Volumenarbeit pV: H = U + pV.          \r\n         Entropie\r\n          \r\n           \r\n          MaÃ für die molekulare Unordnung bzw. MaÃ für die molekulare Beweglichkeit, die in einem System herrscht. Der Entropieinhalt S eines Systems vergröÃert sich mit der Zunahme der molekularen Unordnung, z. B. durch Erhöhung der Temperatur.          \r\n         Equilibrium\r\n          \r\n           \r\n          lat., Gleichgewicht.          \r\n         Expansion\r\n          \r\n           \r\n          Ausdehnung; Thermisch bedingte Volumenänderung, die von der Art des Stoffes, der Ausgangstemperatur und der Temperaturerhöhung abhängt.          \r\n         Expansionsfaktor\r\n          \r\n             \r\n        Thermischer Volumenausdehnungskoeffizient von Stoffen b = 1/V·(¶V/¶T)p;. Steht mit dem Temperaturkoeffizienten der Oberflächenspannung in enger Beziehung. Für viele molekulare Flüssigkeiten gilt in guter Näherung: dg(T)/dT = 4b·g(T)\r\n \r\n        \r\n        \r\n        \r\n        F\r\n         Film\r\n\r\n             \r\n        Siehe Oberflächenfilm.\r\n         Füllstoff  \r\n          Füllmaterial, das einem anderen Stoff zugefügt wird, z. B. Polymer, Papier, Gummi, um dessen Eigenschaften zu verbessern oder dessen Preis zu senken.          \r\n         Flockulation  \r\n          Entstehung von Flocken durch Hinzugabe von bestimmten Substanzen (Flockulationsmittel) oder durch Ãnderung der physikalischen Bedingungen (Temperatur, pH) des flockulationsfähigen Systems; häufig auch Vernetzung von Einzelpartikeln durch adsorbierte Makromoleküle (Polymere).\r\n          Fluktuation  \r\n          Zeitweise und sprungartige Veränderung einer oder mehrerer Eigenschaften eines physikalischen Systems.          \r\n         Fluktuationsgrad  \r\n          Zeitliche Rate an spontan auftretenden oder induzierten Fluktuationen eines physikalischen Systems.          \r\n         Flüssigkeitslamelle  \r\n          Siehe Lamelle.          \r\n         Fortschreitewinkel\r\n          \r\n             \r\n        Dynamischer Kontaktwinkel, der beim Eintauchen einer festen Substanz in eine Flüssigkeit oder bei fortschreitender Dosierung eines Tropfens auf einer Oberfläche gemessen wird.          \r\n         Freie Oberflächenenergie  \r\n          Siehe Oberflächenenergie. \r\n        \r\n        \r\n        \r\n        G\r\n         Gel  \r\n          Gallertartiges Kolloid.\r\n         Goniometer  \r\n          Optisches KontaktwinkelmeÃgerät mit manueller Auswertung des Tropfenbildes mittels einer Winkelskala.          \r\n         Grenzfläche  \r\n          a) Im erweiterten Sinne jegliche Phasengrenze zwischen verschiedenen kondensierten und auch zwischen kondensierten und nichtkondensierten Phasen.\r\n             \r\n          b) Im engere Sinne eine Phasengrenze zwischen zwei Festkörpern, zwei Flüssigkeiten oder einem Festkörper und einer Flüssigkeit.        \r\n         Grenzflächenanalyse  \r\n          Untersuchung einer Grenzfläche zur Bestimmung deren chemischen und physikalischen Eigenschaften.          \r\n         Grenzflächenaktive Substanz  \r\n          Siehe oberflächenaktive Substanz.          \r\n         Grenzflächenchemie  \r\n          Interdisziplinäre Wissenschaft, die sich mit den spezifischen Wechselwirkungen und Eigenschaften der Stoffe an Phasengrenzen und den physikalisch-chemischen Veränderungen aufgrund dieser Wechselwirkungen befaÃt.          \r\n         Grenzflächenspannung\r\n          \r\n           \r\n          Wichtiger thermodynamischer Parameter zur Charakterisierung von Phasengrenzen. Bezeichnet die Menge der Arbeit pro Einheitsfläche, die aufgewendet werden muÃ, um ein Stück neuer Grenzfläche zwischen zwei Phasen zu erzeugen (SI-Einheit: mN/m und mJ/m²).\r\n        \r\n        \r\n        \r\n        H\r\n         Heterogenität  \r\n          Bezeichnet Uneinheitlichkeit; Stoffaufbau aus verschiedenen Phasen, auch chemische Uneinheitlichkeit an Phasengrenzflächen.\r\n         Hochgeschwindigkeitskamera\r\n          \r\n           \r\n          Kamera mit einer hoher Bildaufnahmerate, typischerweise ab einigen Hundert Bildern/sec; in der Grenzflächenchemie häufig benutzte Technik zur Charakterisierung von schnellen Absorptionsprozessen an saugenden Papieren und Non Woven-Materialien \r\n         Höhen-Breiten-Methode (H/W-Methode)\r\n          \r\n           \r\n          Berechnungsmethode für Kontaktwinkel an Tropfen und Blasen nach der Gleichung: tan Q = 2H/W;  siehe auch Circle-Fit. \r\n         Homogen\r\n          \r\n           \r\n          Bezeichnet Einheitlichkeit; Stoffaufbau aus einer einzigen Phase. \r\n         Hydratation \r\nUmhüllung eines geladenen oder ungeladenen Moleküls oder Atoms mit Wassermolekülen. \r\n         Hydrophil\r\n          \r\n           \r\n          Wasserfreundlich.\r\n          Hydrophob  \r\n          Wasserabweisend. \r\n         Hydrotropie\r\n          \r\n           \r\n          Phänomen, daÃ eine schwerlösliche Substanz in Gegenwart einer zweiten Komponente, die selbst kein Lösungsmittel darstellt, wasserlöslich wird. Substanzen, die eine derartige Löslichkeitsverbesserung bewirken, werden als Hydrotrope oder Hydrotropika bezeichnet. \r\n         Hysterese  \r\n          Siehe Kontaktwinkelhysterese.\r\n        \r\n        \r\n        \r\n        I\r\n         Keine Einträge verfügbar.\r\n        \r\n        \r\n        \r\n        J\r\n         Joule  \r\n          1 Joule ist die Arbeit, die aufgewandt werden muÃ, wenn der Angriffspunkt der Kraft 1 N in Richtung der Kraft um 1 m verschoben wird.\r\n        \r\n        \r\n        \r\n        K\r\n         Kapillaren\r\n\r\n         \r\n        Sehr feine, meist langgestreckte Hohlräume; insbesondere für Röhrchen mit kleinen Innendurchmessern.\r\n\r\n         Kapillardruck  \r\n          An gekrümmten Objekten (wie Tropfen, Blasen, Lamellen) auftretender Druck in Normalenrichtung zur Grenzfläche, der von den Hauptkrümmungsradien R1 und R2 dieser geometrischen Objekte sowie der an der Phasengrenze herrschenden Ober- bzw. Grenzflächenspannung gL abhängt:  Dp = gL (1/R1+1/R2).\r\n          \r\n         Kapillarkräfte  \r\n          Bezeichnung für die Kräfte, die durch das Wirken der Oberflächenspannung von Flüssigkeiten in Kapillaren, engen Spalten und Poren entstehen.\r\n          \r\n         Koagulation  \r\n          Gerinnung. Abgeleitete Bezeichnung für die Umwandlung eines Sols in ein Gel unter Ausflockung\r\n          \r\n         Kohäsion  \r\n          Durch Anziehungskraft bewirkter Zusammenhalt der Moleküle einer Phase.\r\n         Kohäsionsenergie\r\n          \r\n           \r\n          Ist die Energie, die bei Kohäsionsvorgängen entsteht oder aufgewendet werden muÃ.\r\n         Kolloid (Thomas Graham, 1861)  \r\n          Kolloide sind Zustände von Stoffen, die in charakteristischer Weise besonders fein zerteilt sind (üblicherweise in einem Dispersionsmittel), meist in einer Dimension zwischen 1 bis maximal 1000 nm; siehe auch Nanopartikel.\r\n          Kolloidchemie  \r\n        Bezeichnung für das Teilgebiet der Physikalischen Chemie, das den kolloidalen Zustand der Materie erfaÃt.          \r\n         Kompressibilität  \r\n          Unter der Kompressibilität versteht man die Eigenschaft aller Festkörper, Flüssigkeiten und Gase, ihr Volumen unter dem EinfluÃ eines (allseitig) wirkenden Drucks zu verkleinern.          \r\n         Kondensation  \r\n          Auch Verdichtung. Phasenumwandlung  von bestehenden Dämpfen oder Gasen in Flüssigkeiten oder feste Stoffe (allgemein kondensierte Stoffe) durch Abkühlung.\r\n          Kontaktwinkel  \r\n        Winkel, den eine Tangente an die Tropfenkontur im Drei-Phasen-Punkt zur Oberfläche des Festkörpers bildet und somit ein MaÃ für die Benetzbarkeit einer Oberfläche oder Grenzfläche durch eine andere Phase darstellt. Je gröÃer der Kontaktwinkel, umso schlechter ist die Benetzung.\r\n          \r\n          Kontaktwinkelhysterese\r\n          \r\n          Differenz zwischen dem dynamischen Fortschreite- und Rückzugskontaktwinkel.          \r\n         KontaktwinkelmeÃgerät\r\n          \r\n           \r\n          Gerät zur Messung des Kontaktwinkels an Festkörpern oder Flüssigkeiten, meist mit Hilfe einer optischen Methode.          \r\n         Konzentration  \r\n          Anzahl von Teilchen (in Mol) pro Volumeneinheit (in Liter) eines Stoffes.\r\n          Kritische Mizellbildungskonzentration (CMC)  \r\n          Die Konzentration einer Lösung, bei der eine oberflächenaktive Substanz beginnt, Agglomerate (Mizellen) bildet, die im kinetischen Gleichgewicht mit den Einzelmolekülen stehen. Bei Ãberschreitung der CMC - als einem sog. Phasenübergangspunkt -  ändert sich der Verlauf der physikalischen Eigenschaften der Lösung mit der Konzentration, insbesondere auch die Oberflächen- oder Grenzflächenspannung des Systems, die dann einen Knickpunkt besitzt.          \r\n         Kritische Oberflächenenergie  \r\n          Die Oberflächenenergie eines Festkörpers, die der kritischen Oberflächenspannung entspricht. Siehe auch Zisman-Plot.          \r\n         Kritische Oberflächenspannung  \r\n          Ist die gröÃte Oberflächenspannung einer Flüssigkeit, die gerade noch auf einer Festkörperoberfläche spreitet. Siehe auch Zisman-Plot. \r\n        \r\n        \r\n        \r\n        L\r\n         Lamelle \r\n        Beim Eintauchen oder Herausziehen eines Gegenstandes aus einer Flüssigkeit entstehende, konvex oder konkav gekrümmte Flüssigkeitssäule am eintauchenden Umfang des Gegenstandes.\r\n          Lamellenmethode  \r\n          Methode zur Bestimmung der Oberflächenspannung von hochviskosen Flüssigkeiten, z. B. Polymerschmelzen auch bei höheren Temperaturen aus der gekrümmten Kontur von Lamellen.          \r\n         Laminare Strömung\r\n          \r\n           \r\n          Nichtturbulente Strömung; es treten dabei keine Verwirbelungen des Fluidstroms auf.          \r\n         Lichtstreuung  \r\n          Streuung von sichtbarem Licht z.B. durch kolloidal verteilte Teilchen in Lösung. Sie ist oft schon ohne experimentelle Hilfsmittel durch Trübung und Opaleszenz erkennbar.\r\n          Lipophil  \r\n        Fettfreundlich.\r\n          Lipophob  \r\n          Fettabweisend.\r\n          Löslichkeit \r\n        Siehe Lösungen. \r\n         Lösungen\r\n          \r\n          Homogene  \r\n          Gemische verschiedener Stoffe, wobei auch noch mikroskopische Teilvolumina der Lösungen eine gleichartige Zusammensetzung aufweisen. Unter Lösungen versteht man im engeren Sinne flüssige Gemische aus mindestens 2 Komponenten. \r\n        \r\n        \r\n        \r\n        M\r\n         Makroemulsion\r\n\r\n         \r\n        Siehe Emulsion.\r\n         Makromolekül \r\nBezeichnet ein Molekül, für die eine vielfache Wiederholung einer oder mehrerer Arten von einer meist kovalent verknüpften Gruppe von Atomen (Monomere) charakteristisch ist. Deren Anzahl ist bereits so groÃ, daÃ sich die Eigenschaften der Makromoleküle durch Hinzufügen oder Abspalten einiger weniger Monomere nicht wesentlich verändern. \r\n         Maximaler Blasendruck\r\n          \r\n           \r\n          Der maximale innere Druck pmax in der aus einer Kapillare austretenden Gasblase ist dann erreicht, wenn sie die näherungsweise Form einer Halbkugel mit dem Radius rB = rK (rK - Halbkugelradius) annimmt. Dann läÃt sich die Oberflächenspannung gLV der umgebenden Flüssigkeit über die Gleichung gLV = Dpmax/(2 rB) berechnen. \r\n         Medium\r\n          \r\n           \r\n          Begriff, der verwendet wird, um die Bedingungen, unter denen sich eine chemische Reaktion abspielt, näher zu charakterisieren. \r\n         Mizellbildung  \r\n          Siehe Mizelle. \r\n         Mizelle \r\nAls Mizellen werden orientierte, oftmals kugelförmige oder stäbchenförmige Aggregate oberflächenaktiver Substanzen bezeichnet. Es formieren sich meist 10 bis 100 einzelne Moleküle, die sich so ausrichten, daÃ die hydrophilen oder lipophilen Anteile des Moleküls die AuÃenseite des Aggregates bilden, je nach umgebenden Lösemittel. Die Lösemittelmoleküle werden aus dem Innneren der Mizelle verdrängt.          \r\n         Mizellkonzentration \r\nSiehe Kritische Mizellbildungskonzentration (CMC).\r\n          Mizellisierung  \r\n          Beschreibt den Vorgang der Mizellbildung.\r\n          Mikroemulsion  \r\n          Thermodynamisch stabile, transparente Lösung von Mizellen, die mit Lösungshilfsmitteln versetzt ist.          \r\n         Monodispers\r\n          \r\n           \r\n          Sind die in einer Lösung dispergierten Teilchen alle von derselben GröÃe, spricht man von einem monodispersen System.          \r\n         Monolayer \r\n          Siehe Monoschicht.\r\n          Monoschicht  \r\n          Monomolekularer Film, der auf einem festen oder flüssigen Untergrund adsorbiert ist.\r\n          Multilayer  \r\n          Mehrschichtiges Adsorbat auf einem meist festen Untergrund.\r\n        \r\n        \r\n        \r\n        N\r\n         Nanopartikel  \r\n          ultrafeine, feste Partikel im kolloidalen GröÃenbereich mit hohem Oberflächen- zu Volumenverhältnis, die Eigenschaften der Partikel werden wesentlich durch die hohe spezifische Oberfläche bestimmt; die Herstellung und die Eigenschaften von Nanopartikeln sind Gegenstand der Nanotechnologie und ihrer wissenschaftlichen Basis, der Kolloidchemie\r\n         nichtbenetzend  \r\n          Eigenschaft einer Flüssigkeit auf der Oberfläche eines Festkörpers (oder einer anderen Flüssigkeit) nicht zu spreiten wenn der Kontaktwinkel gröÃer oder gleich 90° ist. \r\n        \r\n        \r\n        \r\n        O\r\n         \r\n        Oberfläche \r\n\r\n\r\nPhasengrenze zwischen einem Festkörper oder einer Flüssigkeit und einem Gas.\r\n\r\n\r\n         Oberflächenaktive Substanz \r\n          \r\n          \r\n          Chemische Verbindung, die sowohl hydrophile als auch lipophile Anteile besitzt und sich bevorzugt an Oberflächen und Grenzflächen anreichert, z. B. Tenside.\r\n          \r\n          \r\n         Oberflächenanalyse \r\n          \r\n          \r\n          Untersuchung einer Oberfläche zur Bestimmung derer physikalischen und chemischen Eigenschaften.\r\n          \r\n          Oberflächenchemie \r\n          \r\n          \r\n          Siehe auch Grenzflächenchemie; Wissenschaft, die sich mit der Bildung, der chemischen Struktur, den chemischen Veränderungen sowie dem physikalischen Verhalten von den Oberflächen befaÃt.\r\n          \r\n          \r\n         Oberflächenenergie (Freie Oberflächenenergie) \r\n          \r\n          \r\n          MaÃ für den gesamten Energieinhalt einer Festkörperoberfläche, das ein Ãquivalent zu Oberflächenspannung bei Flüssigkeiten darstellt. Die Gesamtenergie kann für viele praktische Anwendungen, als aus polaren und dispersen Anteilen zusammengesetzt betrachtet werden (SI-Einheiten: mN/m und mJ/m²).\r\n          \r\n          \r\n         Oberflächenfilm \r\n          \r\n          \r\n          Dünne Schicht von Molekülen auf Flüssigkeits- oder Festkörperoberfächen mit spezifischen Eigenschaften, die sich von der Volumenphase unterscheiden; meist als Bereich mit ausgeprägter Adsorption (dann als Adsorptionsfilm).\r\n          \r\n          Oberflächenkonzentration \r\n          \r\n          \r\n          Konzentration eines an einer Oberfläche adsorbierten Monolayers.\r\n          \r\n          \r\n         Oberflächenrauhigkeit \r\n          \r\n          \r\n          Beschreibt die Topographie einer Oberfläche.\r\n          \r\n          \r\n         Oberflächenspannung \r\n          \r\n          \r\n          Entspricht der Menge an Arbeit (pro Einheitsfläche), die aufgewendet werden muÃ, um eine Stück neue Oberfläche zu erzeugen und ist somit ein MaÃ für den Energiegehalt einer Oberfläche bei Flüssigkeiten (SI-Einheiten: mN/m und mJ/m²).\r\n          \r\n          \r\n         Osmose \r\n          \r\n          \r\n          Allgemeine Bezeichnung für einen Materialtransport zwischen zwei unterschieldich konzentrierte flüssigen Phasen (Lösungen) durch eine dritte, die sie voneinander trennt. Die trennende Phase wird meist als Membran bezeichnet. Die Osmose läuft bis zum vollständigen Konzentrationsausgleich bzw. bis zum Ausgleich des sog. osmotischen Drucks.\r\n          \r\n        \r\n        \r\n        \r\n        \r\n        P\r\n         \r\nPartielle Benetzung \r\n\r\n\r\nUnvollständige Benetzung; eine Substanz breitet sich nur unvollständig und nicht an allen Stellen auf einer  Oberfläche aus.\r\n         Passivierung \r\n          \r\n          \r\n          Behandlung einer Metalloberfläche, um die  Korrosionsanfälligkeit durch Herabsetzen der bei Metallen relativ hohen Oberflächenenergie zu reduzieren.LäÃt sich durch die Erzeugung von anti-korrosiven, stabilen Oxidschichten erreichen, z.B. aus Chrom(oxiden).          \r\n         Pendant Drop Methode \r\n          \r\n          \r\n          Methode zur Bestimmung der Oberflächenspannung von Flüssigkeiten gegen eine Gasphase mittels eines an einer Kapillare hängenden Tropfens. Die Oberflächenspannung wird durch Auswertung der Tropfenkontur ermittelt. Die Methode ist auch bei höheren  Drücken, Temperaturen und auch für die Bestimmung von Grenzflächenspannungen nicht vollständig mischbarer Flüssigkeiten anwendbar.          \r\n         Penetration \r\n          \r\n          \r\n          Eindingen von Partikeln in eine kondensierte Phase; spez.: Eindringen eines oberflächenaktiven Moleküls einer Substanz in eine auf ihr verteilte Monoschicht. Es tritt durch den Vorgang eine Ãnderung der Eigenschaften der Monoschicht ein.          \r\n         Permeabilität \r\n          \r\n          \r\n          Durchlässigkeit eines Stoffes für andere Moleküle oder Teilchen.\r\n          \r\n          Phase \r\n          \r\n          \r\n          Durch eine Fläche abgegrenzter Bereich, in dem keine sprunghaften Ãnderungen der physikalischen-chemischen Volumeneigenschaften auftreten, z.B. Dichte, chemische Zusammensetzung.          \r\n         Phasendiagramm \r\n          \r\n          \r\n          Auch Zustandsdiagramm. Zeigt die verschiedenen Existenzbereiche von Phasen (Aggregatzustände) einschlieÃlich der Phasenexistenzgrenzen eines gegebenen physikalisch-chemischen Systems in Abhängigkeit von mindestens zwei thermodynamischen Parametern an, wie z.B. Temperatur, Druck und Konzentration.          \r\n         Phasengrenzfläche \r\n          \r\n          \r\n          Geschlossene, einfach oder auch mehrfach zusammenhängende Begrenzungsfläche einer Phase gegenüber anderen Phasen.          \r\n         Physikalische Adsorption \r\n          \r\n          \r\n          Siehe Chemisorption.          \r\n         Polare Anteile \r\n          \r\n          \r\n          Die Anteile in der Oberflächenenergie oder Grenz- bzw. Oberflächenspannung , die ihre Ursache in der polaren Wechselwirkung an der Phasengrenzfläche finden.          \r\n         Polare Wechselwirkung \r\n          \r\n          \r\n          intermolekulare Anziehung zwischen induzierten und permanenten Dipolen (permanent - permanent, induziert - permanent), sowie hervorgerufen durch Wasserstoffbrückenbindungskräfte\r\n         Polarisierung \r\n          \r\n          \r\n          Auftreten eines Dipolmoments in einer Atombindung, z. B. durch unterschiedliche Elektronegatitvität.          \r\n         Polydispers \r\n          \r\n          \r\n          In einer Dispersion verteilte Teilchen von unterschiedlicher GröÃe; Gegenteil von monodispers.\r\n          Pore \r\n          \r\n          \r\n          Hohlraum, Blase in einem Material.\r\n          PorengröÃe \r\n          \r\n          \r\n          GröÃe der Poren.\r\n          \r\n          Porosität \r\n          \r\n          \r\n          Volumenverhältnis von Poren zum Material.\r\n         Pulverkontaktwinkel Methode \r\n          \r\n          \r\n          Methode zur Bestimmung des Kontaktwinkels von Pulvern und Faserbündeln sowie zur Bestimmung des Adsorptionsverhaltens von Flüssigkeiten an benetzbaren Fasern aus der tensiometrischen Messung der Adsoprtionsgeschwinigkeit (auch Modifizierte Washburn Methode)..\r\n        \r\n        \r\n        \r\n        Q\r\n         Keine Einträge verfügbar.\r\n        \r\n        \r\n        \r\n        R\r\n         Randwinkel  \r\n          Siehe Kontaktwinkel.\r\n         Rückzugswinkel\r\n          \r\n           \r\n          Dynamischer Kontaktwinkel, der beim Herausziehen einer festen Substanz aus einer Flüssigkeit oder bei wieder Einsaugen des bereits liegenden Tropfens auf einer Oberfläche zur Bestimmung der Entnetzungseigenschaften gemessen wird. \r\n        \r\n        \r\n        \r\n        S\r\n         \r\nSediment \r\nIn einer Flüssigkeit aufgeschlämmter Feststoff, der sich nach einer bestimmten Zeit absetzt.\r\n\r\n\r\n      \r\n         Sedimentation \r\n          \r\n          \r\n          Vorgang, der das Absetzen eines in einer Flüssigkeit aufgeschlemmten Feststoffes in Abhängigkeit von der Zeit beschreibt.          \r\n         Semipermeable Membran \r\n          \r\n          \r\n          Halbdurchlässige Membran.          \r\n         Sessile Drop Methode \r\n          \r\n          \r\n          Methode des "liegenden Tropfens" zur Ermittlung des Kontaktwinkels auf einer horizontal ausgerichteten Festkörperoberfläche aus der Tropfenkontur          \r\n         Sol \r\n          \r\n          \r\n          Bezeichnung für eine kolloidale Lösung. In dieser Lösung ist ein fester oder flüssiger Stoff in feinster Verteilung in einem festen, flüssigen oder gasförmigen Medium dispergiert.\r\n          Sorption \r\n          \r\n          \r\n          Technischer Begriff zur Beschreibung eines unbekannten Adsorptionsmechanismus.          \r\n         Spinning Drop Methode \r\n          \r\n          \r\n          Methode zur Bestimmung der statischen Grenzflächenspannung oder auch der dynamischen Grenzflächenspannung von Flüssigkeiten, insbesondere im niedrigen Bereich; die Messung erfolgt anhand eines bei verschiedenen Geschwindigkeiten rotierenden Tropfens, der durch die Fliehkräfte  eine zur Rotationsachse hin gestreckte Form annimmt. Der Durchmesser des Tropfens korreliert mit der Grenzflächenspannung.          \r\n         Spreiten \r\n          \r\n          \r\n          Benetzt eine Flüssigkeit eine Oberfläche so, daÃ ein Kontaktwinkel von näherungsweise 0° entsteht, spricht man von Spreiten; es findet eine vollständige Benetzung der Oberfläche statt.\r\n         Stabilität \r\n          \r\n          \r\n          Haltbarkeit einer Dispersion oder Emulsion, d. h. es findet keine Entmischung der dispergierten bzw. emulgierten Phase von der umgebenden Phase statt.          \r\n         Steighöhenmethode  \r\n          Beim Eintauchen einer Kapillare in eine Flüssigkeit steigt der Flüssigkeitspegel in der Kapillare bedingt durch die Oberflächenspannung an. Dieser Anstieg wird durch den hydrostatischen Druck begrenzt. Die Oberflächenspannung kann mittels der GauÃ-Laplace''schen Gleichung berechnet werden.          \r\n         Suspension \r\n          \r\n          \r\n          Aufschwemmung, siehe Dispersion.        \r\n         \r\n        \r\n        \r\n        \r\n        T\r\n         Tangentenmethode \r\n\r\n\r\nMethode zur Bestimmung des  Kontaktwinkels am  Dreiphasenpunkt  eines zu vermessenden Tropfens.\r\n\r\n         Tenside \r\n          \r\n          \r\n          Oberflächenaktive Stoffe mit heterogen aufgebauten Molekülen mit einem  hydrophilen Teil (meistens der Kopf) und einem hydrophoben Teil (Alkylkette).\r\n          \r\n         Tensiometer \r\n          \r\n          \r\n          MeÃgerät zur Bestimmung von Oberflächen- und Grenzflächenspannungen, z. B. nach der Wilhelmy Methode oder Du Noüy Ring Methode. Moderne elektronische Tensiometer können Benetzungseigenschaften auch nach der Washburn Methode und andere komplexere Aufgaben durchführen.\r\n          \r\n         Tropfenbasisdurchmesser \r\n          \r\n          \r\n          Durchmesser des Tropfens an der Berührungsfläche (Grenzfläche) zum benetzten Material, bei achsensymmetrischen Tropfen auch der Abstand zwischen dem rechten und linken Dreiphasenpunkt.\r\n          Tropfenkontur \r\n          \r\n          \r\n          Optische Projektion (UmriÃ) eines Tropfens.\r\n          \r\n         Tropfenvolumen \r\n          \r\n          \r\n          Volumen eines Tropfen; kann bei bestimmten Voraussetzungen zur (axialen) Symmetrie des Tropfens auch aus der Tropfenkontur berechnet werden.\r\n          Tropfenvolumen Methode \r\n          \r\n          \r\n          Methode zur Bestimmung der dynamischen Grenzflächenspannung als Funktion der Dosierrate bzw. des Tropfenalters, der Temperatur oder Konzentration.\r\n        \r\n        \r\n        \r\n        \r\n        U\r\n         Unpolare Anteile\r\n\r\n                 \r\n        Siehe Disperse Anteile.\r\n         Umgekehrte Osmose  \r\n          Bezeichnet ein Verfahren zur Entsalzung von Lösungen, z. B. Meerwasser, mit Hilfe einer speziell strukturierten Membran. \r\n        \r\n        \r\n        \r\n        V\r\n         Van der Waalssche Kräfte (nach J. D. van der Waals, 1873)  \r\n          Atomare und molekulare Anziehungskräfte, die durch den Dipoleffekt (zwischen permanenten Dipolen), den Induktionseffekt (zwischen permanenten und induzierten Dipolen) und den Dispersionseffekt (zwischen induzierten Dipolen) hervorgerufen werden. Siehe auch Israelachvili (1991): Chap. 11 - van der Waals Forces between Surfaces.\r\n         Viskosität\r\n          \r\n           \r\n          Rheologische GröÃe, die den FlieÃwiderstand einer Substanz beschreibt.\r\n        \r\n        \r\n        \r\n        W\r\n         \r\n \r\nWashburn-Gleichung (nach E.W. Washburn, 1921) \r\n\r\n\r\nbeschreibt das Fortschreiten einer Flüssigkeitsfront h(t) in  benetzbaren porösen Materialien: h(t) ~ Ãt.\r\n         Washburn Methode (modifizierte Washburn Methode) \r\n          \r\n          \r\n          MeÃmethode zur Bestimmung der Benetzungsgeschwindigkeit und des Kontaktwinkels von adsorbierenden Materialien (Pulver, Faserbündel usw.) mittels der Washburn-Gleichung oder einer ihrer Modifikationen.          \r\n         Wasserstoffbrücken-Bindung \r\n          \r\n          \r\n          Bezeichnet eine wichtige Form von Nebenvalenzbindung. Diese bildet sich zwischen einem an elektronegativen Element kovalent gebundenen Wasserstoff-Atom und dem einsamen Elektronenpaar eines anderen elektronegativen Atoms aus.          \r\n         Wechselwirkung \r\n          \r\n          \r\n          Intramolekulare oder intermolekulare Anziehung oder AbstoÃung von verschiedenen oder gleichartigen Teilchen.          \r\n         Wilhelmy-Methode (nach L. Wilhelmy, 1863) \r\n          \r\n          \r\n          Methode zur Bestimmung der statischen und der langsam veränderlichen Oberflächen- und Grenzflächenspannung von Flüssigkeiten mittels einer gut benetzbaren Platte. Es wird die Kraft gemessen, die bei der Benetzung der vertikal freihängenden Wilhelmy-Platte auftritt.          \r\n         Wilhelmy-Platte \r\n          \r\n          \r\n          Gerade oder zylindrische Platte aus Platin oder anderen gut benetzbaren, stabilen Materialien zur Durchführung von Messungen nach der Wilhelmy-Methode.\r\n        \r\n        \r\n        \r\n        X\r\n         Keine Einträge verfügbar.\r\n        \r\n        \r\n        \r\n        Y\r\n         Youngsche Gleichung (nach T. Young, 1805)  \r\n          Relation zwischen der  Oberflächenenergie   gS, der Grenzflächenenergie  gSL, eines ebenen Festkörpers sowie dem  Kontaktwinkel   Q   und der  Oberflächenspannung gLV  eines darauf befindlichen Flüssigkeitstropfens:  gS  = gSL  + gLV  cos Q\r\n         Young-Duprésche Gleichung (nach T. Young, 1805 und A. Dupré, 1869)\r\n          \r\n           \r\n          Beziehung zwischen der Adhäsionsarbeit WA, der Oberflächenspannung gLV und dem Kontaktwinkel Q: WA= gLV (1+cosQ) eines liegenden Tropfens.          \r\n         Young-Laplace Fit\r\n          \r\n           \r\n          Numerische Anpassung einer Tropfenkontur (z.B. für liegende, hängende und rotierende Tropfen oder Flüssigkeitslamellen) an die Lösung der Young-Laplace-Gleichung zur Ermittlung des Kontaktwinkels oder der Ober- bzw. Grenzflächenspannung.          \r\n         Young-Laplace-Gleichung (nach P. S. de Laplace, 1806 und T. Young, 1805)  \r\n          Statische oder auch dynamische Gleichgewichtsrelation zwischen dem auftretenden Kapillardruck Dp sowie zusätzlichen wirkenden Drücken (Schweredruck, Inertialdruck usw.) an gekrümmten Phasengrenzen, wie Tropfen, Blasen und Lamellen.\r\n        \r\n        \r\n        \r\n        Z\r\n         Zisman-Plot (nach W.A. Zisman, 1950)\r\n\r\n         \r\n        Graphische Darstellung zur Ermittlung der  kritischen Oberflächenspannung (eigentlich der Freien Oberflächenenergie) eines Festkörpers nach W.A. Zisman (1950-52), Die Abszisse (x-Achse)  trägt die  Oberflächenspannung der benutzten Testflüssigkeiten, die Ordinate (y-Achse) dagegen den Kosinus der gemessenen  Kontaktwinkel  cos Q. Der Wert auf der Achse der Oberflächenspannungen, bei dem die Bedingung des  Spreitens für den aufgebrachten Tropfen auf dem Festkörper erstmalig eintritt (cos Q    = 1), wird  kritische Oberflächenspannung genannt. \r\n        \r\n        \r\n       \r\n \r\n   \r\n      \n', '', 'page', '', '', 1429537054, 229, 12, 'Glossar', 0, '', ''),
(208, 5, 1431355495, 1428385888, 0, 0, 0, '', '231', '\nSoftware Download\r\n Benötigen Sie ein Software-Update einmal besonders schnell, so können wir es auf unserem FTP-Server für Sie zum Download bereitstellen.\n', '', 'page', '', '', 1429687566, 231, 12, 'Software-Updates', 0, '', ''),
(209, 5, 1431355495, 1428385888, 0, 0, 0, '', '232', '\nDataphysics-Hotline\r\nWir sind unter der Telefonnummer&nbsp;+ 49 (0)711-770556-55&nbsp;\r\nMo - Fr&nbsp;von&nbsp;9.00&nbsp;bis17.00 Uhr&nbsp;für Sie da.\r\n\r\nHausanschrift:\r\nDataPhysics Instruments GmbH&nbsp; Raiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0 Telefax: ++49 (0)711 770556-99 Internet: www.dataphysics.de&nbsp;  info@dataphysics.de\n', '', 'page', '', '', 1429687284, 232, 12, 'Service', 0, '', ''),
(236, 5, 1431355495, 1431328019, 0, 0, 0, '', '255', 'SVT 20 N\nSpinning Drop Video Tensiometer.\r\n The spinning drop video tensio­meter, SVT 20N, is a special purpose optical instrument for measuring of conventional to extremely low interfacial tensions.&nbsp; The control/data analysis software is designed for intuative use and fast access to all control elements.&nbsp; Combined with the software packages SVTS 20, SVTS 21, and SVTS 22 the SVT 20N provides the following functions: Automatic control of the electronic tilting base, for positioning of the rotating droplet along the rotary axis of the capillary Automatic variation of the rotation speed Measurement and analysis of the static, time, and temperature dependent interfacial tension (1·10-6&nbsp;to 2·103&nbsp;mN/m) according to the spinning&nbsp; drop method and oscillating spinning&nbsp; drop method. Measurement and analysis of the 2D and 3D extensional relaxation of visco-elastic liquids and membrane encapsulated liquids. drops by variation of the rotational speed. Determination of the yield point of liquid crystalline. Determination of the dispersive and polar parts of liquids on the basis of interfacial tension, with error limits. Calculation of the geometry of rotating drops related to the rotary axis.  Features of the basic instrument configuration: High dynamic, electronically commutated direct current motor with software controlled speed control (0 to 20?000 rev/min; ± 0.001 rev/min)&nbsp; to realise spped steps (± 2?000 rev/min/s) and sinusoidal speed variations (0.01 to 200 Hz) for relaxation measurements. High-performance 6-fold power zoom lens with an integrated continuous fine focus and adjustable observation angle. Video measuring system with USB-camera (max. 146 images/s sample rate). USB connection to PC (or Notebook). LED lighting with software controlled adjustable intensity and strobe frequency. Integrated touch screen for controlling basic functions. Automatic, software controlled position and tilt adjustment. Automatic calibration function for correcting optical distortions caused by the capillaries (cylindrical lens effects).  Optional SVT accessories: Easy mountable liquid/gas-temperature controlled measuring cell MC TFC 130; temperature range&nbsp; -10 to 130 °C. Easy mountable peltier/gas-temperature controlled measuring cell MC TPC 180; temperature range -30 to 180 °C; ±0.1 K resolution with a heat up and cool down rate of ±1 K/s. Fast-exchangeable special high-temperature glass capillaries (up to 180 °C) FEC&nbsp;622/4 0-HT. Cost-effective (disposable) fast-exchange capillaries made of polycarbonate (up to 60 °C) FEC&nbsp;780/400-P. Automatic dosing system ES-D/S. Calibration standard DCS-SVT. The available software modules for the SVT 20N are: SVTS 20&nbsp; - Interfacial Tension. SVTS 21 - Oscillation. SVTS 22 - Geometry Data Sheet SVT 20 N\n', '', 'page', '', '', 1429699559, 255, 181, 'SVT 20 N', 3, '', ''),
(211, 5, 1431355495, 1428385888, 0, 0, 0, '', '235', 'Travail d''équipe - Clé à notre succès\nPour jouer bien comme une organisation innovatrice, nous maintenons un ouvert coopèrent la structure dans laquelle chaque individu est encouragé à développer ses/ses propres connaissances.&nbsp;La grande portée est fournie pour permettre l''application intégrale de nouvelles idées créatrices.\nSeulement le meilleur pour vous!\nLa satisfaction de client est la clé de notre philosophie. Nos instruments, accessoires et modules de logiciel sont spécifiques composé aux exigences individuelles et aux demandes.&nbsp;Nous travaillons continuellement pour arriver au meilleur résultat pour l''existant et ''yet pour venir'' les défis.&nbsp;Chaque point levé pendant le processus des ventes initial ou pendant l''utilisation d''instrumentation installée sera rencontré avec une analyse détaillée de sentier de solution et d''exigences.\r\nEn conséquence de ces efforts nous sommes capables de rester au premier rang de cette science de développement.Nous plaçons de très hautes demandes sur notre qualité et intégrité. Nous voulons garantir votre succès - donc nous fournissons notre meilleur, chaque jour.\r\nPrenez-nous à notre mot!\n', '', 'page', '', '', 1429537138, 235, 13, 'Philosophie', 3, '', ''),
(212, 5, 1431355495, 1428385888, 0, 0, 0, '', '235', 'Teamwork - Key to our success\nIn order to perform well as an (at all times) innovative organisation, we maintain an open cooperate structure in which each individual is encouraged to develop his/her own skills.&nbsp;\r\nGreat scope is provided to enable the full implementation of new creative ideas and integrate new acquired know-how within the ''group''.\nOnly the best for our customers\nCustomer satisfaction is key to the philosophy or our company. Wherever possible our instrument, accessory and software development is led by specific customer requirements and requests. Those of you who have trusted your companies money and your own reputations to DataPhysics and its products can be assured that we will continually work with you to arrive at the very best outcome for existing and ''yet to come'' challenges. Each point raised during the initial sales process or during use of installed instrumentation will be met with a thorough analysis of requirements and exhaustive, application led, solution path.\r\nAs a consequence of these efforts and the faultless lines of communication within all departments within our company, DataPhysics (and yourselves) are able to remain at the forefront of this developing science.\r\nTo ensure the most rapid return on your investment and minimise/eliminate on-going costs to your businesses, we place very high demands on our quality and reliability. We want to ensure your success - therefore we provide our best, each and every day.\r\nTake us at our word!\n', '', 'page', '', '', 1429537138, 235, 13, 'Philosophy', 2, '', ''),
(213, 5, 1431355495, 1428385888, 0, 0, 0, '', '236', 'Understanding Interfaces\nThe advent of nanotechnology has resulted in an increased understanding of solid-liquid interactions such as surface wetting. This knowledge has important implications for preventing wear on engines, paintings and buildings, for medical products such as contact lenses, surgical tools and prosthetics, as well as for self-cleaning windows, cars and clothes. Our ability to observe and create specific nanoscale surface topographies has enabled to us to dramatically improve our understanding and control surface wetting.\nDataPhysics in brief\nThere are many things to tell you about our company. This internet presence, as well as the brochures you can download from this web-site, will give you a comprehensive overview on our competences, innovative power and unique product range. A short summary to introduce our company:\r\n Our company was founded in 1997 in Filderstadt near Stuttgart in the south of Germany. Our core competence is the application of the newest measurement techniques for surface and interfacial science. Development, manufacturing and worldwide sales of laboratory and process instruments for surface and interfacial science. Broad product range of manual and automated instruments for the measurement of surface properties. Our own application centre for surface science: Individual training, contract measurements and excellent after-sales application support. We are a highly respected partner, worldwide, within research institutes, universities and industry.\n', '', 'page', '', '', 1429537281, 236, 13, 'A propos de nous', 3, '', ''),
(214, 5, 1431355495, 1428385888, 0, 0, 0, '', '236', 'Understanding Interfaces\nThe advent of nanotechnology has resulted in an increased understanding of solid-liquid interactions such as surface wetting. This knowledge has important implications for preventing wear on engines, paintings and buildings, for medical products such as contact lenses, surgical tools and prosthetics, as well as for self-cleaning windows, cars and clothes. Our ability to observe and create specific nanoscale surface topographies has enabled to us to dramatically improve our understanding and control surface wetting.\nDataPhysics in brief\nThere are many things to tell you about our company. This internet presence, as well as the brochures you can download from this web-site, will give you a comprehensive overview on our competences, innovative power and unique product range. A short summary to introduce our company:\r\n Our company was founded in 1997 in Filderstadt near Stuttgart in the south of Germany. Our core competence is the application of the newest measurement techniques for surface and interfacial science. Development, manufacturing and worldwide sales of laboratory and process instruments for surface and interfacial science. Broad product range of manual and automated instruments for the measurement of surface properties. Our own application centre for surface science: Individual training, contract measurements and excellent after-sales application support. We are a highly respected partner, worldwide, within research institutes, universities and industry.\n', '', 'page', '', '', 1429537281, 236, 13, 'Figures & Facts', 2, '', '');
INSERT INTO `tx_kesearch_index` (`uid`, `pid`, `tstamp`, `crdate`, `cruser_id`, `starttime`, `endtime`, `fe_group`, `targetpid`, `content`, `params`, `type`, `tags`, `abstract`, `sortdate`, `orig_uid`, `orig_pid`, `title`, `language`, `directory`, `hash`) VALUES
(215, 5, 1431355495, 1428385888, 0, 0, 0, '', '237', 'Références utiles\nIci vous trouvez quelques références qui pourraient vous aider à obtenir plus d''information de la science d''interface et de la surface.&nbsp;\r\n http://fim.iccas.ac.cn/\r\nSite internet d''Interface material group fonctionnelle d''académie chinoise de science\r\n http://www.igb.fraunhofer.de/en.html\r\nSite internet de l''Institut de Fraunhofer allemand pour l''ingénierie interdu visage et la biotechnologie\nFoires & Conférences \nSaisissez l''occasion pour nous rencontrer directement! Nous assistons régulièrement à plusieurs expositions et à conférences scientifiques. \r\nTrouvez s''il vous plaît au-dessous d''une liste pour les événements prochains où nous allons en présentant nos appareils de mesure hauts et innovateurs.\nForum LABO & BIOTECH 2015, France - 21.04. - 23.04.2015\n Le commerce équitable avec DataPhysics Instruments surviendra à Paris l''expo Porte de Versailles.\r\n \r\n Les produits, les solutions et les  innovations seront présentés dans les champs de recherche, analyse et  contrôle. Les domaines d''utilisation sont &nbsp;pour examples de les  biotechnologies et l''industrie chimique.&nbsp;\n', '', 'page', '', '', 1429694621, 237, 14, 'Foires & Conférences', 3, '', ''),
(216, 5, 1431355495, 1428385888, 0, 0, 0, '', '237', 'Useful Links\nContained here are links which might help you to find more information on surfaces and interfaces.&nbsp;\r\n www.igb.fraunhofer.de\r\nWebsite of the Fraunhofer Institute for Interfacial engineering and Biotechnology\r\n www.mpip-mainz.mpg.de\r\nWebsite of the Max Planck Institute for Polymer Research\r\n http://fim.iccas.ac.cn/\r\nWebsite of functional Interface Material Group of Chinese Academy of Science\nExhibitions & conferences\nAn excellent opportunity to meet with us, directly. We plan to attend several exhibitions and scientific conferences in the coming months and will update this site as time passes. Please find below a list for the upcoming events where we will be available.\nForum LABO & Biotech 2015, France - 31st of March - 2nd of April\nAt 31st - 2nd of April this years Forum LABO Expo takes place in Paris.  DataPhysics Instruments will be at the location, repesented at  the  exhibition by our partner Apollo Instruments.\nSurface Technology 2015, Germany - 13th - 17th of April\nSurface Technology 2015 The Surface Technology 2015 will take place in Hannover from 13th to 17th of April. DataPhysics will be on the spot an present you the newest advance is surface scienes and surface charackterisation.\r\n\r\nWe will be happy to welcome you at our booth F34.\nEuropean Coating Show 2015, Germany - 21st - 23rd of April\nECS 2015 At the European Coating Show 2015 in Nürnberg DataPhysics will prenset to you the newst instruments in the field of coating and surface charackterisation and give you an introduction in the related measurment methods.\nSPE Western Regional Meeting 2015, USA - 27th - 30th of April\nSPE From 27th-30th of April the Society of Petroleum Engineers will hold the Western Regional Meeting in California. DataPhysics will be there to present you the newest instrumentation for interface characterisation. We are looking forward to see you at our booth (Nr. 27).\n15th Conference of the International Association of Colloid and Interface Scientists, Germany - 24th - 29th of May\nThe 15th Conference of the International Association of Colloid and Interface Scientists is held from May 24-29 in Mainz. IACIS 2015 is coorganized together with the German Colloid Society. DataPhysics Instruments will be there to present you the newest instrumentation in the field.\r\n\nCESIO 2015, Turkey - 1st - 3rd of June\nCESIO 2015 This event is organised on behalf of the European Committee of Surfactants and their Organic Intermediates (CESIO), a sector group of CEFIC. We will be on the spot to present you the newst measuring devices for surface analyses.\nColloid and Surface Science Symposium, USA - 15th - 17th of June\nThe 89th annual meeting of the American Chemical Society Division of Colloid and Surface Chemistry will take place in Pittsburgh this year. DataPhysics represantative Future Digital Scientific will welcome you there to discuss the newest advances in the field.\nECOSS 31, Spain - 31st of August - 4th of September\nThe 31. European Conference on Surface Science will take place in Barcelona. DataPhysics Instruments will be represented by our partner NEURTEK. We will be happy to see you at our booth No. 31!\nThementage Grenz- und Oberflächentechnik - Germany - 15th - 17th of September\nThGOT At the 11th conference on surface and interface techniques in Zeulenroda DataPhysics Instruments will be at the location and present to you the newest instrumentation in this field of expertise.\r\n\r\nDataPhysics will be sponsoring the event this year again and also will participate with a scientific talk.\n', '', 'page', '', '', 1429694621, 237, 14, 'Exhibitions & Conferences', 2, '', ''),
(217, 5, 1431355495, 1428385888, 0, 0, 0, '', '238', 'Adress\nDataPhysics Instruments GmbH\r\nRaiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0&nbsp; Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de E-Mail:&nbsp; sales@dataphysics.de\n\n Directions\n\n Représentation mondiale\nN''hésitez pas s''il vous plaît à nous contacter directement. Nous sommes heureux d''aider vous par téléphone ou l''e-mail.\nVentes\r\nppa. Nils Langer Phone: + 49-711-770556-51 Fax: + 49-711-770556-99 Email:&nbsp; n.langer@dataphysics.de\r\nDr. Peter Oberschachtsiek Phone: + 49-711-770556-52 Fax: + 49-711-770556-99 Email:&nbsp; p.oberschachtsiek@dataphysics.de\r\nMarina Langer Phone: + 49-711-770556-53 Fax: + 49-711-770556-99 Email:&nbsp; m.langer@dataphysics.de&nbsp;\r\nDaniel Scholz Phone: + 49-711-770556-56 Fax: + 49-711-770556-99 Email:&nbsp; d.scholz@dataphysics.de&nbsp;\r\nDina Otto Phone: + 49-711-770556-58 Fax: + 49-711-770556-99 Email:&nbsp; d.otto@dataphysics.de&nbsp;\r\n\r\nSoutien technique\r\nGerhard Maier Phoe: + 49-711-770556-55 Fax: + 49-711-770556-99 Email:&nbsp; service@dataphysics.de\r\n\r\nLaboratoire d''application\r\nNorbert Heil Phone: + 49-711-770556-66 Fax: + 49-711-770556-99 Email:&nbsp; n.heil@dataphysics.de&nbsp;\r\n\r\nDirection générale\r\nHorst Rau Phone: + 49-711-770556-10 Fax: + 49-711-770556-99 Email:&nbsp; h.rau@dataphysics.de\r\nTorsten Holz Phone: + 49-711-770556-30 Fax: + 49-711-770556-99 Email:&nbsp; t.holz@dataphysics.de\r\n\r\nAdministration\r\nJulia Scheler Phone: + 49-711-770556-0 Fax: + 49-711-770556-99 Email:&nbsp; j.scheler@dataphysics.de\n', '', 'page', '', '', 1429352452, 238, 15, 'Interlocuteur', 3, '', ''),
(218, 5, 1431355495, 1428385888, 0, 0, 0, '', '238', 'Adress\nDataPhysics Instruments GmbH\r\nRaiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0&nbsp; Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de E-Mail:&nbsp; sales@dataphysics.de\n\n Directions\n\n Distributors Worldwide\nPlease fell free to contact us, directly. We are happy to assist you by telephone or email\nSales:&nbsp;\r\nNils Langer Phone: + 49-711-770556-51 Fax: + 49-711-770556-99 Email:&nbsp; n.langer@dataphysics.de\r\nDr. Peter Oberschachtsiek Phone: + 49-711-770556-52 Fax: + 49-711-770556-99 Email:&nbsp; p.oberschachtsiek@dataphysics.de\r\nDaniel Scholz Phone: + 49-711-770556-56 Fax: + 49-711-770556-99 Email:&nbsp; d.scholz@dataphysics.de&nbsp;\r\nDina Otto&nbsp; Phone: + 49-711-770556-58 Fax: + 49-711-770556-99 Email:&nbsp; d.otto@dataphysics.de\r\n\r\nTechnical Support:\r\nGerhard Maier Phoe: + 49-711-770556-55 Fax: + 49-711-770556-99 Email:&nbsp; service@dataphysics.de\r\n\r\nApplication Laboratory:\r\nNorbert Heil Phone: + 49-711-770556-66 Fax: + 49-711-770556-99 Email:&nbsp; n.heil@dataphysics.de&nbsp;\r\n\r\nGeneral Management:\r\nHorst Rau Phone: + 49-711-770556-10 Fax: + 49-711-770556-99 Email:&nbsp; h.rau@dataphysics.de Torsten Holz Phone: + 49-711-770556-30 Fax: + 49-711-770556-99 Email:&nbsp; t.holz@dataphysics.de\r\n\r\nAdministration:\r\nJulia Scheler Phone: + 49-711-770556-0 Fax: + 49-711-770556-99 Email:&nbsp; j.scheler@dataphysics.de\r\nInga Saier Phone: + 49-711-770556-0 Fax: + 49-711-770556-99 Email:&nbsp; i.saier@dataphysics.de\n', '', 'page', '', '', 1429352452, 238, 15, 'Contact persons', 2, '', ''),
(219, 5, 1431355495, 1428385888, 0, 0, 0, '', '238', 'Ansprechpartner\nGeschäftsleitung\r\nTorsten Holz Telefon: +49 (0)711 770556-30  t.holz@dataphysics.de\r\nHorst Rau Telefon: +49 (0)711 770556-10  h.rau@dataphysics.de\r\n\r\nVertrieb\r\nNils Langer Telefon: +49 (0)711 770556-51  n.langer@dataphysics.de\r\nDr. Peter Oberschachtsiek Telefon: +49 (0)711 770556-52  p.oberschachtsiek@dataphysics.de\r\nDaniel Scholz Telefon: +49 (0)711 770556-56  d.scholz@dataphysics.de\r\nDina Otto Telefon: +49 (0)711 770556-58  d.otto@dataphysics.de\r\n\r\nTechnical Support / Service\r\nGerhard Maier Telefon: +49 (0)711 770556-45  g.maier@dataphysics.de\r\n\r\nApplikationslabor\r\nNorbert Heil Telefon +49 (0)711 770556-66  n.heil@dataphysics.de\r\n\r\nAuftragsabwicklung\r\nJulia Scheler Telefon: +49 (0)711 770556-12  j.scheler@dataphysics.de\n\nHausanschrift\r\nDataPhysics Instruments GmbH Raiffeisenstrasse 34 D 70794 Filderstadt Telefon: ++49 (0)711 770556-0 Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de  sales@dataphysics.de\n\n Anfahrt\n\n Vertretungen weltweit\n', '', 'page', '', '', 1429352452, 238, 15, 'Ansprechpartner', 0, '', ''),
(220, 5, 1431355495, 1428385888, 0, 0, 0, '', '239', 'Adress\nDataPhysics Instruments GmbH\r\nRaiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0&nbsp; Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de E-Mail:&nbsp; sales@dataphysics.de\nReprésentation mondiale Directions\n\n\n\r\n        \r\n        \r\n        \r\n        \r\n        \r\n          \r\n        \r\n        \r\n        \r\n          Se il vous plaît choisir un continent\r\n          Europe\r\n            Amérique\r\n            Afrique\r\n            Asie\r\n            Australie\r\n        \r\n        \r\n        \r\n        \r\n          Se il vous plaît choisir un pays\r\n          Belgique, Luxembourg\r\n            Danemark\r\n            Allemagne\r\n            France\r\n            Royaume-Uni\r\n            Irlande\r\n            Italie\r\n            Pays-Bas\r\n            Norvège, la Suède, la Finlande\r\n            Pologne\r\n            Portugal\r\n            La Russie et la CEI\r\n            Suisse, le Liechtenstein, Austria\r\n            Espagne\r\n            Hongrie\r\n        \r\n        \r\n        \r\n          Se il vous plaît choisir un pays\r\n          Argentine\r\n            Brésil\r\n            Chili\r\n            Equateur\r\n            Canada\r\n            Mexique\r\n            Ãtats-Unis\r\n        \r\n        \r\n        \r\n          Se il vous plaît choisir un pays\r\n          Afrique du Sud\r\n        \r\n        \r\n        \r\n          Se il vous plaît choisir un pays\r\n          Chine\r\n            Inde\r\n            Iran\r\n            Israël\r\n            Japon\r\n            Malaisie\r\n            Singapour, l''Indonésie\r\n            Corée du Sud\r\n            Taiwan\r\n            Thaïlande\r\n        \r\n        \r\n        \r\n          Se il vous plaît choisir un pays\r\n          Australie\r\n        \r\n        \r\n        \r\n        \r\n        Belgium / Luxembourg\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-56 \r\n        Fax: +49-711-770556-99 \r\n        d.scholz@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Contact: Daniel Scholz \r\n        \r\n        \r\n        \r\n        Denmark\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Contact: Dr. Peter Oberschachtsiek\r\n\r\n        \r\n        \r\n        \r\n        Germany\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-0 \r\n        Fax: +49-711-770556-99  \r\n\r\n        \r\n        Region: South / Central / East Germany (postcode areas: 0, 1, 33-37, 5, 6, 7, 8, 9) \r\n        Contact: Dr. Peter Oberschachtsiek, dial -52 \r\n        p.oberschachtsiek@dataphysics.de  \r\n        \r\n        Region: North / West Germany (postcode areas: 2, 30-32, 38-39, 4) \r\n        Contact: Daniel Scholz, dial -56 \r\n        d.scholz@dataphysics.de\r\n\r\n        \r\n        \r\n        \r\n        France\r\n        Apollo Instruments \r\n        3 rue Clément Ader \r\n        ZAC Mercières \r\n        F-60200 Compiègne \r\n        Telephone: +33-3-44426619 \r\n        Fax: +33-3-44852472 \r\n        info@apolloinstruments.fr \r\n        Internet: http://www.apolloinstruments.fr \r\n        Contact: Dr. Romuald Dez  \r\n\r\n        \r\n        \r\n        \r\n        UK\r\n        Gerhardt UK \r\n        C. Gerhardt UK Ltd. \r\n        Unit 5, Avonbury Court \r\n        County road, Brackley \r\n        Northants, NN 13 7AX \r\n        Telefon: +44-1280-706772 \r\n        Fax: +44-1280-706088 \r\n          info@gerhardtonline.co.uk \r\n        Internet: http://www.cgerhardt.com \r\n        Contact: Paul Campbell \r\n\r\n        \r\n        \r\n        \r\n        Ireland\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: + 49-711-770556-51 \r\n        Fax: + 49-711-770556-99 \r\n        n.langer@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Contact: Nils Langer \r\n\r\n\r\n        \r\n        \r\n        \r\n        Italy\r\n        NG Labtec Srl. \r\n        Via Carlo Serassi, 21 \r\n        IT-24100 Bergamo \r\n        Telefon: +39-035-4175268 \r\n        Fax:  +39-035-4534445 \r\n        n.guerra@nglabtec.com \r\n        Internet: http://www.nglabtec.com \r\n        Contact: Norberto Guerra  \r\n\r\n        \r\n        \r\n        \r\n        The Netherlands\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-56 \r\n        Fax: +49-711-770556-99 \r\n        d.scholz@dataphysics.de \r\n        Web: www.dataphysics.de \r\n        Contact: Daniel Scholz\r\n\r\n        \r\n        \r\n        \r\n        Norway, Sweden, Finland\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact: Dr. Peter Oberschachtsiek \r\n\r\n        \r\n        \r\n        \r\n        Poland\r\n        Spectro-Lab \r\n        ul. Warszawska 100/102 \r\n        05-092 Lomianki \r\n        Telefon: +48 22 675 25 67 \r\n        Fax: +48 22 811 98 18 \r\n        artem.khvorostov@spectro-lab.pl \r\n          Internet: www.spectro-lab.com \r\n        Contact: Artem Khvorostov\r\n\r\n        \r\n        \r\n        \r\n        Portugal\r\n        Norleq - Equipamentos e Servicos \r\n        Rua Formosa 49 \r\n        3-sala 6 \r\n        4000-250 Porto \r\n        Telefon: +351-222081483 \r\n        Telefax: +351-222081768 \r\n        info@norleq.com \r\n        Internet: http://www.norleq.com \r\n        Contact: Adao Araujo\r\n\r\n      \r\n        \r\n        \r\n        Russia\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70704 Filderstadt \r\n        Phone: +49-711-770556-53 \r\n        Fax: +49-711-770556-99 \r\n        m.langer@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact:  Marina Langer (spricht Russisch)\r\n\r\n      \r\n        \r\n        \r\n        Switzerland, Liechtenstein, Austria\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact: Dr. Peter Oberschachtsiek\r\n\r\n        \r\n        \r\n        \r\n        Spain\r\n        NEURTEK S. A. \r\n        Pol. Ind Azitain Parc. 3a \r\n        ES-20600 Eibar \r\n        Telefon: +34-902-420082 / +34-943-8200-82 \r\n        Fax: +34-943-820157 \r\n        rafael@neurtek.es \r\n        Internet: http://www.neurtek.es \r\n        Contact: Rafael M. Atxurra\r\n\r\n        \r\n        \r\n        \r\n        Hungary\r\n        Laborexport Ltd. \r\n        Ibrahim str. 8 \r\n        HU-1113 Budapest \r\n        Telefon: +36-1-209-6424 \r\n        Fax: +36-1-209-6424 \r\n        sales@laborexport.hu \r\n        Internet: http://www.laborexport.hu \r\n        Contact: Dr. Tibor Müller\r\n\r\n      \r\n        \r\n        \r\n        Argentina\r\n        LAB CONTROL Ltda. \r\n        Rua José de Carvalho, 294 \r\n        Cep 04714-020 \r\n        Chácara Santo Antonio \r\n        Sao Paulo - Brasil \r\n        Telefon: +55-11-5181-1173 \r\n        vendas@labcontrol.com.br \r\n        Internet: http://www.labcontrol.com.br \r\n        Contact: Carlos-Alberto Maciel\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Brazil\r\n        LAB CONTROL Ltda. \r\n        Rua José de Carvalho, 294 \r\n        Cep 04714-020 \r\n        Chácara Santo Antonio \r\n        Sao Paulo - Brasil \r\n        Telefon: +55-11-5181-1173 \r\n        vendas@labcontrol.com.br \r\n        Internet: http://www.labcontrol.com.br \r\n        Contact: Carlos-Alberto Maciel \r\n\r\n  \r\n      \r\n        \r\n        \r\n        Chile\r\n        Melvyn Becerra &amp; Cia. Ltda. \r\n        Instrumentos Cientificos \r\n        Dr. Torres Boonen 834 \r\n        CL-Providencia Santiago \r\n        Telefon: +56-2-3414-233 / 2093-492 \r\n        Fax: +56-2-2059-596 \r\n        melbec1@vtr.net \r\n        Internet: http://www.melvynbecerra.cl \r\n        Contact: Melvyn Beccera \r\n\r\n  \r\n      \r\n        \r\n        \r\n        Ecuador\r\n        Jose M. Jalil Haas/ JJ&amp;H  Cia Ltda. \r\n        Pasaje Frey Joaquin Auz 181 (E07-08) \r\n        Quito \r\n        Telefon: +593-2-2244583, +593-2-2244583 \r\n        Fax: +593-2-2442380 \r\n        josehalil@jhjalil.com \r\n        Internet: www.jhjalil.com \r\n        Contact: Jose Jalil\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Canada\r\n        Folio Instruments Inc. \r\n        1645 St-Patrick \r\n        Montrela, Quebec, H3K3G9 \r\n        Telefon: +1-514-937-7931 \r\n        Fax: +1-514-937-4533 \r\n        rbarriere@folioinstruments.com \r\n        Internet: http://www.folioinstruments.com \r\n        Contact: Rhett Barriere\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Mexico\r\n        Conceptos e Instrumentos S.A. de C.V. \r\n        Creston 212-4 \r\n        Jardines del Pedregal \r\n        Mexico D. F. 01900 \r\n        Telefon: +52-55-55681671 oder 55681295 \r\n        conceptoseinstrumentos@gmail.com \r\n        Internet: www.conceptoseinstrumentos.com \r\n        Contact: José Antonio Blanco\r\n\r\n  \r\n      \r\n        \r\n        \r\n        USA\r\n        Future Digital Scientific Corp. \r\n        320 Endo Blvd. Suite 2B \r\n        Garden City \r\n        US-New York 11530 \r\n        Telefon: +1-516-349-0663 \r\n        Fax: +1-516-349-0642 \r\n        spencer@FDSC.com \r\n        Internet: http://www.FDSC.com \r\n        Contact: Dr. Spencer Wu\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        South Africa\r\n        Micron Scientific \r\n        175 Princes Avenue \r\n        Benoni, Gauteng, ZA \r\n        Telefon: +27-11-422-4303 \r\n        Fax: +27-11-845-2923 \r\n        trutera@ms.co.za \r\n        Internet: http://www.micronscientific.co.za \r\n        Contact: Albrecht Truter \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        China\r\n        EDIC\r\n        Beijing Eastern-Dataphy Instruments Co. Ltd. (EDIC) \r\n        Room 1010, Arstonal Tower \r\n        No.69 Zi Zhu Yuan Road Street \r\n        CN-100089 Beijing, Haidian District \r\n        Telefon: +86-10-68920269, 689202-57, -75, -76, -77  \r\n        Fax: +86-10-6892-0275 \r\n        robert_wang@edcc.com.cn \r\n        Internet: http://www.edcc.com.cn \r\n        Contact: Robert Wang \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        India\r\n        Goettfert DataPhysics Instruments India Pvt. Ltd. \r\n        P-84A, C.I.T. Road \r\n        Scheme-IV M \r\n        Calcutta-700 010 \r\n        Telefon: +919339867536 \r\n        basu@gdinstruments.co.in \r\n        Internet: http://www.gdinstruments.co.in \r\n        Contact: Bedanta Basu\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Iran\r\n        Mahar Fan Abzar Co. \r\n        Unit 16, No 40, Alley 13 \r\n        Miremad Street \r\n        Tehran, Iran \r\n        P.O. Box 15875-3781 \r\n        Telefon: ++98-21-88522420 and 88507022 \r\n        Fax: ++98-21-88522421 \r\n        piroozmarashi@maharfan.com \r\n        Contact: Dr. Pirooz Marashi\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Israel\r\n        Tetra Sense Scientific Tools \r\n        35 Hamoshav Rd \r\n        POB 887 \r\n        Carcur 37108 \r\n        Telefon: (+972)04 6272720 \r\n        Fax: (+972) 04 6272719 \r\n        moshe@tetrasense.co.il \r\n        Internet: http://www.tetrasense.co.il \r\n        Contact: Moshe Shmilovitz \r\n\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Japan\r\n        EKO Corporation \r\n        Sasazuka Center Bldg. 1-6 \r\n        Sasazuka 2-chome Sibuya-ku \r\n        JP-Tokyo 151 \r\n        Telefon: +81-3-5352-2911 \r\n        Fax: +81-3-5252-2917 \r\n        yazaki@eko.co.jp \r\n        Internet: http://www.eko.co.jp \r\n        Contact: Toshiaki Yazaki \r\n        \r\n        \r\n       \r\n        \r\n        \r\n        Malaysia\r\n        LMS\r\n        LMS Scientific Solutions Sdn Bhd. \r\n        D10-3-G, Block D10, \r\n        Pusat Perdagangan Dana 1, \r\n        Jalan PJU 1A/46, \r\n        47301 Petaling Jaya, \r\n        Selangor, Malaysia \r\n        Telefon: +603-7843 6533 \r\n        Fax: +603-7843 6522 \r\n         info@lmsscientific.com \r\n        Internet: www.lmsscientific.com \r\n        Contact: Jaez Lee\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Singapore\r\n        LMS Technologies Pte Ltd. \r\n        6 Tagore Drive \r\n        #04-05 Tagore Industrial Building \r\n        SG-787 623 Singapore \r\n        Telefon: +65-6-451-1123 \r\n        Fax: +65-6-452-1011 \r\n        anghg@lmstech.com.sg \r\n        Internet: www.lmsscientific.com \r\n        Contact: Francis Ang \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        South Korea\r\n        MCIK Co. - Material Characterization Instruments Korea \r\n        85-17 (2/2) Yeonhee-2 Dong \r\n        Seodaemun-Gu \r\n        KR-120-824 Seoul \r\n        Telefon: +82-2-3143-2740 / -1979 \r\n        Fax: +82-2-3143-2753 \r\n        srkim@mcik.co.kr \r\n        Internet: http://www.mcik.co.kr \r\n        Contact: Dr. Kim, Seungrok \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Taiwan\r\n        Toptical Scientific Corp. \r\n        7F-2, No 421, Sungshan Road \r\n        Taipei 11083, ROC \r\n        Telefon: + 886-2-2346-1510 \r\n        Fax: + 886-2-2346-1520 \r\n        toptical@ms17.hinet.net \r\n        Internet: www.toptical.com.tw \r\n        Contact: Reg Chen \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Thailand\r\n        LMS Instruments Co Ltd. \r\n        128/73 Phyathai Plaza Bldg.7th Floor \r\n        Phyathai Road, Thung-Phyathai \r\n        Radthavee \r\n        TH-Bangkok 10400 \r\n        Telefon: ++66-2-612-9091 \r\n        Fax: ++66-2-612-9093 \r\n        info@lmsinstr.com \r\n        Internet: www.lmsscientific.com \r\n        Contact: Sumate Tiratrakoonwichaya\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Australia\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-51 \r\n        Fax: +49-711-770556-99 \r\n        n.langer@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact: Nils Langer \r\n        \r\n        \r\n       \r\n      \n', '', 'page', '', '', 1429700971, 239, 15, 'Représentation mondiale', 3, '', ''),
(221, 5, 1431355495, 1428385888, 0, 0, 0, '', '239', 'Adress\nDataPhysics Instruments GmbH\r\nRaiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0&nbsp; Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de E-Mail:&nbsp; sales@dataphysics.de\nDistributors Worldwide\n\n\n\r\n        \r\n        \r\n        \r\n        \r\n        \r\n          \r\n        \r\n        \r\n        \r\n          Please choose a continent\r\n          Europe\r\n            America\r\n            Africa\r\n            Asia\r\n            Australia\r\n        \r\n        \r\n        \r\n        \r\n          Please choose a country\r\n          Belgium, Luxembourg\r\n            Denmark\r\n            Germany\r\n            France\r\n            UK\r\n            Ireland\r\n            Italy\r\n            Netherlands\r\n            Norway, Sweden, Finland\r\n            Poland\r\n            Portugal\r\n            Russia and CIS\r\n            Switzerland, Liechtenstein, Austri\r\n            Spain\r\n            Hungary\r\n        \r\n        \r\n        \r\n          Please choose a country\r\n          Argentina\r\n            Brazil\r\n            Chile\r\n            Ecuador\r\n            Canada\r\n            Mexico\r\n            USA\r\n        \r\n        \r\n        \r\n          Please choose a country\r\n          South Africa\r\n        \r\n        \r\n        \r\n          Please choose a country\r\n          China\r\n            India\r\n            Iran\r\n            Israel\r\n            Japan\r\n            Malaysia\r\n            Singapore, Indonesia\r\n            South Korea\r\n            Taiwan\r\n            Thailand\r\n        \r\n        \r\n        \r\n          Please choose a country\r\n          Australia\r\n        \r\n        \r\n        \r\n        \r\n        Belgium / Luxembourg\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-56 \r\n        Fax: +49-711-770556-99 \r\n        d.scholz@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Contact: Daniel Scholz \r\n        \r\n        \r\n        \r\n        Denmark\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Contact: Dr. Peter Oberschachtsiek\r\n\r\n        \r\n        \r\n        \r\n        Germany\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-0 \r\n        Fax: +49-711-770556-99  \r\n\r\n        \r\n        Region: South / Central / East Germany (postcode areas: 0, 1, 33-37, 5, 6, 7, 8, 9) \r\n        Contact: Dr. Peter Oberschachtsiek, dial -52 \r\n        p.oberschachtsiek@dataphysics.de  \r\n        \r\n        Region: North / West Germany (postcode areas: 2, 30-32, 38-39, 4) \r\n        Contact: Daniel Scholz, dial -56 \r\n        d.scholz@dataphysics.de\r\n\r\n        \r\n        \r\n        \r\n        France\r\n        Apollo Instruments \r\n        3 rue Clément Ader \r\n        ZAC Mercières \r\n        F-60200 Compiègne \r\n        Telephone: +33-3-44426619 \r\n        Fax: +33-3-44852472 \r\n        info@apolloinstruments.fr \r\n        Internet: http://www.apolloinstruments.fr \r\n        Contact: Dr. Romuald Dez  \r\n\r\n        \r\n        \r\n        \r\n        UK\r\n        Gerhardt UK \r\n        C. Gerhardt UK Ltd. \r\n        Unit 5, Avonbury Court \r\n        County road, Brackley \r\n        Northants, NN 13 7AX \r\n        Telefon: +44-1280-706772 \r\n        Fax: +44-1280-706088 \r\n          info@gerhardtonline.co.uk \r\n        Internet: http://www.cgerhardt.com \r\n        Contact: Paul Campbell \r\n\r\n        \r\n        \r\n        \r\n        Ireland\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: + 49-711-770556-51 \r\n        Fax: + 49-711-770556-99 \r\n        n.langer@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Contact: Nils Langer \r\n\r\n\r\n        \r\n        \r\n        \r\n        Italy\r\n        NG Labtec Srl. \r\n        Via Carlo Serassi, 21 \r\n        IT-24100 Bergamo \r\n        Telefon: +39-035-4175268 \r\n        Fax:  +39-035-4534445 \r\n        n.guerra@nglabtec.com \r\n        Internet: http://www.nglabtec.com \r\n        Contact: Norberto Guerra  \r\n\r\n        \r\n        \r\n        \r\n        The Netherlands\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-56 \r\n        Fax: +49-711-770556-99 \r\n        d.scholz@dataphysics.de \r\n        Web: www.dataphysics.de \r\n        Contact: Daniel Scholz\r\n\r\n        \r\n        \r\n        \r\n        Norway, Sweden, Finland\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact: Dr. Peter Oberschachtsiek \r\n\r\n        \r\n        \r\n        \r\n        Poland\r\n        Spectro-Lab \r\n        ul. Warszawska 100/102 \r\n        05-092 Lomianki \r\n        Telefon: +48 22 675 25 67 \r\n        Fax: +48 22 811 98 18 \r\n        artem.khvorostov@spectro-lab.pl \r\n          Internet: www.spectro-lab.com \r\n        Contact: Artem Khvorostov\r\n\r\n        \r\n        \r\n        \r\n        Portugal\r\n        Norleq - Equipamentos e Servicos \r\n        Rua Formosa 49 \r\n        3-sala 6 \r\n        4000-250 Porto \r\n        Telefon: +351-222081483 \r\n        Telefax: +351-222081768 \r\n        info@norleq.com \r\n        Internet: http://www.norleq.com \r\n        Contact: Adao Araujo\r\n\r\n      \r\n        \r\n        \r\n        Russia\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70704 Filderstadt \r\n        Phone: +49-711-770556-53 \r\n        Fax: +49-711-770556-99 \r\n        m.langer@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact:  Marina Langer (spricht Russisch)\r\n\r\n      \r\n        \r\n        \r\n        Switzerland, Liechtenstein, Austria\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact: Dr. Peter Oberschachtsiek\r\n\r\n        \r\n        \r\n        \r\n        Spain\r\n        NEURTEK S. A. \r\n        Pol. Ind Azitain Parc. 3a \r\n        ES-20600 Eibar \r\n        Telefon: +34-902-420082 / +34-943-8200-82 \r\n        Fax: +34-943-820157 \r\n        rafael@neurtek.es \r\n        Internet: http://www.neurtek.es \r\n        Contact: Rafael M. Atxurra\r\n\r\n        \r\n        \r\n        \r\n        Hungary\r\n        Laborexport Ltd. \r\n        Ibrahim str. 8 \r\n        HU-1113 Budapest \r\n        Telefon: +36-1-209-6424 \r\n        Fax: +36-1-209-6424 \r\n        sales@laborexport.hu \r\n        Internet: http://www.laborexport.hu \r\n        Contact: Dr. Tibor Müller\r\n\r\n      \r\n        \r\n        \r\n        Argentina\r\n        LAB CONTROL Ltda. \r\n        Rua José de Carvalho, 294 \r\n        Cep 04714-020 \r\n        Chácara Santo Antonio \r\n        Sao Paulo - Brasil \r\n        Telefon: +55-11-5181-1173 \r\n        vendas@labcontrol.com.br \r\n        Internet: http://www.labcontrol.com.br \r\n        Contact: Carlos-Alberto Maciel\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Brazil\r\n        LAB CONTROL Ltda. \r\n        Rua José de Carvalho, 294 \r\n        Cep 04714-020 \r\n        Chácara Santo Antonio \r\n        Sao Paulo - Brasil \r\n        Telefon: +55-11-5181-1173 \r\n        vendas@labcontrol.com.br \r\n        Internet: http://www.labcontrol.com.br \r\n        Contact: Carlos-Alberto Maciel \r\n\r\n  \r\n      \r\n        \r\n        \r\n        Chile\r\n        Melvyn Becerra &amp; Cia. Ltda. \r\n        Instrumentos Cientificos \r\n        Dr. Torres Boonen 834 \r\n        CL-Providencia Santiago \r\n        Telefon: +56-2-3414-233 / 2093-492 \r\n        Fax: +56-2-2059-596 \r\n        melbec1@vtr.net \r\n        Internet: http://www.melvynbecerra.cl \r\n        Contact: Melvyn Beccera \r\n\r\n  \r\n      \r\n        \r\n        \r\n        Ecuador\r\n        Jose M. Jalil Haas/ JJ&amp;H  Cia Ltda. \r\n        Pasaje Frey Joaquin Auz 181 (E07-08) \r\n        Quito \r\n        Telefon: +593-2-2244583, +593-2-2244583 \r\n        Fax: +593-2-2442380 \r\n        josehalil@jhjalil.com \r\n        Internet: www.jhjalil.com \r\n        Contact: Jose Jalil\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Canada\r\n        Folio Instruments Inc. \r\n        1645 St-Patrick \r\n        Montrela, Quebec, H3K3G9 \r\n        Telefon: +1-514-937-7931 \r\n        Fax: +1-514-937-4533 \r\n        rbarriere@folioinstruments.com \r\n        Internet: http://www.folioinstruments.com \r\n        Contact: Rhett Barriere\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Mexico\r\n        Conceptos e Instrumentos S.A. de C.V. \r\n        Creston 212-4 \r\n        Jardines del Pedregal \r\n        Mexico D. F. 01900 \r\n        Telefon: +52-55-55681671 oder 55681295 \r\n        conceptoseinstrumentos@gmail.com \r\n        Internet: www.conceptoseinstrumentos.com \r\n        Contact: José Antonio Blanco\r\n\r\n  \r\n      \r\n        \r\n        \r\n        USA\r\n        Future Digital Scientific Corp. \r\n        320 Endo Blvd. Suite 2B \r\n        Garden City \r\n        US-New York 11530 \r\n        Telefon: +1-516-349-0663 \r\n        Fax: +1-516-349-0642 \r\n        spencer@FDSC.com \r\n        Internet: http://www.FDSC.com \r\n        Contact: Dr. Spencer Wu\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        South Africa\r\n        Micron Scientific \r\n        175 Princes Avenue \r\n        Benoni, Gauteng, ZA \r\n        Telefon: +27-11-422-4303 \r\n        Fax: +27-11-845-2923 \r\n        trutera@ms.co.za \r\n        Internet: http://www.micronscientific.co.za \r\n        Contact: Albrecht Truter \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        China\r\n        EDIC\r\n        Beijing Eastern-Dataphy Instruments Co. Ltd. (EDIC) \r\n        Room 1010, Arstonal Tower \r\n        No.69 Zi Zhu Yuan Road Street \r\n        CN-100089 Beijing, Haidian District \r\n        Telefon: +86-10-68920269, 689202-57, -75, -76, -77  \r\n        Fax: +86-10-6892-0275 \r\n        robert_wang@edcc.com.cn \r\n        Internet: http://www.edcc.com.cn \r\n        Contact: Robert Wang \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        India\r\n        Goettfert DataPhysics Instruments India Pvt. Ltd. \r\n        P-84A, C.I.T. Road \r\n        Scheme-IV M \r\n        Calcutta-700 010 \r\n        Telefon: +919339867536 \r\n        basu@gdinstruments.co.in \r\n        Internet: http://www.gdinstruments.co.in \r\n        Contact: Bedanta Basu\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Iran\r\n        Mahar Fan Abzar Co. \r\n        Unit 16, No 40, Alley 13 \r\n        Miremad Street \r\n        Tehran, Iran \r\n        P.O. Box 15875-3781 \r\n        Telefon: ++98-21-88522420 and 88507022 \r\n        Fax: ++98-21-88522421 \r\n        piroozmarashi@maharfan.com \r\n        Contact: Dr. Pirooz Marashi\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Israel\r\n        Tetra Sense Scientific Tools \r\n        35 Hamoshav Rd \r\n        POB 887 \r\n        Carcur 37108 \r\n        Telefon: (+972)04 6272720 \r\n        Fax: (+972) 04 6272719 \r\n        moshe@tetrasense.co.il \r\n        Internet: http://www.tetrasense.co.il \r\n        Contact: Moshe Shmilovitz \r\n\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Japan\r\n        EKO Corporation \r\n        Sasazuka Center Bldg. 1-6 \r\n        Sasazuka 2-chome Sibuya-ku \r\n        JP-Tokyo 151 \r\n        Telefon: +81-3-5352-2911 \r\n        Fax: +81-3-5252-2917 \r\n        yazaki@eko.co.jp \r\n        Internet: http://www.eko.co.jp \r\n        Contact: Toshiaki Yazaki \r\n        \r\n        \r\n       \r\n        \r\n        \r\n        Malaysia\r\n        LMS\r\n        LMS Scientific Solutions Sdn Bhd. \r\n        D10-3-G, Block D10, \r\n        Pusat Perdagangan Dana 1, \r\n        Jalan PJU 1A/46, \r\n        47301 Petaling Jaya, \r\n        Selangor, Malaysia \r\n        Telefon: +603-7843 6533 \r\n        Fax: +603-7843 6522 \r\n         info@lmsscientific.com \r\n        Internet: www.lmsscientific.com \r\n        Contact: Jaez Lee\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Singapore\r\n        LMS Technologies Pte Ltd. \r\n        6 Tagore Drive \r\n        #04-05 Tagore Industrial Building \r\n        SG-787 623 Singapore \r\n        Telefon: +65-6-451-1123 \r\n        Fax: +65-6-452-1011 \r\n        anghg@lmstech.com.sg \r\n        Internet: www.lmsscientific.com \r\n        Contact: Francis Ang \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        South Korea\r\n        MCIK Co. - Material Characterization Instruments Korea \r\n        85-17 (2/2) Yeonhee-2 Dong \r\n        Seodaemun-Gu \r\n        KR-120-824 Seoul \r\n        Telefon: +82-2-3143-2740 / -1979 \r\n        Fax: +82-2-3143-2753 \r\n        srkim@mcik.co.kr \r\n        Internet: http://www.mcik.co.kr \r\n        Contact: Dr. Kim, Seungrok \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Taiwan\r\n        Toptical Scientific Corp. \r\n        7F-2, No 421, Sungshan Road \r\n        Taipei 11083, ROC \r\n        Telefon: + 886-2-2346-1510 \r\n        Fax: + 886-2-2346-1520 \r\n        toptical@ms17.hinet.net \r\n        Internet: www.toptical.com.tw \r\n        Contact: Reg Chen \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Thailand\r\n        LMS Instruments Co Ltd. \r\n        128/73 Phyathai Plaza Bldg.7th Floor \r\n        Phyathai Road, Thung-Phyathai \r\n        Radthavee \r\n        TH-Bangkok 10400 \r\n        Telefon: ++66-2-612-9091 \r\n        Fax: ++66-2-612-9093 \r\n        info@lmsinstr.com \r\n        Internet: www.lmsscientific.com \r\n        Contact: Sumate Tiratrakoonwichaya\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Australia\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-51 \r\n        Fax: +49-711-770556-99 \r\n        n.langer@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Contact: Nils Langer \r\n        \r\n        \r\n       \r\n      \n', '', 'page', '', '', 1429700971, 239, 15, 'Distributors Worldwide', 2, '', '');
INSERT INTO `tx_kesearch_index` (`uid`, `pid`, `tstamp`, `crdate`, `cruser_id`, `starttime`, `endtime`, `fe_group`, `targetpid`, `content`, `params`, `type`, `tags`, `abstract`, `sortdate`, `orig_uid`, `orig_pid`, `title`, `language`, `directory`, `hash`) VALUES
(222, 5, 1431355495, 1428385888, 0, 0, 0, '', '239', 'Vertretungen weltweit\n\n\n\r\n        \r\n        \r\n        \r\n        \r\n        \r\n          \r\n        \r\n        \r\n        \r\n          Bitte wählen Sie einen Kontinent\r\n          Europa\r\n            Amerika\r\n            Afrika\r\n            Asien\r\n            Australien\r\n        \r\n        \r\n        \r\n        \r\n          Bitte wählen Sie ein Land\r\n          Belgien, Luxemburg\r\n            Dänemark\r\n            Deutschland\r\n            Frankreich\r\n            GroÃbritannien\r\n            Irland\r\n            Italien\r\n            Niederlande\r\n            Norwegen, Schweden, Finnland\r\n            Polen\r\n            Portugal\r\n            Russland und GUS\r\n            Schweiz, Liechtenstein, Ãsterreich\r\n            Spanien\r\n            Ungarn\r\n        \r\n        \r\n        \r\n          Bitte wählen Sie ein Land\r\n          Argentinien\r\n            Brasilien\r\n            Chile\r\n            Ecuador\r\n            Kanada\r\n            Mexico\r\n            USA\r\n        \r\n        \r\n        \r\n          Bitte wählen Sie ein Land\r\n          Südafrika\r\n        \r\n        \r\n        \r\n          Bitte wählen Sie ein Land\r\n          China\r\n            Indien\r\n            Iran\r\n            Israel\r\n            Japan\r\n            Malaysia\r\n            Singapur, Indonesien\r\n            Süd Korea\r\n            Taiwan\r\n            Thailand\r\n        \r\n        \r\n        \r\n          Bitte wählen Sie ein Land\r\n          Australien\r\n        \r\n        \r\n        \r\n        \r\n        Belgien / Luxemburg\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-56 \r\n        Fax: +49-711-770556-99 \r\n        d.scholz@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Kontakt: Daniel Scholz \r\n        \r\n        \r\n        \r\n        Dänemark\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Kontakt: Dr. Peter Oberschachtsiek\r\n\r\n        \r\n        \r\n        \r\n        Deutschland\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-0 \r\n        Fax: +49-711-770556-99  \r\n\r\n        \r\n        Region: Süd-/Mittel-/Ostdeutschland (PLZ-Gebiete: 0, 1, 33-37, 5, 6, 7, 8, 9) \r\n        Kontakt: Dr. Peter Oberschachtsiek, Durchwahl -52 \r\n        p.oberschachtsiek@dataphysics.de  \r\n        \r\n        Region: Nord-/Westdeutschland (PLZ-Gebiete 2, 30-32, 38-39, 4) \r\n        Kontakt: Daniel Scholz, Durchwahl -56 \r\n        d.scholz@dataphysics.de\r\n\r\n        \r\n        \r\n        \r\n        Frankreich\r\n        Apollo Instruments \r\n        3 rue Clément Ader \r\n        ZAC Mercières \r\n        F-60200 Compiègne \r\n        Telephone: +33-3-44426619 \r\n        Fax: +33-3-44852472 \r\n        info@apolloinstruments.fr \r\n        Internet: http://www.apolloinstruments.fr \r\n        Kontakt: Dr. Romuald Dez  \r\n\r\n        \r\n        \r\n        \r\n        GroÃbritannien\r\n        Gerhardt UK \r\n        C. Gerhardt UK Ltd. \r\n        Unit 5, Avonbury Court \r\n        County road, Brackley \r\n        Northants, NN 13 7AX \r\n        Telefon: +44-1280-706772 \r\n        Fax: +44-1280-706088 \r\n          info@gerhardtonline.co.uk \r\n        Internet: http://www.cgerhardt.com \r\n        Kontakt: Paul Campbell \r\n\r\n        \r\n        \r\n        \r\n        Irland\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: + 49-711-770556-51 \r\n        Fax: + 49-711-770556-99 \r\n        n.langer@dataphysics.de \r\n        Internet:www.dataphysics.de \r\n        Kontakt: Nils Langer \r\n\r\n\r\n        \r\n        \r\n        \r\n        Italien\r\n        NG Labtec Srl. \r\n        Via Carlo Serassi, 21 \r\n        IT-24100 Bergamo \r\n        Telefon: +39-035-4175268 \r\n        Fax:  +39-035-4534445 \r\n        n.guerra@nglabtec.com \r\n        Internet: http://www.nglabtec.com \r\n        Kontakt: Norberto Guerra  \r\n\r\n        \r\n        \r\n        \r\n        Niederlande\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-56 \r\n        Fax: +49-711-770556-99 \r\n        d.scholz@dataphysics.de \r\n        Web: www.dataphysics.de \r\n        Kontakt: Daniel Scholz\r\n\r\n        \r\n        \r\n        \r\n        Norwegen, Schweden, Finnland\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Kontakt: Dr. Peter Oberschachtsiek \r\n\r\n        \r\n        \r\n        \r\n        Polen\r\n        Spectro-Lab \r\n        ul. Warszawska 100/102 \r\n        05-092 Lomianki \r\n        Telefon: +48 22 675 25 67 \r\n        Fax: +48 22 811 98 18 \r\n        artem.khvorostov@spectro-lab.pl \r\n          Internet: www.spectro-lab.com \r\n        Kontakt: Artem Khvorostov\r\n\r\n        \r\n        \r\n        \r\n        Portugal\r\n        Norleq - Equipamentos e Servicos \r\n        Rua Formosa 49 \r\n        3-sala 6 \r\n        4000-250 Porto \r\n        Telefon: +351-222081483 \r\n        Telefax: +351-222081768 \r\n        info@norleq.com \r\n        Internet: http://www.norleq.com \r\n        Kontakt: Adao Araujo\r\n\r\n      \r\n        \r\n        \r\n        Russland\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70704 Filderstadt \r\n        Phone: +49-711-770556-53 \r\n        Fax: +49-711-770556-99 \r\n        m.langer@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Kontakt:  Marina Langer (spricht Russisch)\r\n\r\n      \r\n        \r\n        \r\n        Schweiz, Lichtenstein, Ãsterreich\r\n        DataPhysics Instruments GmbH \r\n        Raiffeisenstr. 34 \r\n        70794 Filderstadt \r\n        Telefon: +49-711-770556-52 \r\n        Fax: +49-711-770556-99 \r\n        p.oberschachtsiek@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Kontakt: Dr. Peter Oberschachtsiek\r\n\r\n        \r\n        \r\n        \r\n        Spanien\r\n        NEURTEK S. A. \r\n        Pol. Ind Azitain Parc. 3a \r\n        ES-20600 Eibar \r\n        Telefon: +34-902-420082 / +34-943-8200-82 \r\n        Fax: +34-943-820157 \r\n        rafael@neurtek.es \r\n        Internet: http://www.neurtek.es \r\n        Kontakt: Rafael M. Atxurra\r\n\r\n        \r\n        \r\n        \r\n        Ungarn\r\n        Laborexport Ltd. \r\n        Ibrahim str. 8 \r\n        HU-1113 Budapest \r\n        Telefon: +36-1-209-6424 \r\n        Fax: +36-1-209-6424 \r\n        sales@laborexport.hu \r\n        Internet: http://www.laborexport.hu \r\n        Kontakt: Dr. Tibor Müller\r\n\r\n      \r\n        \r\n        \r\n        Argentinien\r\n        LAB CONTROL Ltda. \r\n        Rua José de Carvalho, 294 \r\n        Cep 04714-020 \r\n        Chácara Santo Antonio \r\n        Sao Paulo - Brasil \r\n        Telefon: +55-11-5181-1173 \r\n        vendas@labcontrol.com.br \r\n        Internet: http://www.labcontrol.com.br \r\n        Kontakt: Carlos-Alberto Maciel\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Brasilien\r\n        LAB CONTROL Ltda. \r\n        Rua José de Carvalho, 294 \r\n        Cep 04714-020 \r\n        Chácara Santo Antonio \r\n        Sao Paulo - Brasil \r\n        Telefon: +55-11-5181-1173 \r\n        vendas@labcontrol.com.br \r\n        Internet: http://www.labcontrol.com.br \r\n        Kontakt: Carlos-Alberto Maciel \r\n\r\n  \r\n      \r\n        \r\n        \r\n        Chile\r\n        Melvyn Becerra &amp; Cia. Ltda. \r\n        Instrumentos Cientificos \r\n        Dr. Torres Boonen 834 \r\n        CL-Providencia Santiago \r\n        Telefon: +56-2-3414-233 / 2093-492 \r\n        Fax: +56-2-2059-596 \r\n        melbec1@vtr.net \r\n        Internet: http://www.melvynbecerra.cl \r\n        Kontakt: Melvyn Beccera \r\n\r\n  \r\n      \r\n        \r\n        \r\n        Ecuador\r\n        Jose M. Jalil Haas/ JJ&amp;H  Cia Ltda. \r\n        Pasaje Frey Joaquin Auz 181 (E07-08) \r\n        Quito \r\n        Telefon: +593-2-2244583, +593-2-2244583 \r\n        Fax: +593-2-2442380 \r\n        josehalil@jhjalil.com \r\n        Internet: www.jhjalil.com \r\n        Kontakt: Jose Jalil\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Kanada\r\n        Folio Instruments Inc. \r\n        1645 St-Patrick \r\n        Montrela, Quebec, H3K3G9 \r\n        Telefon: +1-514-937-7931 \r\n        Fax: +1-514-937-4533 \r\n        rbarriere@folioinstruments.com \r\n        Internet: http://www.folioinstruments.com \r\n        Kontakt: Rhett Barriere\r\n\r\n  \r\n      \r\n        \r\n        \r\n        Mexico\r\n        Conceptos e Instrumentos S.A. de C.V. \r\n        Creston 212-4 \r\n        Jardines del Pedregal \r\n        Mexico D. F. 01900 \r\n        Telefon: +52-55-55681671 oder 55681295 \r\n        conceptoseinstrumentos@gmail.com \r\n        Internet: www.conceptoseinstrumentos.com \r\n        Kontakt: José Antonio Blanco\r\n\r\n  \r\n      \r\n        \r\n        \r\n        USA\r\n        Future Digital Scientific Corp. \r\n        320 Endo Blvd. Suite 2B \r\n        Garden City \r\n        US-New York 11530 \r\n        Telefon: +1-516-349-0663 \r\n        Fax: +1-516-349-0642 \r\n        spencer@FDSC.com \r\n        Internet: http://www.FDSC.com \r\n        Kontakt: Dr. Spencer Wu\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Südafrika\r\n        Micron Scientific \r\n        175 Princes Avenue \r\n        Benoni, Gauteng, ZA \r\n        Telefon: +27-11-422-4303 \r\n        Fax: +27-11-845-2923 \r\n        trutera@ms.co.za \r\n        Internet: http://www.micronscientific.co.za \r\n        Kontakt: Albrecht Truter \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        China\r\n        EDIC\r\n        Beijing Eastern-Dataphy Instruments Co. Ltd. (EDIC) \r\n        Room 1010, Arstonal Tower \r\n        No.69 Zi Zhu Yuan Road Street \r\n        CN-100089 Beijing, Haidian District \r\n        Telefon: +86-10-68920269, 689202-57, -75, -76, -77  \r\n        Fax: +86-10-6892-0275 \r\n        robert_wang@edcc.com.cn \r\n        Internet: http://www.edcc.com.cn \r\n        Kontakt: Robert Wang \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Indien\r\n        Goettfert DataPhysics Instruments India Pvt. Ltd. \r\n        P-84A, C.I.T. Road \r\n        Scheme-IV M \r\n        Calcutta-700 010 \r\n        Telefon: +919339867536 \r\n        basu@gdinstruments.co.in \r\n        Internet: http://www.gdinstruments.co.in \r\n        Kontakt: Bedanta Basu\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Iran\r\n        Mahar Fan Abzar Co. \r\n        Unit 16, No 40, Alley 13 \r\n        Miremad Street \r\n        Tehran, Iran \r\n        P.O. Box 15875-3781 \r\n        Telefon: ++98-21-88522420 and 88507022 \r\n        Fax: ++98-21-88522421 \r\n        piroozmarashi@maharfan.com \r\n        Kontakt: Dr. Pirooz Marashi\r\n        \r\n        \r\n        \r\n        \r\n        \r\n        Israel\r\n        Tetra Sense Scientific Tools \r\n        35 Hamoshav Rd \r\n        POB 887 \r\n        Carcur 37108 \r\n        Telefon: (+972)04 6272720 \r\n        Fax: (+972) 04 6272719 \r\n        moshe@tetrasense.co.il \r\n        Internet: http://www.tetrasense.co.il \r\n        Kontakt: Moshe Shmilovitz \r\n\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Japan\r\n        EKO Corporation \r\n        Sasazuka Center Bldg. 1-6 \r\n        Sasazuka 2-chome Sibuya-ku \r\n        JP-Tokyo 151 \r\n        Telefon: +81-3-5352-2911 \r\n        Fax: +81-3-5252-2917 \r\n        yazaki@eko.co.jp \r\n        Internet: http://www.eko.co.jp \r\n        Kontakt: Toshiaki Yazaki \r\n        \r\n        \r\n       \r\n        \r\n        \r\n        Malaysia\r\n        LMS\r\n        LMS Scientific Solutions Sdn Bhd. \r\n        D10-3-G, Block D10, \r\n        Pusat Perdagangan Dana 1, \r\n        Jalan PJU 1A/46, \r\n        47301 Petaling Jaya, \r\n        Selangor, Malaysia \r\n        Telefon: +603-7843 6533 \r\n        Fax: +603-7843 6522 \r\n         info@lmsscientific.com \r\n        Internet: www.lmsscientific.com \r\n        Kontakt: Jaez Lee\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Singapur\r\n        LMS Technologies Pte Ltd. \r\n        6 Tagore Drive \r\n        #04-05 Tagore Industrial Building \r\n        SG-787 623 Singapore \r\n        Telefon: +65-6-451-1123 \r\n        Fax: +65-6-452-1011 \r\n        anghg@lmstech.com.sg \r\n        Internet: www.lmsscientific.com \r\n        Kontakt: Francis Ang \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Südkorea\r\n        MCIK Co. - Material Characterization Instruments Korea \r\n        85-17 (2/2) Yeonhee-2 Dong \r\n        Seodaemun-Gu \r\n        KR-120-824 Seoul \r\n        Telefon: +82-2-3143-2740 / -1979 \r\n        Fax: +82-2-3143-2753 \r\n        srkim@mcik.co.kr \r\n        Internet: http://www.mcik.co.kr \r\n        Kontakt: Dr. Kim, Seungrok \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Taiwan\r\n        Toptical Scientific Corp. \r\n        7F-2, No 421, Sungshan Road \r\n        Taipei 11083, ROC \r\n        Telefon: + 886-2-2346-1510 \r\n        Fax: + 886-2-2346-1520 \r\n        toptical@ms17.hinet.net \r\n        Internet: www.toptical.com.tw \r\n        Kontakt: Reg Chen \r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Thailand\r\n        LMS Instruments Co Ltd. \r\n        128/73 Phyathai Plaza Bldg.7th Floor \r\n        Phyathai Road, Thung-Phyathai \r\n        Radthavee \r\n        TH-Bangkok 10400 \r\n        Telefon: ++66-2-612-9091 \r\n        Fax: ++66-2-612-9093 \r\n        info@lmsinstr.com \r\n        Internet: www.lmsscientific.com \r\n        Kontakt: Sumate Tiratrakoonwichaya\r\n        \r\n        \r\n        \r\n         \r\n        \r\n        Australien\r\n        DataPhysics Instruments GmbH \r\n        RaiffeisenstraÃe 34 \r\n        D-70794 Filderstadt \r\n        Telefon: +49-711-770556-51 \r\n        Fax: +49-711-770556-99 \r\n        n.langer@dataphysics.de \r\n        Internet: www.dataphysics.de \r\n        Kontakt: Nils Langer \r\n        \r\n        \r\n       \r\n      \n\nHausanschrift\r\nDataPhysics Instruments GmbH Raiffeisenstrasse 34 D 70794 Filderstadt Telefon: ++49 (0)711 770556-0 Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de  sales@dataphysics.de\n', '', 'page', '', '', 1429700971, 239, 15, 'Vertretungen weltweit', 0, '', ''),
(223, 5, 1431355495, 1428385888, 0, 0, 0, '', '241', 'Your way by car to DataPhysics\nDriving from Munich or Karlsruhe/Heilbronn/Singen\r\n A8-Exit Stuttgart-Degerloch&nbsp;(No. 52b); Direction: &quot;Tübingen&quot; Follow the B27 directionTübingen until you reach Exit: Filderstadt-Ost-Bonlanden. On the first traffic light straight onwards First junction turn to the right After 100 m on the left hand side you will find three office buildings No. 34 is the sternmost building Parking lot directly in front of building No. 34 DataPhysics is located on the ground level (Level 0)\nAdress\nDataPhysics Instruments GmbH\r\nRaiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0&nbsp; Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de E-Mail:&nbsp; sales@dataphysics.de\n (copy 6)\n Directions\n (copy 3)\n Représentation mondiale\n', '', 'page', '', '', 1429624821, 241, 15, 'Directions', 3, '', ''),
(224, 5, 1431355495, 1428385888, 0, 0, 0, '', '241', 'Your way by car to DataPhysics\nDriving from Munich or Karlsruhe/Heilbronn/Singen\r\n A8-Exit Stuttgart-Degerloch&nbsp;(No. 52b); Direction: &quot;Tübingen&quot; Follow the B27 directionTübingen until you reach Exit: Filderstadt-Ost-Bonlanden. On the first traffic light straight onwards First junction turn to the right After 100 m on the left hand side you will find three office buildings No. 34 is the sternmost building Parking lot directly in front of building No. 34 DataPhysics is located on the ground level (Level 0)\nAdress\nDataPhysics Instruments GmbH\r\nRaiffeisenstrasse 34&nbsp; D 70794 Filderstadt Telefon: ++49 (0)711 770556-0&nbsp; Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de E-Mail:&nbsp; sales@dataphysics.de\n (copy 5)\n Directions\n (copy 2)\n Distributors Worldwide\n', '', 'page', '', '', 1429624821, 241, 15, 'Directions', 2, '', ''),
(225, 5, 1431355495, 1428385888, 0, 0, 0, '', '241', '\nHausanschrift\r\nDataPhysics Instruments GmbH Raiffeisenstrasse 34 D 70794 Filderstadt Telefon: ++49 (0)711 770556-0 Telefax: ++49 (0)711 770556-99 Internet:&nbsp; www.dataphysics.de  sales@dataphysics.de\n (copy 4)\n Anfahrt\n (copy 1)\n Vertretungen weltweit\nSo finden Sie uns\nVon München und Karlsruhe/Heilbronn/Singen kommend\r\n A8-AnschluÃstelle Stuttgart-Degerloch (Nr. 52b); Richtung: &quot;Tübingen&quot; Der B27 Richtung Tübingen folgen bis Ausfahrt Filderstadt-Ost-Bonlanden. Danach an erster Ampel geradeaus Erste StraÃe rechts Nach 150 m links 3 hintereinanderstehende Bürogebäude Nr. 34 ist das hintere Gebäude Parkplatz vor dem Gebäude Nr. 34 DataPhysics befindet sich im ErdgeschoÃ (Ebene 0)\n', '', 'page', '', '', 1429624821, 241, 15, 'Anfahrt', 0, '', ''),
(226, 5, 1431355495, 1428385888, 0, 0, 0, '', '242', 'OCA 25\nThe&nbsp;video-based optical contact angle measuring system, OCA 25, is our most versatile instrument for the measurement and&nbsp;analysis of contact angle and drop shape. The OCA 25 is available in the variants: OCA 25L, and OCA 25LHT.&nbsp;\r\nFeatures of the basic instrument configuration: \r\n Sample table with manual adjustment in three axes, via high-precision mechanics, for the precise and accurate positioning of the sample. High performance 6x parfocal zoom lens with integrated continuous fine focus, and adjustable observation and camera tilt angle. Video measuring system with high-resolution USB camera (max. 123 images/s sample rate). Lighting of the optical path with software controlled intensity adjustment, without hysteresis. Max. sample dimensions (L x W x H):&nbsp; OCA 25: 220 x ¤&nbsp;x 70 mm. OCA 25L: 330 x ¤&nbsp;x 70 mm. OCA 25LHT:330 x ¤&nbsp;x 70 mm, HTFQ 1200 or HTFC 1800.\r\n Optional OCA accessories: Some of the most important components and accessories are listed below:&nbsp;\r\n High-speed option for USB camera (max. 215 or 311 images/s sample rate), High-speed video system UpHSV1220 (max. 1220 images/s sample rate), or UpHSV3000 (sample rate 3000 images/s). Software controlled electronic multiple dosing systems, E-MDr, for the precise automatic positioning of up to six dosing needles. Direct dosing system, SD-DM, DD-DM, SD-DE or DDE/x, Up to six electronic syringe units, ESr, with precise adjustable dosing volume (min. 50 nl) and dosing speed (0.06 µl/sâ¦26.4 µl/s). Electronic tilting base unit, TBU 95, for a maximum tilt angle of 90° and manual tilting base attachment, TBA 60M, for a maximum tilt angle of 70°. Electronic turn table with vacuum fixation, ETTr/VAC, with wide range of top plates. Temperature and environmental control systems (-30â¦1800 °C und 1 mPaâ¦ 5·106&nbsp;Pa). Wide range of sample holding and positioning units. Inc holders for foils or papers, FSH 30 and FSC 80/150, for single fibres, FHO 40plus, or the suction plate, SP 100, for holding thin flexible samples flat on the stage, with an adjustable suction area. Oscillating drop generators, ODG 20 and ODG 20P, for the measurement of surface elasticities and for relaxational studies at phase boundaries. Electro-wetting platform, EWP 100, for the analysis of sessile and pendant drops in an electrical field. Top view video system, TV-VS, for the qualitative view and image capture of the of the drop position (USB camera with 52 images/s sample rate, 6x parfocal zoom lens and adjustable observation angle). The available software modules for the OCA 25 are: SCA 20 â contact angle. SCA 21 â surface free energy. SCA 22 â pendant drop. SCA 23 â lamella contour. SCA 26 â oscillation / relaxation. &nbsp;\r\n Data Sheet OCA 25\n', '', 'page', '', '', 1429619078, 242, 185, 'OCA 25', 3, '', ''),
(227, 5, 1431355495, 1428385888, 0, 0, 0, '', '242', 'OCA 25\nThe&nbsp;video-based optical contact angle measuring system, OCA 25, is our most versatile instrument for the measurement and&nbsp;analysis of contact angle and drop shape. The OCA 25 is available in the variants: OCA 25L, and OCA 25LHT.&nbsp;\r\nFeatures of the basic instrument configuration: \r\n Sample table with manual adjustment in three axes, via high-precision mechanics, for the precise and accurate positioning of the sample. High performance 6x parfocal zoom lens with integrated continuous fine focus, and adjustable observation and camera tilt angle. Video measuring system with high-resolution USB camera (max. 123 images/s sample rate). Lighting of the optical path with software controlled intensity adjustment, without hysteresis. Max. sample dimensions (L x W x H):&nbsp; OCA 25: 220 x ¤&nbsp;x 70 mm. OCA 25L: 330 x ¤&nbsp;x 70 mm. OCA 25LHT:330 x ¤&nbsp;x 70 mm, HTFQ 1200 or HTFC 1800.\r\n Optional OCA accessories: Some of the most important components and accessories are listed below:&nbsp;\r\n High-speed option for USB camera (max. 215 or 311 images/s sample rate), High-speed video system UpHSV1220 (max. 1220 images/s sample rate), or UpHSV3000 (sample rate 3000 images/s). Software controlled electronic multiple dosing systems, E-MDr, for the precise automatic positioning of up to six dosing needles. Direct dosing system, SD-DM, DD-DM, SD-DE or DDE/x, Up to six electronic syringe units, ESr, with precise adjustable dosing volume (min. 50 nl) and dosing speed (0.06 µl/sâ¦26.4 µl/s). Electronic tilting base unit, TBU 95, for a maximum tilt angle of 90° and manual tilting base attachment, TBA 60M, for a maximum tilt angle of 70°. Electronic turn table with vacuum fixation, ETTr/VAC, with wide range of top plates. Temperature and environmental control systems (-30â¦1800 °C und 1 mPaâ¦ 5·106&nbsp;Pa). Wide range of sample holding and positioning units. Inc holders for foils or papers, FSH 30 and FSC 80/150, for single fibres, FHO 40plus, or the suction plate, SP 100, for holding thin flexible samples flat on the stage, with an adjustable suction area. Oscillating drop generators, ODG 20 and ODG 20P, for the measurement of surface elasticities and for relaxational studies at phase boundaries. Electro-wetting platform, EWP 100, for the analysis of sessile and pendant drops in an electrical field. Top view video system, TV-VS, for the qualitative view and image capture of the of the drop position (USB camera with 52 images/s sample rate, 6x parfocal zoom lens and adjustable observation angle). The available software modules for the OCA 25 are: SCA 20 â contact angle. SCA 21 â surface free energy. SCA 22 â pendant drop. SCA 23 â lamella contour. SCA 26 â oscillation / relaxation. &nbsp;\r\n Data Sheet OCA 25\n', '', 'page', '', '', 1429619078, 242, 185, 'OCA 25', 2, '', ''),
(228, 5, 1431355495, 1428385888, 0, 0, 0, '', '243', 'OCA 50AF\nThe&nbsp;automatic video-based optical contact angle&nbsp;and&nbsp;drop contour analysis&nbsp;system OCA 50 AF is the instrument of choice when your application demands for the fully automated, timesaving, set-up and analysis of the&nbsp;homogeneity/wettability of solid surfaces&nbsp;and/or the&nbsp;surface free energy&nbsp;(inc. polar/disperse components) of solids. The automated sequence of tests and the video-based optical image processing facilitates the analysis of simple and complex sample structures at the âpush of a buttonâ.\r\n\r\nFor bigger samples (e.g. 12â Wafers) the OCA 50 AF L with long X-axis and Y-axis is available.\r\nFeatures of the base instrument:\r\n Sample table, software controlled, adjustment in all three axis (X-, Y- and Z-axis). High performance 6x parfocal zoom lens with automated focus and camera tilt angle. Video measuring system with high-resolution USB camera (max. 1220 images/s sample rate). Lighting of the optical path with software controlled intensity adjustment, without hysteresis. Max. sample dimensions (L x W x H):&nbsp; OCA 50 AF: 220 x ? x 70 mm.\r\n&nbsp; Optional OCA accessories: Some of the most important components and accessories are listed below:\r\n High-speed video system upHSV 3000 (sample rate 3000 images/s). Software controlled electronic multiple dosing systems, E-MDr, for the precise automatic positioning of up to six dosing needles. Direct dosing system, SD-DM, DD-DM, SD-DE or DDE/x. Up to six electronic syringe units, ESr, with precise adjustable dosing speed (0.4 nl/sâ¦174 µl/s). Electronic tilting base unit, TBU 95, for a maximum tilt angle of 95° and manual tilting base attachment, TBA 60M, for a maximum tilt angle of 70°. Electronic turn table with vacuum fixation, ETTr/VAC, with a wide range of top plates. Temperature and environmental control systems (-30â¦700 °C). Wide range of sample holding and positioning units. Inc. holders for foils or papers, FSH 30 and FSC 80/150, for single fibres, FHO 40plus, or the suction plate, SP 100, for holding thin flexible samples flat on the stage, with an adjustable suction area. Oscillating drop generators, ODG 20 for the measurement of surface elasticities and for relaxational studies at phase boundaries. Electro-wetting platform, EWP 100, for the analysis of sessile and pendant drops in an electrical field. Top view video system, TV-VS, for the viewing and image capture of the drop position (USB camera with 52 images/s sample rate, 6x parfocal zoom lens and adjustable observation angle). The available software modules for the OCA 50 AF are: SCA 20 â contact angle. SCA 21 â surface free energy. SCA 22 â pendant drop. SCA 23 â lamella contour. SCA 26 â oscillation / relaxation.\n', '', 'page', '', '', 1429619853, 243, 185, 'OCA 50AF', 3, '', ''),
(229, 5, 1431355495, 1428385888, 0, 0, 0, '', '243', 'OCA 50 AF\nThe&nbsp;automatic video-based optical contact angle&nbsp;and&nbsp;drop contour analysis&nbsp;system OCA 50 AF is the instrument of choice when your application demands for the fully automated, timesaving, set-up and analysis of the&nbsp;homogeneity/wettability of solid surfaces&nbsp;and/or the&nbsp;surface free energy&nbsp;(inc. polar/disperse components) of solids. The automated sequence of tests and the video-based optical image processing facilitates the analysis of simple and complex sample structures at the âpush of a buttonâ.\r\n\r\nFor bigger samples (e.g. 12â Wafers) the OCA 50 AF L with long X-axis and Y-axis is available.\r\nFeatures of the base instrument:\r\n Sample table, software controlled, adjustment in all three axis (X-, Y- and Z-axis). High performance 6x parfocal zoom lens with automated focus and camera tilt angle. Video measuring system with high-resolution USB camera (max. 1220 images/s sample rate). Lighting of the optical path with software controlled intensity adjustment, without hysteresis. Max. sample dimensions (L x W x H):&nbsp; OCA 50 AF: 220 x ¤&nbsp;x 70 mm.\r\n&nbsp; Optional OCA accessories: Some of the most important components and accessories are listed below:\r\n High-speed video system upHSV 3000 (sample rate 3000 images/s). Software controlled electronic multiple dosing systems, E-MDr, for the precise automatic positioning of up to six dosing needles. Direct dosing system, SD-DM, DD-DM, SD-DE or DDE/x. Up to six electronic syringe units, ESr, with precise adjustable dosing speed (0.4 nl/sâ¦174 µl/s). Electronic tilting base unit, TBU 95, for a maximum tilt angle of 95° and manual tilting base attachment, TBA 60M, for a maximum tilt angle of 70°. Electronic turn table with vacuum fixation, ETTr/VAC, with a wide range of top plates. Temperature and environmental control systems (-30â¦700 °C). Wide range of sample holding and positioning units. Inc. holders for foils or papers, FSH 30 and FSC 80/150, for single fibres, FHO 40plus, or the suction plate, SP 100, for holding thin flexible samples flat on the stage, with an adjustable suction area. Oscillating drop generators, ODG 20 for the measurement of surface elasticities and for relaxational studies at phase boundaries. Electro-wetting platform, EWP 100, for the analysis of sessile and pendant drops in an electrical field. Top view video system, TV-VS, for the viewing and image capture of the drop position (USB camera with 52 images/s sample rate, 6x parfocal zoom lens and adjustable observation angle). The available software modules for the OCA 50 AF are: SCA 20 â contact angle. SCA 21 â surface free energy. SCA 22 â pendant drop. SCA 23 â lamella contour. SCA 26 â oscillation / relaxation.\r\n Data Sheet OCA 50 AF\n', '', 'page', '', '', 1429619853, 243, 185, 'OCA 50AF', 2, '', ''),
(242, 5, 1431355495, 1431355495, 0, 0, 0, '', '261', 'Test\nLorem Ipsum\n', '', 'page', '', '', 1431352711, 261, 185, 'OCA Test', 0, '', ''),
(231, 5, 1431355495, 1428385888, 0, 0, 0, '', '244', 'OCA 50Micro\nThe fully&nbsp;automatic contact angle measuring system for microstructures, OCA 50 Micro, is the optical instrument with the highest degree of automation. It provides for the accurate, precise and reproducible measurement of contact angles of minute liquid drops and for the evaluation of the wetting properties of&nbsp;microscopic&nbsp;andmacroscopic&nbsp;structures. State-of-the-art optics, precise mechanics, fast electronic controllers as well as high-resolution video measurement technology ensures the appropriate drop placement and image capture/analysis, in microscopic dimensions, in any situation.\r\nFeatures of the basic instrument configuration:\r\n Measuring lens with 55-fold (optional 137.5-fold)&nbsp; zoom and software controlled, motorized focus and adjustment of the observation angle. Video measuring system with high resolution USB camera (max. 1220 images/s sample rate). Measuring table with, software controlled, motorized X-, Y- and Z-axes for the highly precise positioning of solid samples to ensure droplet placement/pick-up in the required area. High efficiency lighting of the optical path with software control of the light intensity, without hysteresis, and special heat eliminating routine.&nbsp;\r\n\r\nOptional OCA accessories:\r\nThe applicability of the OCA 50 Micro can be enhanced with the majority of the components within the OCA accessory range (with the notable exception of the high temperature furnaces).\r\n\r\nSome of the most important components and accessories are listed below:\r\n High-speed video system upHSV 3000 (max. 3000 images/s sample rate). Software controlled electronic multiple dosing systems, E-MD, for the precise automatic positioning of up to six dosing needles. Direct dosing systems, SD-DM, DD-DM, SD-DE and DDE/x. Up to six electronic syringe units, ESr, with precise adjustable dosing speed (0.4 nl/sâ¦174&nbsp;µl/s). Electronic micro dosing systems for the precise dosing of the test liquids in the nanoliter and picoliter range. Electronic tilting base unit, TBU 95, for a maximum tilt angle of 95° and manual tilting base attachment, TBA 60M, for a maximum tilt angle of 70°. Electronic turn table with vacuum fixation, ETT/VAC, with top plates up to 200 mm diameter. Temperature and environmental control systems (-30â¦700 °C). Wide range of sample holding and positioning units. Inc. holders for foils or papers, FSH 30 and FSC 80/150, for single fibres, FHO 40plus, or the suction plate, SP 100, for holding thin flexible samples flat on the stage, with an adjustable suction area. Oscillating drop generators, ODG 20 for the measurement of surface elasticities and for relaxational studies at phase boundaries. Electro-wetting platform, EWP 100, for the analysis of sessile and pendant drops in an electrical field. Top view video system TV-VS for the viewing and image capture of the drop (USB camera with 52 images/s sample rate, 6x parfocal zoom lens and adjustable observation angle). The available software modules for the OCA 50 Micro are: SCA 20 â contact angle. SCA 21 ­â surface free energy. SCA 22 â pendant drop. SCA 23 â lamella contour. SCA 24 â drop on fibre. SCA 26 â oscillation / relaxation.\r\n Data Sheet OCA 50 Micro\n', '', 'page', '', '', 1429619994, 244, 185, 'OCA 50Micro', 2, '', ''),
(239, 5, 1431355495, 1431328019, 0, 0, 0, '', '259', '\nPresse\r\nWas gibt es über DataPhysics zu berichten und was schreibt die Fachpresse über uns? Aktuelle Pressemitteilungen sowie ausgewählte, bereits publizierte Artikel können Sie hier nachlesen und bei Bedarf herunterladen.\r\n\r\n05.11.2014\r\n\r\nZuverlässiger Alleskönner der Grenzflächenchemie! - Das optimierte Tensiometer aus der DCAT-Reihe mit neuer Software und Messmethoden von DataPhysics Instruments\r\nDie dynamischen Kontaktwinkelmessgeräte und Tensiometer der DCAT-Baureihe von DataPhysics Instruments präsentieren sich ab sofort mit einer neu entwickelten, intuitiv zu bedienenden Software sowie einer optimierten Auswahl an Messmöglichkeiten. Die gewichtsbasierten Messungen der Oberflächen- und Grenzflächenspannung werden nach der Du Noüy-Ring- oder der Wilhelmy Platten-Methode vollautomatisch gesteuert und ausgewertet. Für die unterschiedlichen dynamischen, statischen, zeit- und temperaturabhängigen Analyseoptionen steht Ihnen nun auch ein Langmuir-Blodgett-Trog als Prüfkörper zur Verfügung. Damit lassen&nbsp;sich&nbsp;Oberflächenbeschichtungen ebenso wie molekulare Selbstorganisationsprozesse von chemischen Substanzen untersuchen. Innovatives Design und die hochwertigen Gerätekomponenten garantieren Ihnen eine optimale Unterstützung in der Forschung, Entwicklung und Qualitätskontrolle.&nbsp;\r\n\r\nOptische Tropfenkonturanalyse â hochpräzise und innovativ\r\n\r\n\r\nDie neue Generation Kontaktwinkelmessgeräte von DataPhysics Instruments, präsentiert auf CHEMIE.de!\r\n Chemie.de\nVeranstaltungen & Seminare\nDie Bedeutung der Grenzflächenchemie wächst unaufhaltsam und damit auch  das Anwendungsspektrum der DataPhysics-Produkte. Mit unseren  Veranstaltungen und Seminaren halten wir Sie auf dem neuesten Stand.\r\nIndividuelle, auf Ihre Bedürfnisse und Anwendungsproblematiken  abgestimmte Schulungen führen wir in unserem Anwendungszentrum in  Filderstadt oder bei Ihnen vor Ort durch. Bitte setzen Sie sich mit uns  in Verbindung, um einen passenden Termin mit Ihnen abzustimmem.\nSeminar mit Theorie & Praxis - 16.06. - 17.06.2015\nModerne Messmethoden für die Grenzflächenchemie\r\nUnser Seminar richtet sich an alle Interessierten, die sich einen Ãberblick über den aktuellen Stand der Gerätetechnik in der Grenzflächenanalyse verschaffen möchten. Neben der Vertiefung von theoretischem Grundwissen werden auch anwendungsbezogene Probleme praktisch behandelt. Das Seminar findet in den neuen Schulungsräumen der DataPhysics Instruments GmbH in Filderstadt bei Stuttgart statt.\r\n\r\n Seminarflyer, Juni 2015\n', '', 'page', '', '', 1429694733, 259, 14, 'Seminare & Presse', 0, '', ''),
(240, 5, 1431355495, 1431328019, 0, 0, 0, '', '259', 'Communiqués de presse\n31.10.2014\r\nSystème d''analysant de contour de goutte optique â extrêmement exact et innovateur\r\nThe new optical contact angle measuring devices from DataPhysics Instruments    The measurement of contact angles on solid bodies done by the sessile drop method and the surface tension of liquids done by the pendant drop method are standard procedures in the industrial quality management as well as in research and development, nowadays. For instance, functional surfaces running through cleaning processes or adherence properties of glues can be analyzed with the new optical contact angle measurement devices from DataPhysics Instruments more precisely, time-efficient and user-friendly.  \r\n Special features:\r\n Touchpad for controlling all electrical components, e.g. sample stage, tilting base unit, dosing of liquids User-friendly and easy comprehensible software, arranged in modules Motorized sample stage with x-, y- and z-axes for an exact, fully automatic sample positioning as well as drop intake. Electric multiple direct dosing system without tubes so that cleaning is not necessary anymore High-speed camera with a maximum of 3000 pictures per second and full-automatic focus to investigate wetting processes in a millisecond range Different temperature and environment chambers (?30 to 700 °C; rel. humidity: 5 to 90%) Automatic test runs Electronic tilting base unit for the determination of the roll of behaviour of liquid drops (max. tilt angle 95°)\r\n   Application fields are laboratory analysis of the wetting behaviour of/on e.g. coatings, colours, fibres, glues, varnishes, micro-mechanical construction components, surface structures, paper and cellulose, inks and much more.  For further information please visit our product website, download our product brochure or request information now!\nAteliers & Séminaires\nNous offrons l''entraînement spécifique d''application et les séminaires  dans votre environnement de travail familier ou à notre propre facilité  en Allemagne. Vous-même et votre équipe sera formée tant dans les  aspects pratiques que dans théoriques de la science pertinente, en  utilisant l''instrumentation individuelle et les méthodes.    L''entraînement spécifique d''application, sur votre instrument/à votre  endroit ou notre laboratoire d''application garant le soutien le plus  efficace sur tous les aspects pertinents de vos applications  spécifiques.    Nous fournissons aussi le soutien téléphonique et e-mail sur le  développement et l''application de procédures de mesure et le soutien le  plus rapide possible en faveur de tous, l''application rattachée, les  questions de logiciel.    Donc n''hésitez pas à nous  contacter.\nA NEUF!! Séminare d''application allemand - 03.03.-04.03.2015\nNotre prochain séminaire d''application allemand a lieu du 3.3.-4.3.2015  &nbsp;dans nos nouveaux espaces&nbsp;dans Filderstadt.&nbsp;Nous voudrions vous inviter  cordialement. Dans le séminaire, ils apprendront la théorie et  pratique.&nbsp;Vous recevez les autres informations  ici.\n', '', 'page', '', '', 1429694733, 259, 14, 'Séminaires & Presse', 3, '', ''),
(241, 5, 1431355495, 1431328019, 0, 0, 0, '', '259', 'Training, Seminars, Workshops\nWe offer, application specific, training and seminars either within your  familiar work environment or at our own facility in Germany. Yourself  and your team will be trained in both practical and theoretical aspects  of the relevant science, using the relevant instrumentation and  developed methods.    Application specific training, on your instrument/at your location, or  our application laboratory guarantees the most effective support on all  relevant aspects of your specific applications.    Additionally we provide telephone and e-mail support on the development  and application of measurement procedures and the quickest possible  support for all, application related, software questions.    Please get in  contact with us.\n', '', 'page', '', '', 1429694733, 259, 14, 'Seminars & Press', 2, '', '');

-- --------------------------------------------------------

--
-- Tabellenstruktur für Tabelle `tx_kesearch_indexerconfig`
--

CREATE TABLE IF NOT EXISTS `tx_kesearch_indexerconfig` (
  `uid` int(11) NOT NULL AUTO_INCREMENT,
  `pid` int(11) NOT NULL DEFAULT '0',
  `tstamp` int(11) NOT NULL DEFAULT '0',
  `crdate` int(11) NOT NULL DEFAULT '0',
  `cruser_id` int(11) NOT NULL DEFAULT '0',
  `deleted` tinyint(4) NOT NULL DEFAULT '0',
  `hidden` tinyint(4) NOT NULL DEFAULT '0',
  `title` tinytext,
  `storagepid` text,
  `targetpid` text,
  `startingpoints_recursive` text,
  `single_pages` text,
  `sysfolder` text,
  `index_content_with_restrictions` text,
  `index_passed_events` text,
  `type` varchar(90) NOT NULL DEFAULT '',
  `index_news_category_mode` tinyint(4) NOT NULL DEFAULT '0',
  `index_news_category_selection` text,
  `index_extnews_category_selection` text,
  `index_news_useHRDatesSingle` tinyint(4) NOT NULL DEFAULT '0',
  `index_news_useHRDatesSingleWithoutDay` tinyint(4) NOT NULL DEFAULT '0',
  `index_dam_categories` text,
  `index_dam_without_categories` tinyint(4) NOT NULL DEFAULT '0',
  `index_dam_categories_recursive` tinyint(3) NOT NULL DEFAULT '0',
  `index_use_page_tags` tinyint(3) NOT NULL DEFAULT '0',
  `index_use_page_tags_for_files` tinyint(3) NOT NULL DEFAULT '0',
  `directories` text,
  `fileext` tinytext,
  `commenttypes` tinytext,
  `filteroption` int(11) NOT NULL DEFAULT '0',
  `tvpath` varchar(255) NOT NULL DEFAULT '',
  `fal_storage` int(11) NOT NULL DEFAULT '0',
  PRIMARY KEY (`uid`),
  KEY `parent` (`pid`)
) ENGINE=MyISAM  DEFAULT CHARSET=latin1 AUTO_INCREMENT=2 ;

--
-- Daten für Tabelle `tx_kesearch_indexerconfig`
--

INSERT INTO `tx_kesearch_indexerconfig` (`uid`, `pid`, `tstamp`, `crdate`, `cruser_id`, `deleted`, `hidden`, `title`, `storagepid`, `targetpid`, `startingpoints_recursive`, `single_pages`, `sysfolder`, `index_content_with_restrictions`, `index_passed_events`, `type`, `index_news_category_mode`, `index_news_category_selection`, `index_extnews_category_selection`, `index_news_useHRDatesSingle`, `index_news_useHRDatesSingleWithoutDay`, `index_dam_categories`, `index_dam_without_categories`, `index_dam_categories_recursive`, `index_use_page_tags`, `index_use_page_tags_for_files`, `directories`, `fileext`, `commenttypes`, `filteroption`, `tvpath`, `fal_storage`) VALUES
(1, 248, 1427708651, 1427708651, 1, 0, 0, 'Search Indexer', '5', NULL, '16', '', NULL, 'no', NULL, 'page', 0, NULL, NULL, 0, 0, NULL, 0, 0, 0, 0, NULL, 'pdf,ppt,doc,xls', 'pages', 0, 'field_content', -1);

-- --------------------------------------------------------

--
-- Tabellenstruktur für Tabelle `tx_kesearch_stat_search`
--

CREATE TABLE IF NOT EXISTS `tx_kesearch_stat_search` (
  `uid` int(11) NOT NULL AUTO_INCREMENT,
  `pid` int(11) NOT NULL DEFAULT '0',
  `searchphrase` text,
  `tstamp` int(11) NOT NULL DEFAULT '0',
  `hits` int(11) NOT NULL DEFAULT '0',
  `tagsagainst` text,
  `language` int(11) NOT NULL DEFAULT '0',
  PRIMARY KEY (`uid`)
) ENGINE=MyISAM  DEFAULT CHARSET=latin1 AUTO_INCREMENT=43 ;

--
-- Daten für Tabelle `tx_kesearch_stat_search`
--

INSERT INTO `tx_kesearch_stat_search` (`uid`, `pid`, `searchphrase`, `tstamp`, `hits`, `tagsagainst`, `language`) VALUES
(1, 5, 'test', 1427709033, 7, '', 0),
(2, 5, 'test', 1427709074, 7, '', 0),
(3, 5, 'test', 1427709125, 7, '', 0),
(4, 5, 'test', 1427709420, 7, '', 0),
(5, 5, 'test', 1427709422, 7, '', 0),
(6, 5, 'test', 1427709432, 7, '', 0),
(7, 5, 'test', 1427709578, 7, '', 0),
(8, 5, 'test', 1427709612, 7, '', 0),
(9, 5, 'test', 1427709731, 7, '', 0),
(10, 5, 'test', 1427709733, 7, '', 0),
(11, 5, 'test', 1427709878, 7, '', 0),
(12, 5, 'test', 1427709888, 7, '', 0),
(13, 5, 'test', 1427709912, 7, '', 0),
(14, 5, 'oca50', 1427709931, 0, '', 0),
(15, 5, 'anwed', 1427709938, 0, '', 0),
(16, 5, 'anwend', 1427709940, 35, '', 0),
(17, 5, 'anwend', 1427709945, 35, '', 0),
(18, 5, 'anwend', 1427710019, 35, '', 0),
(19, 5, 'anwend', 1427710084, 35, '', 0),
(20, 5, 'anwend', 1427710165, 35, '', 0),
(21, 5, 'anwend', 1427710183, 35, '', 0),
(22, 5, 'test', 1428084106, 7, '', 0),
(23, 5, 'dca11', 1428162213, 0, '', 0),
(24, 5, 'test', 1428338090, 7, '', 0),
(25, 5, 'test', 1428345495, 7, '', 0),
(26, 5, 'contact', 1428386159, 41, '', 2),
(27, 5, 'contact', 1428386298, 41, '', 2),
(28, 5, 'travail', 1428386315, 2, '', 3),
(29, 5, 'test', 1428391689, 6, '', 0),
(30, 5, 'test', 1428829588, 12, '', 2),
(31, 5, 'nils', 1429191543, 2, '', 2),
(32, 5, 'test', 1429360044, 6, '', 0),
(33, 5, 'test', 1429538645, 6, '', 0),
(34, 5, 'test', 1429538877, 6, '', 0),
(35, 5, 'test', 1429539031, 6, '', 0),
(36, 5, 'dcat', 1429539036, 14, '', 0),
(37, 5, 'dcat', 1429539039, 14, '', 0),
(38, 5, 'dcat', 1429539041, 14, '', 0),
(39, 5, 'rg11', 1429622565, 0, '', 2),
(40, 5, 'dcat', 1429622575, 14, '', 2),
(41, 5, 'test', 1431350306, 6, '', 0),
(42, 5, 'test', 1431355535, 9, '', 3);

-- --------------------------------------------------------

--
-- Tabellenstruktur für Tabelle `tx_kesearch_stat_word`
--

CREATE TABLE IF NOT EXISTS `tx_kesearch_stat_word` (
  `uid` int(11) NOT NULL AUTO_INCREMENT,
  `pid` int(11) NOT NULL DEFAULT '0',
  `word` text,
  `tstamp` int(11) NOT NULL DEFAULT '0',
  `pageid` int(11) NOT NULL DEFAULT '0',
  `resultsfound` int(1) NOT NULL DEFAULT '0',
  `language` int(11) NOT NULL DEFAULT '0',
  PRIMARY KEY (`uid`)
) ENGINE=MyISAM  DEFAULT CHARSET=latin1 AUTO_INCREMENT=43 ;

--
-- Daten für Tabelle `tx_kesearch_stat_word`
--

INSERT INTO `tx_kesearch_stat_word` (`uid`, `pid`, `word`, `tstamp`, `pageid`, `resultsfound`, `language`) VALUES
(1, 5, 'test', 1427709033, 5, 1, 0),
(2, 5, 'test', 1427709074, 5, 1, 0),
(3, 5, 'test', 1427709125, 5, 1, 0),
(4, 5, 'test', 1427709420, 5, 1, 0),
(5, 5, 'test', 1427709422, 5, 1, 0),
(6, 5, 'test', 1427709432, 5, 1, 0),
(7, 5, 'test', 1427709578, 5, 1, 0),
(8, 5, 'test', 1427709612, 5, 1, 0),
(9, 5, 'test', 1427709731, 5, 1, 0),
(10, 5, 'test', 1427709733, 5, 1, 0),
(11, 5, 'test', 1427709878, 5, 1, 0),
(12, 5, 'test', 1427709888, 5, 1, 0),
(13, 5, 'test', 1427709912, 5, 1, 0),
(14, 5, 'oca50', 1427709931, 5, 0, 0),
(15, 5, 'anwed', 1427709938, 5, 0, 0),
(16, 5, 'anwend', 1427709940, 5, 1, 0),
(17, 5, 'anwend', 1427709945, 5, 1, 0),
(18, 5, 'anwend', 1427710019, 5, 1, 0),
(19, 5, 'anwend', 1427710084, 5, 1, 0),
(20, 5, 'anwend', 1427710165, 5, 1, 0),
(21, 5, 'anwend', 1427710183, 5, 1, 0),
(22, 5, 'test', 1428084106, 5, 1, 0),
(23, 5, 'dca11', 1428162213, 5, 0, 0),
(24, 5, 'test', 1428338090, 5, 1, 0),
(25, 5, 'test', 1428345495, 5, 1, 0),
(26, 5, 'contact', 1428386159, 5, 1, 2),
(27, 5, 'contact', 1428386298, 5, 1, 2),
(28, 5, 'travail', 1428386315, 5, 1, 3),
(29, 5, 'test', 1428391689, 5, 1, 0),
(30, 5, 'test', 1428829588, 5, 1, 2),
(31, 5, 'nils', 1429191543, 5, 1, 2),
(32, 5, 'test', 1429360044, 5, 1, 0),
(33, 5, 'test', 1429538645, 5, 1, 0),
(34, 5, 'test', 1429538877, 5, 1, 0),
(35, 5, 'test', 1429539031, 5, 1, 0),
(36, 5, 'dcat', 1429539036, 5, 1, 0),
(37, 5, 'dcat', 1429539039, 5, 1, 0),
(38, 5, 'dcat', 1429539041, 5, 1, 0),
(39, 5, 'rg11', 1429622565, 5, 0, 2),
(40, 5, 'dcat', 1429622575, 5, 1, 2),
(41, 5, 'test', 1431350306, 5, 1, 0),
(42, 5, 'test', 1431355535, 5, 1, 3);

-- --------------------------------------------------------

--
-- Tabellenstruktur für Tabelle `tx_powermail_domain_model_answers`
--

CREATE TABLE IF NOT EXISTS `tx_powermail_domain_model_answers` (
  `uid` int(11) NOT NULL AUTO_INCREMENT,
  `pid` int(11) NOT NULL DEFAULT '0',
  `mail` int(11) unsigned NOT NULL DEFAULT '0',
  `value` text NOT NULL,
  `value_type` int(11) unsigned NOT NULL DEFAULT '0',
  `field` int(11) unsigned NOT NULL DEFAULT '0',
  `tstamp` int(11) unsigned NOT NULL DEFAULT '0',
  `crdate` int(11) unsigned NOT NULL DEFAULT '0',
  `cruser_id` int(11) unsigned NOT NULL DEFAULT '0',
  `deleted` tinyint(4) unsigned NOT NULL DEFAULT '0',
  `hidden` tinyint(4) unsigned NOT NULL DEFAULT '0',
  `starttime` int(11) unsigned NOT NULL DEFAULT '0',
  `endtime` int(11) unsigned NOT NULL DEFAULT '0',
  `t3ver_oid` int(11) NOT NULL DEFAULT '0',
  `t3ver_id` int(11) NOT NULL DEFAULT '0',
  `t3ver_wsid` int(11) NOT NULL DEFAULT '0',
  `t3ver_label` varchar(255) NOT NULL DEFAULT '',
  `t3ver_state` tinyint(4) NOT NULL DEFAULT '0',
  `t3ver_stage` int(11) NOT NULL DEFAULT '0',
  `t3ver_count` int(11) NOT NULL DEFAULT '0',
  `t3ver_tstamp` int(11) NOT NULL DEFAULT '0',
  `t3ver_move_id` int(11) NOT NULL DEFAULT '0',
  `t3_origuid` int(11) NOT NULL DEFAULT '0',
  `sys_language_uid` int(11) NOT NULL DEFAULT '0',
  `l10n_parent` int(11) NOT NULL DEFAULT '0',
  `l10n_diffsource` mediumblob,
  PRIMARY KEY (`uid`),
  KEY `parent` (`pid`),
  KEY `t3ver_oid` (`t3ver_oid`,`t3ver_wsid`),
  KEY `language` (`l10n_parent`,`sys_language_uid`)
) ENGINE=InnoDB  DEFAULT CHARSET=latin1 AUTO_INCREMENT=507 ;

--
-- Daten für Tabelle `tx_powermail_domain_model_answers`
--

INSERT INTO `tx_powermail_domain_model_answers` (`uid`, `pid`, `mail`, `value`, `value_type`, `field`, `tstamp`, `crdate`, `cruser_id`, `deleted`, `hidden`, `starttime`, `endtime`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `sys_language_uid`, `l10n_parent`, `l10n_diffsource`) VALUES
(1, 15, 1, '', 1, 1, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(2, 15, 1, '', 0, 2, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(3, 15, 1, 'test', 0, 3, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(4, 15, 1, '', 0, 4, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(5, 15, 1, 'test', 0, 5, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(6, 15, 1, '', 0, 6, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(7, 15, 1, '', 0, 7, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(8, 15, 1, '', 0, 8, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(9, 15, 1, '', 0, 9, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(10, 15, 1, '', 0, 10, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(11, 15, 1, '123123', 0, 11, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(12, 15, 1, '', 0, 12, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(13, 15, 1, 'a@a.de', 0, 13, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(14, 15, 1, '', 0, 14, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(15, 15, 1, '10', 0, 17, 1428341284, 1428341284, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(16, 251, 2, '', 1, 68, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(17, 251, 2, '', 0, 67, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(18, 251, 2, 'test', 0, 66, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(19, 251, 2, '', 0, 65, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(20, 251, 2, 'test', 0, 64, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(21, 251, 2, '', 0, 63, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(22, 251, 2, '', 0, 62, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(23, 251, 2, '', 0, 61, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(24, 251, 2, '', 0, 60, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(25, 251, 2, '', 0, 59, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(26, 251, 2, '0712-12312', 0, 58, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(27, 251, 2, '', 0, 57, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(28, 251, 2, 'bn@cosabo.de', 0, 56, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(29, 251, 2, '', 0, 55, 1429696765, 1428924040, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(30, 251, 3, 'asda', 0, 86, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(31, 251, 3, 'asdas', 0, 87, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(32, 251, 3, 'asdads', 0, 88, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(33, 251, 3, '1111', 0, 89, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(34, 251, 3, 'asdasd', 0, 90, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(35, 251, 3, 'asdas', 0, 91, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(36, 251, 3, 'asdasd', 0, 92, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(37, 251, 3, 'asdasd', 0, 93, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(38, 251, 3, 'asdasd', 0, 94, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(39, 251, 3, 'asdasd', 0, 95, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(40, 251, 3, 'asdasd', 0, 96, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(41, 251, 3, 'asdasd', 0, 97, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(42, 251, 3, 'asdasd', 0, 98, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(43, 251, 3, 'Steht noch aus', 0, 101, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(44, 251, 3, 'asd', 0, 161, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(45, 251, 3, 'asd', 0, 160, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(46, 251, 3, 'asd', 0, 159, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(47, 251, 3, 'asd', 0, 158, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(48, 251, 3, 'asd', 0, 157, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(49, 251, 3, 'asd', 0, 156, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(50, 251, 3, 'asd', 0, 155, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(51, 251, 3, 'asd', 0, 154, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(52, 251, 3, 'asd', 0, 153, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(53, 251, 3, 'asd', 0, 152, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(54, 251, 3, 'asd', 0, 151, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(55, 251, 3, 'asd', 0, 150, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(56, 251, 3, 'asd', 0, 149, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(57, 251, 3, 'asd', 0, 148, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(58, 251, 3, 'asd', 0, 147, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(59, 251, 3, '', 0, 146, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(60, 251, 3, 'asd', 0, 145, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(61, 251, 3, 'asd', 0, 144, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(62, 251, 3, 'asd', 0, 143, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(63, 251, 3, '', 0, 142, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(64, 251, 3, 'asd', 0, 141, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(65, 251, 3, 'asd', 0, 140, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(66, 251, 3, 'asd', 0, 139, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(67, 251, 3, '', 0, 138, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(68, 251, 3, 'asd', 0, 137, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(69, 251, 3, 'asd', 0, 136, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(70, 251, 3, 'asd', 0, 135, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(71, 251, 3, 'asd', 0, 134, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(72, 251, 3, 'asd', 0, 133, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(73, 251, 3, 'asd', 0, 132, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(74, 251, 3, 'asd', 0, 131, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(75, 251, 3, 'asd', 0, 130, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(76, 251, 3, 'asd', 0, 129, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(77, 251, 3, 'asd', 0, 128, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(78, 251, 3, '', 0, 127, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(79, 251, 3, '', 0, 126, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(80, 251, 3, 'asd', 0, 125, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(81, 251, 3, '', 0, 124, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(82, 251, 3, '', 0, 123, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(83, 251, 3, '', 0, 122, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(84, 251, 3, 'asd', 0, 121, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(85, 251, 3, '', 0, 120, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(86, 251, 3, '', 0, 119, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(87, 251, 3, '', 0, 118, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(88, 251, 3, 'asd', 0, 117, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(89, 251, 3, '', 0, 116, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(90, 251, 3, '', 0, 115, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(91, 251, 3, '', 0, 114, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(92, 251, 3, 'asd', 0, 113, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(93, 251, 3, '', 0, 112, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(94, 251, 3, '', 0, 111, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(95, 251, 3, '', 0, 110, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(96, 251, 3, 'asd', 0, 109, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(97, 251, 3, '', 0, 108, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(98, 251, 3, '', 0, 107, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(99, 251, 3, '', 0, 106, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(100, 251, 3, 'asd', 0, 102, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(101, 251, 3, '', 0, 103, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(102, 251, 3, '', 0, 104, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(103, 251, 3, '', 0, 105, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(104, 251, 3, 'asd', 0, 162, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(105, 251, 3, 'keine Kenntnisse', 0, 163, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(106, 251, 3, 'Grundkenntnisse', 0, 164, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(107, 251, 3, 'verhandlungssicher', 0, 166, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(108, 251, 3, 'asd', 0, 165, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(109, 251, 3, 'gut', 0, 167, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(110, 251, 3, 'asd', 0, 168, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(111, 251, 3, 'Grundkenntnisse', 1, 169, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(112, 251, 3, 'Nein', 0, 170, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(113, 251, 3, 'Ja', 0, 171, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(114, 251, 3, 'asd', 0, 172, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(115, 251, 3, 'asd', 0, 173, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(116, 251, 3, 'Schule', 1, 174, 1429696765, 1429602071, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(117, 251, 4, 'asda', 0, 86, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(118, 251, 4, 'asdas', 0, 87, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(119, 251, 4, 'asdads', 0, 88, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(120, 251, 4, '1111', 0, 89, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(121, 251, 4, 'asdasd', 0, 90, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(122, 251, 4, 'asdas', 0, 91, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(123, 251, 4, 'asdasd', 0, 92, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(124, 251, 4, 'asdasd', 0, 93, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(125, 251, 4, 'asdasd', 0, 94, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(126, 251, 4, 'asdasd', 0, 95, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(127, 251, 4, 'asdasd', 0, 96, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(128, 251, 4, 'asdasd', 0, 97, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(129, 251, 4, 'asdasd', 0, 98, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(130, 251, 4, 'Steht noch aus', 0, 101, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(131, 251, 4, 'asd', 0, 161, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(132, 251, 4, 'asd', 0, 160, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(133, 251, 4, 'asd', 0, 159, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(134, 251, 4, 'asd', 0, 158, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(135, 251, 4, 'asd', 0, 157, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(136, 251, 4, 'asd', 0, 156, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(137, 251, 4, 'asd', 0, 155, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(138, 251, 4, 'asd', 0, 154, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(139, 251, 4, 'asd', 0, 153, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(140, 251, 4, 'asd', 0, 152, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(141, 251, 4, 'asd', 0, 151, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(142, 251, 4, 'asd', 0, 150, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(143, 251, 4, 'asd', 0, 149, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(144, 251, 4, 'asd', 0, 148, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(145, 251, 4, 'asd', 0, 147, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(146, 251, 4, '', 0, 146, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(147, 251, 4, 'asd', 0, 145, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(148, 251, 4, 'asd', 0, 144, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(149, 251, 4, 'asd', 0, 143, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(150, 251, 4, '', 0, 142, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(151, 251, 4, 'asd', 0, 141, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(152, 251, 4, 'asd', 0, 140, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(153, 251, 4, 'asd', 0, 139, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(154, 251, 4, '', 0, 138, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(155, 251, 4, 'asd', 0, 137, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(156, 251, 4, 'asd', 0, 136, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(157, 251, 4, 'asd', 0, 135, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(158, 251, 4, 'asd', 0, 134, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(159, 251, 4, 'asd', 0, 133, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(160, 251, 4, 'asd', 0, 132, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(161, 251, 4, 'asd', 0, 131, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(162, 251, 4, 'asd', 0, 130, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(163, 251, 4, 'asd', 0, 129, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(164, 251, 4, 'asd', 0, 128, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(165, 251, 4, '', 0, 127, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(166, 251, 4, '', 0, 126, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(167, 251, 4, 'asd', 0, 125, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(168, 251, 4, '', 0, 124, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(169, 251, 4, '', 0, 123, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(170, 251, 4, '', 0, 122, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(171, 251, 4, 'asd', 0, 121, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(172, 251, 4, '', 0, 120, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(173, 251, 4, '', 0, 119, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(174, 251, 4, '', 0, 118, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(175, 251, 4, 'asd', 0, 117, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(176, 251, 4, '', 0, 116, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(177, 251, 4, '', 0, 115, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(178, 251, 4, '', 0, 114, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(179, 251, 4, 'asd', 0, 113, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(180, 251, 4, '', 0, 112, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(181, 251, 4, '', 0, 111, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(182, 251, 4, '', 0, 110, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(183, 251, 4, 'asd', 0, 109, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(184, 251, 4, '', 0, 108, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(185, 251, 4, '', 0, 107, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(186, 251, 4, '', 0, 106, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(187, 251, 4, 'asd', 0, 102, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(188, 251, 4, '', 0, 103, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(189, 251, 4, '', 0, 104, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(190, 251, 4, '', 0, 105, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(191, 251, 4, 'asd', 0, 162, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(192, 251, 4, 'keine Kenntnisse', 0, 163, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(193, 251, 4, 'Grundkenntnisse', 0, 164, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(194, 251, 4, 'verhandlungssicher', 0, 166, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(195, 251, 4, 'asd', 0, 165, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(196, 251, 4, 'gut', 0, 167, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(197, 251, 4, 'asd', 0, 168, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(198, 251, 4, 'Grundkenntnisse', 1, 169, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(199, 251, 4, 'Nein', 0, 170, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(200, 251, 4, 'Ja', 0, 171, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(201, 251, 4, 'asd', 0, 172, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(202, 251, 4, 'asd', 0, 173, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(203, 251, 4, 'Schule', 1, 174, 1429696765, 1429602298, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(204, 251, 5, 'asda', 0, 86, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(205, 251, 5, 'asdas', 0, 87, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(206, 251, 5, 'asdads', 0, 88, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(207, 251, 5, '1111', 0, 89, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(208, 251, 5, 'asdasd', 0, 90, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(209, 251, 5, 'asdas', 0, 91, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(210, 251, 5, 'asdasd', 0, 92, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(211, 251, 5, 'asdasd', 0, 93, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(212, 251, 5, 'asdasd', 0, 94, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(213, 251, 5, 'asdasd', 0, 95, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(214, 251, 5, 'asdasd', 0, 96, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(215, 251, 5, 'asdasd', 0, 97, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(216, 251, 5, 'asdasd', 0, 98, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(217, 251, 5, 'Steht noch aus', 0, 101, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(218, 251, 5, 'asd', 0, 161, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(219, 251, 5, 'asd', 0, 160, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(220, 251, 5, 'asd', 0, 159, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(221, 251, 5, 'asd', 0, 158, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(222, 251, 5, 'asd', 0, 157, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(223, 251, 5, 'asd', 0, 156, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(224, 251, 5, 'asd', 0, 155, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(225, 251, 5, 'asd', 0, 154, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(226, 251, 5, 'asd', 0, 153, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(227, 251, 5, 'asd', 0, 152, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(228, 251, 5, 'asd', 0, 151, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(229, 251, 5, 'asd', 0, 150, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(230, 251, 5, 'asd', 0, 149, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(231, 251, 5, 'asd', 0, 148, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(232, 251, 5, 'asd', 0, 147, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(233, 251, 5, '', 0, 146, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(234, 251, 5, 'asd', 0, 145, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(235, 251, 5, 'asd', 0, 144, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(236, 251, 5, 'asd', 0, 143, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(237, 251, 5, '', 0, 142, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(238, 251, 5, 'asd', 0, 141, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(239, 251, 5, 'asd', 0, 140, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(240, 251, 5, 'asd', 0, 139, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(241, 251, 5, '', 0, 138, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(242, 251, 5, 'asd', 0, 137, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(243, 251, 5, 'asd', 0, 136, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(244, 251, 5, 'asd', 0, 135, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(245, 251, 5, 'asd', 0, 134, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(246, 251, 5, 'asd', 0, 133, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(247, 251, 5, 'asd', 0, 132, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(248, 251, 5, 'asd', 0, 131, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(249, 251, 5, 'asd', 0, 130, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(250, 251, 5, 'asd', 0, 129, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(251, 251, 5, 'asd', 0, 128, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(252, 251, 5, '', 0, 127, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(253, 251, 5, '', 0, 126, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(254, 251, 5, 'asd', 0, 125, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(255, 251, 5, '', 0, 124, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(256, 251, 5, '', 0, 123, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(257, 251, 5, '', 0, 122, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(258, 251, 5, 'asd', 0, 121, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(259, 251, 5, '', 0, 120, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(260, 251, 5, '', 0, 119, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(261, 251, 5, '', 0, 118, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(262, 251, 5, 'asd', 0, 117, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(263, 251, 5, '', 0, 116, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(264, 251, 5, '', 0, 115, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(265, 251, 5, '', 0, 114, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(266, 251, 5, 'asd', 0, 113, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(267, 251, 5, '', 0, 112, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(268, 251, 5, '', 0, 111, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(269, 251, 5, '', 0, 110, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(270, 251, 5, 'asd', 0, 109, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(271, 251, 5, '', 0, 108, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(272, 251, 5, '', 0, 107, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(273, 251, 5, '', 0, 106, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(274, 251, 5, 'asd', 0, 102, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(275, 251, 5, '', 0, 103, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(276, 251, 5, '', 0, 104, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(277, 251, 5, '', 0, 105, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(278, 251, 5, 'asd', 0, 162, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(279, 251, 5, 'keine Kenntnisse', 0, 163, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(280, 251, 5, 'Grundkenntnisse', 0, 164, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(281, 251, 5, 'verhandlungssicher', 0, 166, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(282, 251, 5, 'asd', 0, 165, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(283, 251, 5, 'gut', 0, 167, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(284, 251, 5, 'asd', 0, 168, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(285, 251, 5, 'Grundkenntnisse', 1, 169, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(286, 251, 5, 'Nein', 0, 170, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(287, 251, 5, 'Ja', 0, 171, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(288, 251, 5, 'asd', 0, 172, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(289, 251, 5, 'asd', 0, 173, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(290, 251, 5, 'Schule', 1, 174, 1429696765, 1429602547, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(291, 251, 6, 'asda', 0, 86, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(292, 251, 6, 'asdas', 0, 87, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(293, 251, 6, 'asdads', 0, 88, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(294, 251, 6, '1111', 0, 89, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(295, 251, 6, 'asdasd', 0, 90, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(296, 251, 6, 'asdas', 0, 91, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(297, 251, 6, 'asdasd', 0, 92, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(298, 251, 6, 'asdasd', 0, 93, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(299, 251, 6, 'asdasd', 0, 94, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(300, 251, 6, 'asdasd', 0, 95, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(301, 251, 6, 'asdasd', 0, 96, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(302, 251, 6, 'asdasd', 0, 97, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(303, 251, 6, 'asdasd', 0, 98, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(304, 251, 6, 'Steht noch aus', 0, 101, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(305, 251, 6, 'asd', 0, 161, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(306, 251, 6, 'asd', 0, 160, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(307, 251, 6, 'asd', 0, 159, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(308, 251, 6, 'asd', 0, 158, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(309, 251, 6, 'asd', 0, 157, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(310, 251, 6, 'asd', 0, 156, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(311, 251, 6, 'asd', 0, 155, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(312, 251, 6, 'asd', 0, 154, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(313, 251, 6, 'asd', 0, 153, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(314, 251, 6, 'asd', 0, 152, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(315, 251, 6, 'asd', 0, 151, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(316, 251, 6, 'asd', 0, 150, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(317, 251, 6, 'asd', 0, 149, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(318, 251, 6, 'asd', 0, 148, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(319, 251, 6, 'asd', 0, 147, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(320, 251, 6, '', 0, 146, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(321, 251, 6, 'asd', 0, 145, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(322, 251, 6, 'asd', 0, 144, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(323, 251, 6, 'asd', 0, 143, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(324, 251, 6, '', 0, 142, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(325, 251, 6, 'asd', 0, 141, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(326, 251, 6, 'asd', 0, 140, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(327, 251, 6, 'asd', 0, 139, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(328, 251, 6, '', 0, 138, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(329, 251, 6, 'asd', 0, 137, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(330, 251, 6, 'asd', 0, 136, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(331, 251, 6, 'asd', 0, 135, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(332, 251, 6, 'asd', 0, 134, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(333, 251, 6, 'asd', 0, 133, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(334, 251, 6, 'asd', 0, 132, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(335, 251, 6, 'asd', 0, 131, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(336, 251, 6, 'asd', 0, 130, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(337, 251, 6, 'asd', 0, 129, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(338, 251, 6, 'asd', 0, 128, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(339, 251, 6, '', 0, 127, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(340, 251, 6, '', 0, 126, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(341, 251, 6, 'asd', 0, 125, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(342, 251, 6, '', 0, 124, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(343, 251, 6, '', 0, 123, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(344, 251, 6, '', 0, 122, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(345, 251, 6, 'asd', 0, 121, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(346, 251, 6, '', 0, 120, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(347, 251, 6, '', 0, 119, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(348, 251, 6, '', 0, 118, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(349, 251, 6, 'asd', 0, 117, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(350, 251, 6, '', 0, 116, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(351, 251, 6, '', 0, 115, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(352, 251, 6, '', 0, 114, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(353, 251, 6, 'asd', 0, 113, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(354, 251, 6, '', 0, 112, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(355, 251, 6, '', 0, 111, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(356, 251, 6, '', 0, 110, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(357, 251, 6, 'asd', 0, 109, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(358, 251, 6, '', 0, 108, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(359, 251, 6, '', 0, 107, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(360, 251, 6, '', 0, 106, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(361, 251, 6, 'asd', 0, 102, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(362, 251, 6, '', 0, 103, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(363, 251, 6, '', 0, 104, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(364, 251, 6, '', 0, 105, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(365, 251, 6, 'asd', 0, 162, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(366, 251, 6, 'keine Kenntnisse', 0, 163, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(367, 251, 6, 'Grundkenntnisse', 0, 164, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(368, 251, 6, 'verhandlungssicher', 0, 166, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(369, 251, 6, 'asd', 0, 165, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(370, 251, 6, 'gut', 0, 167, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(371, 251, 6, 'asd', 0, 168, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(372, 251, 6, 'Grundkenntnisse', 1, 169, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(373, 251, 6, 'Nein', 0, 170, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(374, 251, 6, 'Ja', 0, 171, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(375, 251, 6, 'asd', 0, 172, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(376, 251, 6, 'asd', 0, 173, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(377, 251, 6, 'Schule', 1, 174, 1429696765, 1429602586, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(378, 251, 7, 'asda', 0, 86, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(379, 251, 7, 'asdas', 0, 87, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(380, 251, 7, 'asdads', 0, 88, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(381, 251, 7, '1111', 0, 89, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(382, 251, 7, 'asdasd', 0, 90, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(383, 251, 7, 'asdas', 0, 91, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(384, 251, 7, 'asdasd', 0, 92, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(385, 251, 7, 'asdasd', 0, 93, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(386, 251, 7, 'asdasd', 0, 94, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(387, 251, 7, 'asdasd', 0, 95, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(388, 251, 7, 'asdasd', 0, 96, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(389, 251, 7, 'asdasd', 0, 97, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(390, 251, 7, 'asdasd', 0, 98, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(391, 251, 7, 'Steht noch aus', 0, 101, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(392, 251, 7, 'asd', 0, 161, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(393, 251, 7, 'asd', 0, 160, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(394, 251, 7, 'asd', 0, 159, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(395, 251, 7, 'asd', 0, 158, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(396, 251, 7, 'asd', 0, 157, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(397, 251, 7, 'asd', 0, 156, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(398, 251, 7, 'asd', 0, 155, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(399, 251, 7, 'asd', 0, 154, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(400, 251, 7, 'asd', 0, 153, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(401, 251, 7, 'asd', 0, 152, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(402, 251, 7, 'asd', 0, 151, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(403, 251, 7, 'asd', 0, 150, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(404, 251, 7, 'asd', 0, 149, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(405, 251, 7, 'asd', 0, 148, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(406, 251, 7, 'asd', 0, 147, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(407, 251, 7, '', 0, 146, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(408, 251, 7, 'asd', 0, 145, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(409, 251, 7, 'asd', 0, 144, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(410, 251, 7, 'asd', 0, 143, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(411, 251, 7, '', 0, 142, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(412, 251, 7, 'asd', 0, 141, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(413, 251, 7, 'asd', 0, 140, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(414, 251, 7, 'asd', 0, 139, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(415, 251, 7, '', 0, 138, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(416, 251, 7, 'asd', 0, 137, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(417, 251, 7, 'asd', 0, 136, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(418, 251, 7, 'asd', 0, 135, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(419, 251, 7, 'asd', 0, 134, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(420, 251, 7, 'asd', 0, 133, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(421, 251, 7, 'asd', 0, 132, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(422, 251, 7, 'asd', 0, 131, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(423, 251, 7, 'asd', 0, 130, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(424, 251, 7, 'asd', 0, 129, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(425, 251, 7, 'asd', 0, 128, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(426, 251, 7, '', 0, 127, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(427, 251, 7, '', 0, 126, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(428, 251, 7, 'asd', 0, 125, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(429, 251, 7, '', 0, 124, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(430, 251, 7, '', 0, 123, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(431, 251, 7, '', 0, 122, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(432, 251, 7, 'asd', 0, 121, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(433, 251, 7, '', 0, 120, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(434, 251, 7, '', 0, 119, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(435, 251, 7, '', 0, 118, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(436, 251, 7, 'asd', 0, 117, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(437, 251, 7, '', 0, 116, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(438, 251, 7, '', 0, 115, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(439, 251, 7, '', 0, 114, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(440, 251, 7, 'asd', 0, 113, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(441, 251, 7, '', 0, 112, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(442, 251, 7, '', 0, 111, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(443, 251, 7, '', 0, 110, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(444, 251, 7, 'asd', 0, 109, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(445, 251, 7, '', 0, 108, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(446, 251, 7, '', 0, 107, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(447, 251, 7, '', 0, 106, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(448, 251, 7, 'asd', 0, 102, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(449, 251, 7, '', 0, 103, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(450, 251, 7, '', 0, 104, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(451, 251, 7, '', 0, 105, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(452, 251, 7, 'asd', 0, 162, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL);
INSERT INTO `tx_powermail_domain_model_answers` (`uid`, `pid`, `mail`, `value`, `value_type`, `field`, `tstamp`, `crdate`, `cruser_id`, `deleted`, `hidden`, `starttime`, `endtime`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `sys_language_uid`, `l10n_parent`, `l10n_diffsource`) VALUES
(453, 251, 7, 'keine Kenntnisse', 0, 163, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(454, 251, 7, 'Grundkenntnisse', 0, 164, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(455, 251, 7, 'verhandlungssicher', 0, 166, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(456, 251, 7, 'asd', 0, 165, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(457, 251, 7, 'gut', 0, 167, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(458, 251, 7, 'asd', 0, 168, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(459, 251, 7, 'Grundkenntnisse', 1, 169, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(460, 251, 7, 'Nein', 0, 170, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(461, 251, 7, 'Ja', 0, 171, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(462, 251, 7, 'asd', 0, 172, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(463, 251, 7, 'asd', 0, 173, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(464, 251, 7, 'Schule', 1, 174, 1429696765, 1429602646, 0, 1, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(465, 15, 8, '["Bite rufen Sie mich an"]', 1, 68, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(466, 15, 8, 'Test', 0, 67, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(467, 15, 8, 'Nübling', 0, 66, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(468, 15, 8, 'Boris', 0, 65, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(469, 15, 8, 'Test', 0, 64, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(470, 15, 8, 'Testweg', 0, 63, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(471, 15, 8, '222', 0, 62, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(472, 15, 8, '12343', 0, 61, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(473, 15, 8, 'Berlin', 0, 60, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(474, 15, 8, 'Deutschalnd', 0, 59, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(475, 15, 8, '0123123122', 0, 58, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(476, 15, 8, '01231231212', 0, 57, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(477, 15, 8, 'info@cosabo.de', 0, 56, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(478, 15, 8, '01231211', 0, 55, 1429684706, 1429684706, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(479, 15, 9, '["Ja"]', 1, 68, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(480, 15, 9, 'estset', 0, 67, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(481, 15, 9, 'estset', 0, 66, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(482, 15, 9, 'setset', 0, 65, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(483, 15, 9, 'setset', 0, 64, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(484, 15, 9, 'setset', 0, 63, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(485, 15, 9, '22', 0, 62, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(486, 15, 9, '2343', 0, 61, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(487, 15, 9, 'sdfsdf', 0, 60, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(488, 15, 9, 'sdfsdf', 0, 59, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(489, 15, 9, '07122-111', 0, 58, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(490, 15, 9, '', 0, 57, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(491, 15, 9, 'bn@cosabo.de', 0, 56, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(492, 15, 9, '', 0, 55, 1429684794, 1429684794, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(493, 15, 10, '["Yes"]', 1, 68, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(494, 15, 10, 'asdasd', 0, 67, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(495, 15, 10, 'asdas', 0, 66, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(496, 15, 10, 'asdasd', 0, 65, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(497, 15, 10, 'asdasd', 0, 64, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(498, 15, 10, '', 0, 63, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(499, 15, 10, '', 0, 62, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(500, 15, 10, '', 0, 61, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(501, 15, 10, '', 0, 60, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(502, 15, 10, '', 0, 59, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(503, 15, 10, '071231', 0, 58, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(504, 15, 10, '', 0, 57, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(505, 15, 10, 'bn@cosabo.de', 0, 56, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL),
(506, 15, 10, '', 0, 55, 1429685883, 1429685883, 0, 0, 0, 0, 0, 0, 0, 0, '', 0, 0, 0, 0, 0, 0, 0, 0, NULL);

-- --------------------------------------------------------

--
-- Tabellenstruktur für Tabelle `tx_powermail_domain_model_fields`
--

