Temperature controlled measuring cell
The SVT 20 can be readily equipped with a liquid/gas-temperature controlled measuring cell, MC-TFC 130, or with a peltier/gas-temperature controlled measuring cell, MC-TPC 180. Inside the measuring cell gas (preferably helium) can be used, as a heat transfer medium, to optimise the transfer of the required temperature regime to the measuring capillary. Every measuring cell is easily mountable/dismountable and includes the water-coolable high-dynamic measuring drive.
Measuring cell MC-TFC 130

MC-TFC 130
Temperature controlled measuring cell for installation onto the tilting stage of the SVT. For measurements lower or higher than room temperature. For use with a liquid circulator, e.g. the liquid cryostat F25 ME. The cell temperature is set by the liquid circulator. In the measurement chamber gas (preferably helium) can be used as a heat transfer medium, to optimise the transfer of the required temperature regime to the measuring capillary. Consisting of:
 Vibration free, electronically commuted, high-dynamic DC-Drive. Liquid/gas temperature controlled chamber with 2 observation windows made from optically clear glass with quick -clamp device for holding the capillary. PT100 temperature measurement sensor.
 Technical data:
 Temperature range -10 to 130 °C. Heating up and cooling down rate: dependant on the liquid circulator.

MC-TPC 180
Temperature controlled measuring cell for installation onto the tilting stage of the SVT. For measurements lower or higher than room temperature. For use with the PID Temperature controller, TC 180/MC-TPC and a liquid circulator, e.g. the liquid cryostat F25 ME. The temperature setting can be done at the TC 180/MC-TPC or at the touch panel of the SVT. In the measurement chamber gas (preferably helium) can be used as a heat transfer medium. Needs counter cooling with fluid circulator.
 Consisting of:  Vibration free, electronically commuted, high-dynamic DC-Drive. Peltier/gas temperature controlled chamber with 2 observation windows made from optically clear glass with quick clamp device for holding the capillary. Two resistance thermometers Pt 100 as measuring and control sensors.
 Technical data:
 Temperature range: -30 to 180 °C; ±0.1 K. Heat up and cool down rate: ±1 K/s.
