<p style="font-size: 11.9999990463257px; ">Device for the heating of gastight syringe DS xx/GT mounted in the electronic syringe module ES. For use in conjunction with a liquid circulator.</p>
<h2>TFC 100</h2>
<p style="font-size: 11.9999990463257px; ">Liquid temperature control unit for measurement within the range - 10 to 130 °C. Requires an external liquid circulator bath and includes the pipe work etc. to allow for a protective atmosphere (inert gas inlet) to surround the sample working area. The triple diffuser, TDI 100, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas flow towards the chamber windows.&nbsp;<br />An optional humidity sensor expands the applicability of the TFC 100 even further.</p>
<h2>TPC 150</h2>
<p style="font-size: 11.9999990463257px; ">Temperature control chamber with a peltier system for the software controlled temperature setting from -30 t0 160 °C (heat up and cool down rate ±1 K/s). Also includes the pipe work etc. to allow for a protective atmosphere to surround the working area. The triple diffuser, TDI 150, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas towards the chamber windows.</p>
<p style="font-size: 11.9999990463257px; ">An optional humidity sensor expands the applicability of the TPC 150 even further.</p>
<h2>TEC 300XL</h2>
<p style="font-size: 11.9999990463257px; ">For mounting to the XL-variants of the OCA measuring devices. Temperature control chamber with electric resistance heater for manual controlled temperature setting up to 300 °C. Temperature display on controller and in SCA-software. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>
<h2>TEC 400</h2>
<p style="font-size: 11.9999990463257px; ">Temperature control chamber with twin electric resistance heater (above and below the sample plate) for software controlled temperature setting up to 400 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>
<h2>TEC 700</h2>
<p style="font-size: 11.9999990463257px; ">Temperature control chambers with twin electric resistance heater (above and below the sample plate) for software control of temperature setting up to 700 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>
<h2>NHD 400 / NHD 700</h2>
<p style="font-size: 11.9999990463257px; ">The NHD 400 and NHD 700 are electrical needle heating devices for generating drops at temperatures up 400 °C and 700 °C respectively (Applications areas include: polymers and hot melts, molten metals, etc.). Also extremely tenacious copolymers, with viscosities in the kPas range, e.g. with elastomer constituents, can be dosed easily at these elevated temperatures.</p>
<h2>MTFQ 1200</h2>
<p style="font-size: 11.9999990463257px; ">Special high temperature furnace for measurements to a maximum of 1200°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the MTFQ 1200 also.</p>
<h2>HTFC 1500/1700</h2>
<p style="font-size: 11.9999990463257px; ">Special high temperature furnace for measurements to a maximum of 1500/1700°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the HTFQ 1500/1700 also.</p>
<h2>HTFC 1500/1800HV</h2>
<p style="font-size: 11.9999990463257px; ">Electrically heated, high temperature, high vacuum tube furnace for use in combination with an OCA 15LHT-SV.&nbsp;</p>
<ul style="font-size: 11.9999990463257px; "><li>Max. temperature: 1500/1800 °C</li><li>Max. temperature for operation under vacuum: 1450°C</li><li>Max. vacuum: 10<sup>-5</sup>&nbsp;mbar</li></ul>
<p style="font-size: 11.9999990463257px; "><br />vacuum tube:</p>
<p style="font-size: 11.9999990463257px; ">Material: C799<br />Outer diameter: 50 mm<br />Inner diameter: 40 mm<br />Length: 990 mm<br />-&nbsp;&nbsp;&nbsp; Prepared for horizontal operation&nbsp;<br />-&nbsp;&nbsp;&nbsp; Observation windows on both sides as additional equipment for gastight flanges<br />-&nbsp;&nbsp;&nbsp; MoSi2 heating elements, mounted vertically for easy replacement<br />-&nbsp;&nbsp;&nbsp; Insulation with vacuum-formed ceramic fibre plates for an energy-saving operation and fast heating up times<br />-&nbsp;&nbsp;&nbsp; Rectangular outer casing with slots for convection cooling<br />-&nbsp;&nbsp;&nbsp; Casing made of sheets of textured stainless steel (non-rusting design)<br />-&nbsp;&nbsp;&nbsp; Type B thermocouple<br />-&nbsp;&nbsp;&nbsp; Power unit with low-voltage transformer and thyristor controller&nbsp;<br />Typically used for the measurement of contact angle and surface tension of materials like:</p>
<div class="indent" style="font-size: 11.9999990463257px; "><ul><li>Compound Materials</li><li>Steel</li><li>Alloys</li><li>Molten Glass</li></ul></div>
<p style="font-size: 11.9999990463257px; ">&nbsp;</p>
<h2>PMC 50S</h2>
<p style="font-size: 11.9999990463257px; ">Pressure cell with two viewing windows and manipulation system to allow the exchange of the sample table. Facilitates the evaluation of contact angles according to the sessile and captive drop method, the measurement of surface tensions in liquid/pressurized gas systems and the interfacial tensions in liquid/liquid systems.&nbsp;</p>
<p style="font-size: 11.9999990463257px; ">Incorporates four dosing needles, for separate sessile and pendant drop measurements via four different injection liquids and a two phase pressure supply system for an external supply of the gas or liquid phases. The first pressure level is used for controlling the pressure of the primary filling liquid or gas. The higher, second pressure level is necessary for the injection dosing of the measuring liquid(s). The primary&nbsp; phase acts also a heat transfer medium.&nbsp;</p>
<ul style="font-size: 11.9999990463257px; "><li>pressure range: 0.03 ... 50 bar</li><li>max. temperature: 80 °C</li></ul>
<h2>HGC 20EC/20/30</h2>
<p style="font-size: 11.9999990463257px; ">For the automated regulation of the humidity of ambient clean air, nitrogen, argon, or any other inert gas in temperature controlled measuring chambers.</p>