INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1102, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 218, 1429629775, 1428312048, 1, 0, 992, 'textpic', 'Software modules SCA', '', 'The SCA software assists you in the intuitive use of all OCA instruments — whether making single CA/IFT measurements with the budget-priced OCA 15EC through to the fully automated contact angle and drop shape contour analysis instrument for microstructures, the OCA 40 Micro. If your application dictates that you choose a ''dynamic'' solution; whether using a standard video­ system with 52 images/s sample rate or the ultra high-speed video system with 2200 images/s sample rate the SCA software will guide you through the steps required to maximise the impact of this invaluable ''movie'' footage and the analysed data sets generated\r\n<link 208 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 218, 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, '', '', 0, 0, '', 0),
(1103, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 217, 1429629775, 1428312048, 1, 0, 1008, 'textpic', 'Sample positioning and holding units', '', 'The OCA range of accessories includes several sample positioning and securing units. In the unlikely event that we cannot identify a unit that will readily address your application, we should be able to adapt a device to facilitate your special requirements.\r\n<link 207 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 217, 0x613a33313a7b733a363a22636f6c506f73223b733a313a2234223b733a353a224354797065223b733a373a2274657874706963223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b733a363a22686561646572223b733a34323a2250726f62656e706f736974696f6e6965722d20756e642050726f62656e68616c74657273797374656d65223b733a31333a226865616465725f6c61796f7574223b733a313a2230223b733a31353a226865616465725f706f736974696f6e223b733a303a22223b733a343a2264617465223b733a313a2230223b733a31313a226865616465725f6c696e6b223b733a303a22223b733a383a22626f647974657874223b733a3139353a22486965722066696e64656e2053696520616c6c6573207761732053696520627261756368656e2c20756d20496872652050726f62656e20696e2064696520726963687469676520506f736974696f6e2066c3bc7220646965204d657373756e67207a75206272696e67656e0d0a3c6c696e6b20323037202d20696e7465726e616c2d6c696e6b20224f70656e7320696e7465726e616c206c696e6b20696e2063757272656e742077696e646f77223e6d65687220657266616872656e3c2f6c696e6b3e223b733a31313a227274655f656e61626c6564223b733a313a2230223b733a353a22696d616765223b733a313a2231223b733a31303a22696d6167655f7a6f6f6d223b733a313a2230223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a333a22313530223b733a31303a22696d6167657769647468223b733a313a2230223b733a31313a22696d616765686569676874223b733a313a2230223b733a31313a22696d616765626f72646572223b733a313a2230223b733a31373a22696d6167655f636f6d7072657373696f6e223b733a313a2230223b733a31333a22696d6167655f65666665637473223b733a313a2230223b733a31313a22696d6167656f7269656e74223b733a323a223137223b733a393a22696d616765636f6c73223b733a313a2232223b733a31323a22696d6167655f6e6f526f7773223b733a313a2230223b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b733a303a22223b733a363a2268696464656e223b733a313a2230223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1104, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 216, 1429629775, 1428312048, 1, 0, 1016, 'textpic', 'Temperature and environmental control unit', '', 'Dataphysics offer a wide range of temperature and environmental control systems that operate in conjunction with the OCA instruments and supportive software. These systems enable the measurement of surface and interfacial parameters across an incredibly diverse group of applications, at temperatures from -30 to 1800 °C and from a high-vacuum to high-pressure atmosphere. Temperature, pressure, humidity, and other physical parameters can be developed, directly, by these OCA instrument accessories, or with the integration of additional hardware.&nbsp;\r\n<link 206 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 216, 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, '', '', 0, 0, '', 0),
(1105, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 214, 1429629775, 1428312048, 1, 0, 1020, 'textpic', 'Liquid dosing', '', 'The most often varied accessories, employed to vary a basic instrument configuration, are electronic syringe units in combination with single or multiple dosing units. If you cannot find a solution to your particular dosing challenges here we will happily demonstrate our willingness to further customise —&nbsp; call us&nbsp;<link 238 - internal-link "Opens internal link in current window">directly</link>.\r\n<link 205 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 214, 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, '', '', 0, 0, '', 0),
(1106, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 211, 1428312193, 1428312116, 1, 0, 2944, 'text', 'OCA Accessory Range', '', '<h2>Special devices to enhance and expand the applicability of the OCA range.</h2>\r\nThe contact angle measuring instruments of the OCA series share a common feature – their modular design enables them to benefit from the many special devices within the highly versatile accessory range. Some are application specific; all have been developed with input from users and partners within academia and industry. This extensive range of accessories&nbsp; consists of various dosing systems, temperature and environment control systems, turn and tilting tables, sample positioning systems, and tilting base units. If you cannot find therein a solution for your uncommon samples or applications, feel free to ask us for special variants of our instruments and accessories. One of our strengths is the customisation of our products to your needs — please&nbsp;<link 238 - internal-link "Opens internal link in current window">call directly</link>&nbsp;or fill out the&nbsp;<link 15 - internal-link "Opens internal link in current window">contact form</link>.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 120, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 211, 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, '', '', 0, 0, '', 0),
(1107, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 213, 1428313564, 1428312196, 1, 0, 2368, 'image', '', '', NULL, '1', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 105, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 213, 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, '', '', 0, 0, '', 0),
(1108, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 804, 1429356958, 1428312196, 1, 0, 63, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 804, 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, '', '', 0, 0, '', 0),
(1109, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 220, 1429629835, 1428313298, 1, 0, 832, 'textpic', 'Accessoires spéciaux', '', 'In close cooperation with customers, research institutes, and universities we have developed a wide range of application specific accessories to complement the OCA series of instruments. In doing so we are continuously broadening the range of surface chemistry applications within both the laboratory and process fields.\r\n<link 210 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 220, 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, '', '', 0, 0, '', 0),
(1110, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 219, 1429629835, 1428313298, 1, 0, 864, 'textpic', 'Systèmes de vidéo de grande vitesse', '', 'All instruments within the OCA series can be upgraded to include a high-speed video system. Your sales representative will consider your application(s), together with yourself, to determine the optimum system for your application.\r\n<link 209 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 219, 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, '', '', 0, 0, '', 0),
(1111, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 218, 1429629835, 1428313298, 1, 0, 880, 'textpic', 'Module de logiciel SCA', '', 'The SCA software assists you in the intuitive use of all OCA instruments — whether making single CA/IFT measurements with the budget-priced OCA 15EC through to the fully automated contact angle and drop shape contour analysis instrument for microstructures, the OCA 40 Micro. If your application dictates that you choose a ''dynamic'' solution; whether using a standard video­ system with 52 images/s sample rate or the ultra high-speed video system with 2200 images/s sample rate the SCA software will guide you through the steps required to maximise the impact of this invaluable ''movie'' footage and the analysed data sets generated\r\n<link 208 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 218, 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, '', '', 0, 0, '', 0),
(1112, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 217, 1429629835, 1428313298, 1, 0, 888, 'textpic', 'Positionnement de promotion et parenthèses', '', 'The OCA range of accessories includes several sample positioning and securing units. In the unlikely event that we cannot identify a unit that will readily address your application, we should be able to adapt a device to facilitate your special requirements.\r\n<link 207 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 217, 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, '', '', 0, 0, '', 0),
(1113, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 216, 1429629835, 1428313298, 1, 0, 892, 'textpic', 'Unités de contrôle de température et de l''environnement', '', 'Dataphysics offer a wide range of temperature and environmental control systems that operate in conjunction with the OCA instruments and supportive software. These systems enable the measurement of surface and interfacial parameters across an incredibly diverse group of applications, at temperatures from -30 to 1800 °C and from a high-vacuum to high-pressure atmosphere. Temperature, pressure, humidity, and other physical parameters can be developed, directly, by these OCA instrument accessories, or with the integration of additional hardware.&nbsp;\r\n<link 206 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 216, 0x613a33313a7b733a363a22636f6c506f73223b733a313a2234223b733a353a224354797065223b733a373a2274657874706963223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b733a363a22686561646572223b733a33303a2254656d706572617475722d20756e64204b6c696d61737465756572756e67223b733a31333a226865616465725f6c61796f7574223b733a313a2230223b733a31353a226865616465725f706f736974696f6e223b733a303a22223b733a343a2264617465223b733a313a2230223b733a31313a226865616465725f6c696e6b223b733a303a22223b733a383a22626f647974657874223b733a3136303a22486965722066696e64656e205369652065696e65205669656c7a61686c20616e2054656d706572617475722d20756e64204b6c696d616b6f6e74726f6c6cc2ad73797374656d656e0d0a3c6c696e6b20323036202d20696e7465726e616c2d6c696e6b20224f70656e7320696e7465726e616c206c696e6b20696e2063757272656e742077696e646f77223e6d65687220657266616872656e3c2f6c696e6b3e223b733a31313a227274655f656e61626c6564223b733a313a2230223b733a353a22696d616765223b733a313a2231223b733a31303a22696d6167655f7a6f6f6d223b733a313a2230223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a333a22313530223b733a31303a22696d6167657769647468223b733a333a22343030223b733a31313a22696d616765686569676874223b733a333a22323033223b733a31313a22696d616765626f72646572223b733a313a2230223b733a31373a22696d6167655f636f6d7072657373696f6e223b733a313a2230223b733a31333a22696d6167655f65666665637473223b733a313a2230223b733a31313a22696d6167656f7269656e74223b733a323a223137223b733a393a22696d616765636f6c73223b733a313a2232223b733a31323a22696d6167655f6e6f526f7773223b733a313a2230223b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b733a303a22223b733a363a2268696464656e223b733a313a2230223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1114, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 214, 1429629835, 1428313298, 1, 0, 894, 'textpic', 'Dosage liquide', '', 'The most often varied accessories, employed to vary a basic instrument configuration, are electronic syringe units in combination with single or multiple dosing units. If you cannot find a solution to your particular dosing challenges here we will happily demonstrate our willingness to further customise —&nbsp; call us&nbsp;<link 238 - internal-link "Opens internal link in current window">directly</link>.\r\n<link 205 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 150, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 214, 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, '', '', 0, 0, '', 0),
(1115, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 804, 1428313742, 1428313713, 1, 0, 4352, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1116, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 805, 1428313742, 1428313713, 1, 0, 4096, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1115, 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, '', '', 0, 0, '', 0),
(1117, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1108, 1428313743, 1428313713, 1, 0, 3840, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1115, 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, '', '', 0, 0, '', 0),
(1118, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 213, 1428313737, 1428313713, 1, 1, 1000000000, 'image', ' (copy 1)', '', NULL, '1', 0, 0, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 105, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0),
(1119, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 752, 1428313737, 1428313713, 1, 1, 1000000000, 'image', ' (copy 2)', '', NULL, '1', 0, 0, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 105, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1118, 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, '', '', 0, 0, '', 0),
(1120, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1107, 1428313737, 1428313713, 1, 1, 1000000000, 'image', ' (copy 3)', '', NULL, '1', 0, 0, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 105, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1118, 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, '', '', 0, 0, '', 0),
(1121, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 214, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Flüssigkeitsdosiersysteme', '', 'Das wichtigste Zubehör des Baukastens sind Dosiermodule in Kombination mit Einfach- oder Mehrfachdosiersystemen.\r\n<link 205 - internal-link "Opens internal link in current window">mehr erfahren</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1122, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1105, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Liquid dosing', '', 'The most often varied accessories, employed to vary a basic instrument configuration, are electronic syringe units in combination with single or multiple dosing units. If you cannot find a solution to your particular dosing challenges here we will happily demonstrate our willingness to further customise —&nbsp; call us&nbsp;<link 238 - internal-link "Opens internal link in current window">directly</link>.\r\n<link 205 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1121, 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, '', '', 0, 0, '', 0),
(1123, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1114, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Dosage liquide', '', 'The most often varied accessories, employed to vary a basic instrument configuration, are electronic syringe units in combination with single or multiple dosing units. If you cannot find a solution to your particular dosing challenges here we will happily demonstrate our willingness to further customise —&nbsp; call us&nbsp;<link 238 - internal-link "Opens internal link in current window">directly</link>.\r\n<link 205 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1121, 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, '', '', 0, 0, '', 0),
(1124, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 216, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Temperatur- und Klimasteuerung', '', 'Hier finden Sie eine Vielzahl an Temperatur- und Klimakontroll­systemen\r\n<link 206 - internal-link "Opens internal link in current window">mehr erfahren</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0),
(1125, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1104, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Temperature and environmental control unit', '', 'Dataphysics offer a wide range of temperature and environmental control systems that operate in conjunction with the OCA instruments and supportive software. These systems enable the measurement of surface and interfacial parameters across an incredibly diverse group of applications, at temperatures from -30 to 1800 °C and from a high-vacuum to high-pressure atmosphere. Temperature, pressure, humidity, and other physical parameters can be developed, directly, by these OCA instrument accessories, or with the integration of additional hardware.&nbsp;\r\n<link 206 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1124, 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, '', '', 0, 0, '', 0),
(1126, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1113, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Unités de contrôle de température et de l''environnement', '', 'Dataphysics offer a wide range of temperature and environmental control systems that operate in conjunction with the OCA instruments and supportive software. These systems enable the measurement of surface and interfacial parameters across an incredibly diverse group of applications, at temperatures from -30 to 1800 °C and from a high-vacuum to high-pressure atmosphere. Temperature, pressure, humidity, and other physical parameters can be developed, directly, by these OCA instrument accessories, or with the integration of additional hardware.&nbsp;\r\n<link 206 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1124, 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, '', '', 0, 0, '', 0),
(1127, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 217, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Probenpositionier- und Probenhaltersysteme', '', 'Hier finden Sie alles was Sie brauchen, um Ihre Proben in die richtige Position für die Messung zu bringen\r\n<link 207 - internal-link "Opens internal link in current window">mehr erfahren</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0),
(1128, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1103, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Sample positioning and holding units', '', 'The OCA range of accessories includes several sample positioning and securing units. In the unlikely event that we cannot identify a unit that will readily address your application, we should be able to adapt a device to facilitate your special requirements.\r\n<link 207 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1127, 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, '', '', 0, 0, '', 0),
(1129, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1112, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Positionnement de promotion et parenthèses', '', 'The OCA range of accessories includes several sample positioning and securing units. In the unlikely event that we cannot identify a unit that will readily address your application, we should be able to adapt a device to facilitate your special requirements.\r\n<link 207 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1127, 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, '', '', 0, 0, '', 0),
(1130, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 218, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Software-Module', '', 'Die SCA Software wurde als modulares Programm für alle Kontaktwinkelmessgeräte der OCA Serie entwickelt, vom preisgünstigen OCA 15EC bis hin zum voll­auto­ma­tischen Konturanalysesystem für Mikrostrukturen OCA 40Micro, für Standard Video­systeme mit 52 Bildern/s Bildaufnahmerate bis hin zu Hochgeschwindigkeits-Bild­ verarbeitungssystemen mit bis zu 2200 Bildern/s Bildaufnahmerate .\r\n<link 208 - internal-link "Opens internal link in current window">mehr erfahren</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0),
(1131, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1102, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Software modules SCA', '', 'The SCA software assists you in the intuitive use of all OCA instruments — whether making single CA/IFT measurements with the budget-priced OCA 15EC through to the fully automated contact angle and drop shape contour analysis instrument for microstructures, the OCA 40 Micro. If your application dictates that you choose a ''dynamic'' solution; whether using a standard video­ system with 52 images/s sample rate or the ultra high-speed video system with 2200 images/s sample rate the SCA software will guide you through the steps required to maximise the impact of this invaluable ''movie'' footage and the analysed data sets generated\r\n<link 208 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1130, 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, '', '', 0, 0, '', 0),
(1132, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1111, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Module de logiciel SCA', '', 'The SCA software assists you in the intuitive use of all OCA instruments — whether making single CA/IFT measurements with the budget-priced OCA 15EC through to the fully automated contact angle and drop shape contour analysis instrument for microstructures, the OCA 40 Micro. If your application dictates that you choose a ''dynamic'' solution; whether using a standard video­ system with 52 images/s sample rate or the ultra high-speed video system with 2200 images/s sample rate the SCA software will guide you through the steps required to maximise the impact of this invaluable ''movie'' footage and the analysed data sets generated\r\n<link 208 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1130, 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, '', '', 0, 0, '', 0),
(1133, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 219, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Hochgeschwindigkeits- Bildverarbeitung', '', 'Alle OCA-Systeme können mit den entsprechenden Highspeed-Optionen für 1/2&quot;-USB-Kamera UpUSB52H (maximale Bildaufnahmerate: 215 Bilder/s) oder 1/3&quot;-USB-Kamera UpUSB87H (maximale Bildaufnahmerate: 311 Bilder/s) erweitert werden, ausserdem mit dem Hochgeschwindigkeits-Kamerasystem UpHSV1220 (maximale Bildaufnahmerate: 1220 Bilder/s) ausgestattet werden. OCA 20, OCA 35 und OCA 40 können außerdem mit dem Hochgeschwindigkeits- Kamerasystem UpHSV3000 (bis zu 3000 Bilder/s) aufgerüstet werden.\r\n<link 209 - internal-link "Opens internal link in current window">mehr erfahren</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0),
(1134, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1101, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'High-speed video systems', '', 'All instruments within the OCA series can be upgraded to include a high-speed video system. Your sales representative will consider your application(s), together with yourself, to determine the optimum system for your application.\r\n<link 209 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1133, 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, '', '', 0, 0, '', 0),
(1135, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1110, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Systèmes de vidéo de grande vitesse', '', 'All instruments within the OCA series can be upgraded to include a high-speed video system. Your sales representative will consider your application(s), together with yourself, to determine the optimum system for your application.\r\n<link 209 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1133, 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, '', '', 0, 0, '', 0),
(1136, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 220, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Sonderzubehör', '', '<span style="font-family: Arial, Helvetica, sans-serif; line-height: 18px; background-color: rgb(244, 244, 244); ">In enger Zusammenarbeit mit Kunden, Forschungsinstituten und Universitäten haben wir eine Vielzahl an Sonderlösungen als Erweiterung für die OCA-Messgeräte entwickelt. Dies ermöglicht die kontinuierliche Erweiterung des Anwendungs­spek­trums der Grenzflächenanalytik in Labor- und Prozessumgebung.</span>\r\n<p style="margin: 10px 0px 0px; padding: 0px; border: 0px; font-family: Arial, Helvetica, sans-serif; font-stretch: inherit; line-height: 20px; vertical-align: baseline; color: rgb(51, 51, 51); "><span style="margin: 0px; padding: 0px; border: 0px; font-family: inherit; font-size: 14px; font-style: inherit; font-variant: inherit; font-weight: inherit; font-stretch: inherit; line-height: inherit; vertical-align: baseline; color: rgb(153, 0, 0); text-decoration: none; "><link 210 - internal-link "Opens internal link in current window">mehr erfahren</link></span></p>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a22636f6c506f73223b4e3b7d, '', '', 0, 0, '', 0),
(1137, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1100, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Special accessories', '', 'In close cooperation with customers, research institutes, and universities we have developed a wide range of application specific accessories to complement the OCA series of instruments. In doing so we are continuously broadening the range of surface chemistry applications within both the laboratory and process fields.\r\n<link 210 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1136, 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, '', '', 0, 0, '', 0),
(1138, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1109, 1428313737, 1428313713, 1, 1, 1000000000, 'textpic', 'Accessoires spéciaux', '', 'In close cooperation with customers, research institutes, and universities we have developed a wide range of application specific accessories to complement the OCA series of instruments. In doing so we are continuously broadening the range of surface chemistry applications within both the laboratory and process fields.\r\n<link 210 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1136, 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, '', '', 0, 0, '', 0),
(1139, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1040, 1429530555, 1428313830, 1, 0, 1000000000, 'textpic', 'Komponenten und Zubehör', '', 'Zu dieser Produktreihe bieten wir eine Vielzahl an Komponenten und ein umfangreiches Zubehörprogramm an.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Zubehör »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 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, '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1140, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1041, 1429530555, 1428313830, 1, 0, 1000000000, 'textpic', ' OCA accessories', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<b><link 186 - internal-link "Opens internal link in current window">OCA accessories</link></b>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1139, 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, '', '', 0, 0, '', 0),
(1141, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 1042, 1429530555, 1428313830, 1, 0, 1000000000, 'textpic', 'OCA Accessoires', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Accessoires »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1139, 0x613a33333a7b733a353a224354797065223b733a373a2274657874706963223b733a363a22636f6c506f73223b733a313a2232223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a363a22686561646572223b733a32343a224b6f6d706f6e656e74656e20756e64205a75626568c3b672223b733a31333a226865616465725f6c61796f7574223b733a313a2230223b733a31353a226865616465725f706f736974696f6e223b733a303a22223b733a343a2264617465223b733a313a2230223b733a31313a226865616465725f6c696e6b223b733a303a22223b733a383a22626f647974657874223b733a3336323a224c6f72656d20697073756d20646f6c6f722073697420616d65742c20636f6e736574657475722073616469707363696e6720656c6974722c20736564206469616d206e6f6e756d79206569726d6f642074656d706f7220696e766964756e74207574206c61626f726520657420646f6c6f7265206d61676e6120616c69717579616d20657261742c20736564206469616d20766f6c75707475612e204c6f72656d20697073756d20646f6c6f722073697420616d65742c20636f6e736574657475722073616469707363696e6720656c6974722c20736564206469616d206e6f6e756d79206569726d6f642074656d706f7220696e766964756e74207574206c61626f726520657420646f6c6f72652e0d0a3c6c696e6b20313836202d20696e7465726e616c2d6c696e6b20224f70656e7320696e7465726e616c206c696e6b20696e2063757272656e742077696e646f77223e4f4341205a75626568c3b67220c2bb3c2f6c696e6b3e223b733a31313a227274655f656e61626c6564223b733a313a2230223b733a353a22696d616765223b733a313a2231223b733a31303a22696d6167655f7a6f6f6d223b733a313a2230223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a333a22313130223b733a31303a22696d6167657769647468223b733a313a2230223b733a31313a22696d616765686569676874223b733a313a2230223b733a31313a22696d616765626f72646572223b733a313a2230223b733a31373a22696d6167655f636f6d7072657373696f6e223b733a313a2230223b733a31333a22696d6167655f65666665637473223b733a313a2230223b733a31313a22696d6167656f7269656e74223b733a323a223137223b733a393a22696d616765636f6c73223b733a313a2232223b733a31323a22696d6167655f6e6f526f7773223b733a313a2230223b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b733a303a22223b733a363a2268696464656e223b733a313a2230223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a393a22737461727474696d65223b4e3b733a373a22656e6474696d65223b4e3b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1142, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 221, 1428313990, 1428313871, 1, 0, 5248, 'text', 'Advanced dosing technology', '', 'The most often varied accessories, employed to vary a basic instrument configuration, are electronic syringe units in combination with single or multiple dosing units. If you cannot find a solution to your particular dosing challenges here we will happily demonstrate our willingness to further customise —&nbsp; call us&nbsp;<link 238 - internal-link "Opens internal link in current window">directly</link>.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 221, 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, '', '', 0, 0, '', 0),
(1143, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 222, 1429620975, 1428313874, 1, 0, 4992, 'text', 'Electronic multiple dosing system E MD/x', '', 'For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning (±0.05 mm) of the dosing needles. Available in four variants, each with the option of further enhancement to the &quot;plus&quot; version with increased positioning accuracy ( ±0.005 mm):\r\n<ul><li>E-MD/2 and E-MD/2plus for max. two dosing needles.</li><li>E-MD/3 and E-MD/3plus for max. three dosing needles.</li><li>E-MD/4 and E-MD/4plus for max. four dosing needles.</li><li>E-MD/6 and E-MD/6plus for max. six dosing needles.</li></ul>\r\n<h2>Electronic multiple direct dosing system TD-DE/3</h2>\r\nFor use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning of up to three direct dosing modules. Combinable with maximum three electronic syringe modules ES-D/M.\r\n<h2>SIngle-Direct dosing systems SD-Dx</h2>\r\nPaints, varnishes, glues and latex dispersions are typical applications for the&nbsp;SD-DMand&nbsp;SD-DE&nbsp;direct dosing systems. Both direct dosing systems can be used to dose either pendant or sessile liquid drops (on a substrate) through disposable syringes and needles without the need for tubing. The SD-DE is an attachment to the E-MD electronic multiple dosing system, whereas the SD-DM is mounted as a stand-alone system directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and a connection cable\r\n<h2>Double-Direct dosing system DD-DM</h2>\r\nPaints, varnishes, glues and latex dispersions are typical applications for the&nbsp;DD-DMdouble direct dosing system. Thedouble direct dosing system can be used to dose two different sessile liquid drops (on a substrate), for the determination of the surface free energy, through disposable syringes and needles without the need for tubing. The DD-DM is mounted directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of two syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and two connection cables.\r\n<h2>Electronic syringe module ES</h2>\r\nSoftware controlled (automatable) dosing volume (min. 50 nl) and dosing rate (0.06&nbsp;µl/s...26.4 µl/s). Also in a vairant for the direct dosing systems SD-DM, SD-DE and TD-DE available. Easy expandable with the syringe heating device SHD.\r\n<h2>Nanoliter dosing system</h2>\r\nThe software controlled nanoliter dosing system -comprising the pressure control unitPCU Nanodrop&nbsp;and up to four nanodrop dosing valves&nbsp;NDV xx- enables the automatic measurement of wetting properties of microscopic samples using droplets in the nanoliter range (1...500 nl). The dosing valves were mounted with the holderDUN/SD-DM&nbsp;for a direct dosing system or&nbsp;DUN/E-MD&nbsp;for the electronic multiple dosing system E-MD/x plus.\r\n<h2>Picoliter dosing system</h2>\r\nThe innovative picoliter dosing system comprises a dosing unit, picodrop DUP/xx and a Picodrop dispenser head, PDH/xx. It enables the analysis of surfaces in sub-microscopic dimensions by dispensing single droplets of 15 to 400 pl precisely to the pre-defined position.\r\n<h2>Refill and rinse system with liquid pump cleaner RRS-LPC 3/1</h2>\r\nFor the automated, software controlled, refill and rinse of the dosing syringes and for the suction cleaning of a sample onto which liquid has been previously dosed. To be used with the following optional components:\r\n<ul><li>electronic multiple dosing system,&nbsp;E-MD/4.</li><li>3 electronic syringe modules,&nbsp;ES.</li><li>suction cleaning system,&nbsp;LPC OCA.</li><li>Vacuum pump,&nbsp;MVP.</li></ul>\r\n<h2>Dosing syringes DS xx</h2>\r\nSubsequent a selection of the most important dosing syringes:\r\n<ul><li>DS 100/GT&nbsp;gas tight 100 µl syringe.</li><li>DS 500/GT&nbsp;gas tight 500 µl syringe.</li><li>DS 1000/GT&nbsp;gas tight 1000 µl syringe.</li><li>DS-D 1000&nbsp;1000 µl disposable syringe.</li><li>DS-D 1000 SF&nbsp;1000 µl disposable syringe, silicone free.</li></ul>\r\n<h2>Dosing needles SN xx</h2>\r\nOur stock needle options comprises of more than 40 different possibilities, made of stainless steel, hydrophobic coated stainless steel and PTFE arranged in groups according to the standard application of the needle:\r\n<ul><li>SNS xx&nbsp;dosing needles for sessile drop measurements.</li><li>SNP xx&nbsp;dosing needles for pendant drop measurements.</li><li>SNC xx&nbsp;upwards bended dosing needles for captive bubble measurements.</li><li>SNS-D xx&nbsp;disposable dosing needles for sessile drop measurements.</li><li>SNP-D xx&nbsp;disposable dosing needles for pendant drop measurements.</li></ul>\r\n<h2>Dosing tubes DT xx</h2>\r\nIf using a standard dosing system (SD, MD/4, or E-MD/x) the dosing needle is connected, to the syringe, by a stainless steel tLuer Lock adapter,&nbsp;LL A. (The Luer Lock adaptor also serves as a means of securing the position of the needle relative to the dosing system)&nbsp; A dosing tube&nbsp;DT xx&nbsp;(made of PTFE) connects the dosing syringe,DS xx, to the needle.&nbsp;<br />HY-Kit\r\n<h2>Hydrophobisation kit for dosing needles SNS/SNP HY-Kit</h2>\r\nSpecial hydrophobic coating kit to prevent the wetting of the outer surface of the dosing needles.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 222, 0x[encoded data, 10184 characters], '', '', 0, 0, '', 0),
(1144, 205, 0, 0, 0, '', 0, 0, 0, 0, 0, 222, 1429621040, 1428313877, 1, 0, 4928, 'text', 'Electronic multiple dosing system E MD/x', '', '<p style="font-size: 11.9999990463257px; ">For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning (±0.05 mm) of the dosing needles. Available in four variants, each with the option of further enhancement to the &quot;plus&quot; version with increased positioning accuracy ( ±0.005 mm):</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>E-MD/2 and E-MD/2plus for max. two dosing needles.</li><li>E-MD/3 and E-MD/3plus for max. three dosing needles.</li><li>E-MD/4 and E-MD/4plus for max. four dosing needles.</li><li>E-MD/6 and E-MD/6plus for max. six dosing needles.</li></ul>\r\n<h2>Electronic multiple direct dosing system TD-DE/3</h2>\r\n<p style="font-size: 11.9999990463257px; ">For use with the OCA 20, OCA 35, OCA&nbsp;40, and their variants. For the software controlled (automatic) selection and positioning of up to three direct dosing modules. Combinable with maximum three electronic syringe modules ES-D/M.</p>\r\n<h2>SIngle-Direct dosing systems SD-Dx</h2>\r\n<p style="font-size: 11.9999990463257px; ">Paints, varnishes, glues and latex dispersions are typical applications for the&nbsp;SD-DMand&nbsp;SD-DE&nbsp;direct dosing systems. Both direct dosing systems can be used to dose either pendant or sessile liquid drops (on a substrate) through disposable syringes and needles without the need for tubing. The SD-DE is an attachment to the E-MD electronic multiple dosing system, whereas the SD-DM is mounted as a stand-alone system directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and a connection cable</p>\r\n<h2>Double-Direct dosing system DD-DM</h2>\r\n<p style="font-size: 11.9999990463257px; ">Paints, varnishes, glues and latex dispersions are typical applications for the&nbsp;DD-DMdouble direct dosing system. Thedouble direct dosing system can be used to dose two different sessile liquid drops (on a substrate), for the determination of the surface free energy, through disposable syringes and needles without the need for tubing. The DD-DM is mounted directly on the illumination housing of all OCA instruments.&nbsp;<br />Consisting of two syringe module carrier with fine adjusting bolts for the axial and horizontal syringe module positioning and two connection cables.</p>\r\n<h2>Electronic syringe module ES</h2>\r\n<p style="font-size: 11.9999990463257px; ">Software controlled (automatable) dosing volume (min. 50 nl) and dosing rate (0.06&nbsp;µl/s...26.4 µl/s). Also in a vairant for the direct dosing systems SD-DM, SD-DE and TD-DE available. Easy expandable with the syringe heating device SHD.</p>\r\n<h2>Nanoliter dosing system</h2>\r\n<p style="font-size: 11.9999990463257px; ">The software controlled nanoliter dosing system -comprising the pressure control unitPCU Nanodrop&nbsp;and up to four nanodrop dosing valves&nbsp;NDV xx- enables the automatic measurement of wetting properties of microscopic samples using droplets in the nanoliter range (1...500 nl). The dosing valves were mounted with the holderDUN/SD-DM&nbsp;for a direct dosing system or&nbsp;DUN/E-MD&nbsp;for the electronic multiple dosing system E-MD/x plus.</p>\r\n<h2>Picoliter dosing system</h2>\r\n<p style="font-size: 11.9999990463257px; ">The innovative picoliter dosing system comprises a dosing unit, picodrop DUP/xx and a Picodrop dispenser head, PDH/xx. It enables the analysis of surfaces in sub-microscopic dimensions by dispensing single droplets of 15 to 400 pl precisely to the pre-defined position.</p>\r\n<h2>Refill and rinse system with liquid pump cleaner RRS-LPC 3/1</h2>\r\n<p style="font-size: 11.9999990463257px; ">For the automated, software controlled, refill and rinse of the dosing syringes and for the suction cleaning of a sample onto which liquid has been previously dosed. To be used with the following optional components:</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>electronic multiple dosing system,&nbsp;E-MD/4.</li><li>3 electronic syringe modules,&nbsp;ES.</li><li>suction cleaning system,&nbsp;LPC OCA.</li><li>Vacuum pump,&nbsp;MVP.</li></ul>\r\n<h2>Dosing syringes DS xx</h2>\r\n<p style="font-size: 11.9999990463257px; ">Subsequent a selection of the most important dosing syringes:</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>DS 100/GT&nbsp;gas tight 100 µl syringe.</li><li>DS 500/GT&nbsp;gas tight 500 µl syringe.</li><li>DS 1000/GT&nbsp;gas tight 1000 µl syringe.</li><li>DS-D 1000&nbsp;1000 µl disposable syringe.</li><li>DS-D 1000 SF&nbsp;1000 µl disposable syringe, silicone free.</li></ul>\r\n<h2>Dosing needles SN xx</h2>\r\n<p style="font-size: 11.9999990463257px; ">Our stock needle options comprises of more than 40 different possibilities, made of stainless steel, hydrophobic coated stainless steel and PTFE arranged in groups according to the standard application of the needle:</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>SNS xx&nbsp;dosing needles for sessile drop measurements.</li><li>SNP xx&nbsp;dosing needles for pendant drop measurements.</li><li>SNC xx&nbsp;upwards bended dosing needles for captive bubble measurements.</li><li>SNS-D xx&nbsp;disposable dosing needles for sessile drop measurements.</li><li>SNP-D xx&nbsp;disposable dosing needles for pendant drop measurements.</li></ul>\r\n<h2>Dosing tubes DT xx</h2>\r\n<p style="font-size: 11.9999990463257px; ">If using a standard dosing system (SD, MD/4, or E-MD/x) the dosing needle is connected, to the syringe, by a stainless steel tLuer Lock adapter,&nbsp;LL A. (The Luer Lock adaptor also serves as a means of securing the position of the needle relative to the dosing system)&nbsp; A dosing tube&nbsp;DT xx&nbsp;(made of PTFE) connects the dosing syringe,DS xx, to the needle.&nbsp;<br />HY-Kit</p>\r\n<h2>Hydrophobisation kit for dosing needles SNS/SNP HY-Kit</h2>\r\n<p style="font-size: 11.9999990463257px; ">Special hydrophobic coating kit to prevent the wetting of the outer surface of the dosing needles.</p>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 222, 0x[encoded data, 10184 characters], '', '', 0, 0, '', 0),
(1145, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 1115, 1428314578, 1428314573, 1, 0, 64, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1146, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 1116, 1428314578, 1428314573, 1, 0, 32, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1145, 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, '', '', 0, 0, '', 0),
(1147, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 1117, 1428314579, 1428314573, 1, 0, 16, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1145, 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, '', '', 0, 0, '', 0),
(1148, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 1139, 1429530641, 1428314573, 1, 0, 1000000000, 'textpic', 'Komponenten und Zubehör ', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Zubehör »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1149, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 1140, 1429530641, 1428314573, 1, 0, 1000000000, 'textpic', ' OCA accessories', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<b><link 186 - internal-link "Opens internal link in current window">OCA accessories</link></b>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1148, 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, '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1150, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 1141, 1429530641, 1428314573, 1, 0, 1000000000, 'textpic', 'OCA Accessoires', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Accessoires »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1148, 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, '', '', 0, 0, '', 0),
(1151, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 1115, 1428314598, 1428314591, 1, 0, 2304, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1152, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 1116, 1428314598, 1428314591, 1, 0, 2048, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1151, 0x613a31373a7b733a353a224354797065223b733a383a2273686f7274637574223b733a363a22636f6c506f73223b733a313a2232223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a363a22686561646572223b733a31333a22496e666f726d6174696f6e656e223b733a373a227265636f726473223b733a323a223134223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a313a2230223b733a363a2268696464656e223b733a313a2230223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a393a22737461727474696d65223b4e3b733a373a22656e6474696d65223b4e3b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1153, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 1117, 1428314599, 1428314591, 1, 0, 1792, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1151, 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, '', '', 0, 0, '', 0),
(1154, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 1139, 1429530668, 1428314591, 1, 0, 1000000000, 'textpic', 'Komponenten und Zubehör ', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Zubehör »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1155, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 1140, 1429530668, 1428314591, 1, 0, 1000000000, 'textpic', ' OCA accessories', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<b><link 186 - internal-link "Opens internal link in current window">OCA accessories</link></b>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1154, 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, '', '', 0, 0, '', 0),
(1156, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 1141, 1429530668, 1428314591, 1, 0, 1000000000, 'textpic', 'OCA Accessoires', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Accessoires »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1154, 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, '', '', 0, 0, '', 0),
(1157, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 1115, 1428314616, 1428314610, 1, 0, 64, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1158, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 1116, 1428314617, 1428314610, 1, 0, 32, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1157, 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, '', '', 0, 0, '', 0),
(1159, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 1117, 1428314618, 1428314610, 1, 0, 16, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1157, 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, '', '', 0, 0, '', 0),
(1160, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 1139, 1429530673, 1428314610, 1, 0, 1000000000, 'textpic', 'Komponenten und Zubehör ', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Zubehör »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1161, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 1140, 1429530673, 1428314610, 1, 0, 1000000000, 'textpic', ' OCA accessories', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<b><link 186 - internal-link "Opens internal link in current window">OCA accessories</link></b>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1160, 0x613a33333a7b733a353a224354797065223b733a373a2274657874706963223b733a363a22636f6c506f73223b733a313a2232223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a363a22686561646572223b733a32343a224b6f6d706f6e656e74656e20756e64205a75626568c3b672223b733a31333a226865616465725f6c61796f7574223b733a313a2230223b733a31353a226865616465725f706f736974696f6e223b733a303a22223b733a343a2264617465223b733a313a2230223b733a31313a226865616465725f6c696e6b223b733a303a22223b733a383a22626f647974657874223b733a3336323a224c6f72656d20697073756d20646f6c6f722073697420616d65742c20636f6e736574657475722073616469707363696e6720656c6974722c20736564206469616d206e6f6e756d79206569726d6f642074656d706f7220696e766964756e74207574206c61626f726520657420646f6c6f7265206d61676e6120616c69717579616d20657261742c20736564206469616d20766f6c75707475612e204c6f72656d20697073756d20646f6c6f722073697420616d65742c20636f6e736574657475722073616469707363696e6720656c6974722c20736564206469616d206e6f6e756d79206569726d6f642074656d706f7220696e766964756e74207574206c61626f726520657420646f6c6f72652e0d0a3c6c696e6b20313836202d20696e7465726e616c2d6c696e6b20224f70656e7320696e7465726e616c206c696e6b20696e2063757272656e742077696e646f77223e4f4341205a75626568c3b67220c2bb3c2f6c696e6b3e223b733a31313a227274655f656e61626c6564223b733a313a2230223b733a353a22696d616765223b733a313a2231223b733a31303a22696d6167655f7a6f6f6d223b733a313a2230223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a333a22313130223b733a31303a22696d6167657769647468223b733a313a2230223b733a31313a22696d616765686569676874223b733a313a2230223b733a31313a22696d616765626f72646572223b733a313a2230223b733a31373a22696d6167655f636f6d7072657373696f6e223b733a313a2230223b733a31333a22696d6167655f65666665637473223b733a313a2230223b733a31313a22696d6167656f7269656e74223b733a323a223137223b733a393a22696d616765636f6c73223b733a313a2232223b733a31323a22696d6167655f6e6f526f7773223b733a313a2230223b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b733a303a22223b733a363a2268696464656e223b733a313a2230223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a393a22737461727474696d65223b4e3b733a373a22656e6474696d65223b4e3b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1162, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 1141, 1429530673, 1428314610, 1, 0, 1000000000, 'textpic', 'OCA Accessoires', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Accessoires »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1160, 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, '', '', 0, 0, '', 0),
(1163, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 1115, 1428314635, 1428314632, 1, 0, 64, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1164, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 1116, 1428314636, 1428314632, 1, 0, 32, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1163, 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, '', '', 0, 0, '', 0),
(1165, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 1117, 1428314636, 1428314632, 1, 0, 16, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1163, 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, '', '', 0, 0, '', 0),
(1166, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 1139, 1429530682, 1428314632, 1, 0, 1000000000, 'textpic', 'Komponenten und Zubehör ', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Zubehör »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1167, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 1140, 1429530682, 1428314632, 1, 0, 1000000000, 'textpic', ' OCA accessories', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<b><link 186 - internal-link "Opens internal link in current window">OCA accessories</link></b>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1166, 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, '', '', 0, 0, '', 0),
(1168, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 1141, 1429530682, 1428314632, 1, 0, 1000000000, 'textpic', 'OCA Accessoires', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Accessoires »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1166, 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, '', '', 0, 0, '', 0),
(1169, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 1115, 1428314650, 1428314645, 1, 0, 64, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1170, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 1116, 1428314650, 1428314645, 1, 0, 32, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1169, 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, '', '', 0, 0, '', 0),
(1171, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 1117, 1428314651, 1428314645, 1, 0, 16, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1169, 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, '', '', 0, 0, '', 0),
(1172, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 1139, 1429530694, 1428314645, 1, 0, 1000000000, 'textpic', 'Komponenten und Zubehör ', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Zubehör »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1173, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 1140, 1429530694, 1428314645, 1, 0, 1000000000, 'textpic', ' OCA accessories', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<b><link 186 - internal-link "Opens internal link in current window">OCA accessories</link></b>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1172, 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, '', '', 0, 0, '', 0),
(1174, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 1141, 1429530694, 1428314645, 1, 0, 1000000000, 'textpic', 'OCA Accessoires', '', 'Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore magna aliquyam erat, sed diam voluptua. Lorem ipsum dolor sit amet, consetetur sadipscing elitr, sed diam nonumy eirmod tempor invidunt ut labore et dolore.\r\n<link 186 - internal-link "Opens internal link in current window">OCA Accessoires »</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 1, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1172, 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, '', '', 0, 0, '', 0),
(1175, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 226, 1428314705, 1428314673, 1, 0, 384, 'text', 'Accessories for the control of temperature and the measurement environment', '', 'Dataphysics offer a wide range of temperature and environmental control systems that operate in conjunction with the OCA instruments and supportive software. These systems enable the measurement of surface and interfacial parameters across an incredibly diverse group of applications, at temperatures from -30 to 1800 °C and from a high-vacuum to high-pressure atmosphere. Temperature, pressure, humidity, and other physical parameters can be developed, directly, by these OCA instrument accessories, or with the integration of additional hardware.&nbsp;', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 226, 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, '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1176, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 227, 1429621289, 1428314676, 1, 0, 192, 'text', 'SHD', '', 'Device for the heating of gastight syringe DS xx/GT mounted in the electronic syringe module ES. For use in conjunction with a liquid circulator.\r\n<h2>TFC 100</h2>\r\nLiquid temperature control unit for measurement within the range - 10 to 130 °C. Requires an external liquid circulator bath and includes the pipe work etc. to allow for a protective atmosphere (inert gas inlet) to surround the sample working area. The triple diffuser, TDI 100, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas flow towards the chamber windows.&nbsp;<br />An optional humidity sensor expands the applicability of the TFC 100 even further.\r\n<h2>TPC 150</h2>\r\nTemperature control chamber with a peltier system for the software controlled temperature setting from -30 t0 160 °C (heat up and cool down rate ±1 K/s). Also includes the pipe work etc. to allow for a protective atmosphere to surround the working area. The triple diffuser, TDI 150, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas towards the chamber windows.\r\nAn optional humidity sensor expands the applicability of the TPC 150 even further.\r\n<h2>TEC 300XL</h2>\r\nFor mounting to the XL-variants of the OCA measuring devices. Temperature control chamber with electric resistance heater for manual controlled temperature setting up to 300 °C. Temperature display on controller and in SCA-software. Work under a protective atmosphere is also facilitated via the inert gas inlet.\r\n<h2>TEC 400</h2>\r\nTemperature control chamber with twin electric resistance heater (above and below the sample plate) for software controlled temperature setting up to 400 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.\r\n<h2>TEC 700</h2>\r\nTemperature control chambers with twin electric resistance heater (above and below the sample plate) for software control of temperature setting up to 700 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.\r\n<h2>NHD 400 / NHD 700</h2>\r\nThe NHD 400 and NHD 700 are electrical needle heating devices for generating drops at temperatures up 400 °C and 700 °C respectively (Applications areas include: polymers and hot melts, molten metals, etc.). Also extremely tenacious copolymers, with viscosities in the kPas range, e.g. with elastomer constituents, can be dosed easily at these elevated temperatures.\r\n<h2>MTFQ 1200</h2>\r\nSpecial high temperature furnace for measurements to a maximum of 1200°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the MTFQ 1200 also.\r\n<h2>HTFC 1500/1700</h2>\r\nSpecial high temperature furnace for measurements to a maximum of 1500/1700°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the HTFQ 1500/1700 also.\r\n<h2>HTFC 1500/1800HV</h2>\r\nElectrically heated, high temperature, high vacuum tube furnace for use in combination with an OCA 15LHT-SV.&nbsp;\r\n<ul><li>Max. temperature: 1500/1800 °C</li><li>Max. temperature for operation under vacuum: 1450°C</li><li>Max. vacuum: 10<sup>-5</sup>&nbsp;mbar</li></ul>\r\n<br />vacuum tube:\r\nMaterial: C799<br />Outer diameter: 50 mm<br />Inner diameter: 40 mm<br />Length: 990 mm<br />-&nbsp;&nbsp;&nbsp; Prepared for horizontal operation&nbsp;<br />-&nbsp;&nbsp;&nbsp; Observation windows on both sides as additional equipment for gastight flanges<br />-&nbsp;&nbsp;&nbsp; MoSi2 heating elements, mounted vertically for easy replacement<br />-&nbsp;&nbsp;&nbsp; Insulation with vacuum-formed ceramic fibre plates for an energy-saving operation and fast heating up times<br />-&nbsp;&nbsp;&nbsp; Rectangular outer casing with slots for convection cooling<br />-&nbsp;&nbsp;&nbsp; Casing made of sheets of textured stainless steel (non-rusting design)<br />-&nbsp;&nbsp;&nbsp; Type B thermocouple<br />-&nbsp;&nbsp;&nbsp; Power unit with low-voltage transformer and thyristor controller&nbsp;<br />Typically used for the measurement of contact angle and surface tension of materials like:\r\n<div class="indent"><ul><li>Compound Materials</li><li>Steel</li><li>Alloys</li><li>Molten Glass</li></ul></div>\r\n\r\n<h2>PMC 50S</h2>\r\nPressure cell with two viewing windows and manipulation system to allow the exchange of the sample table. Facilitates the evaluation of contact angles according to the sessile and captive drop method, the measurement of surface tensions in liquid/pressurized gas systems and the interfacial tensions in liquid/liquid systems.&nbsp;\r\nIncorporates four dosing needles, for separate sessile and pendant drop measurements via four different injection liquids and a two phase pressure supply system for an external supply of the gas or liquid phases. The first pressure level is used for controlling the pressure of the primary filling liquid or gas. The higher, second pressure level is necessary for the injection dosing of the measuring liquid(s). The primary&nbsp; phase acts also a heat transfer medium.&nbsp;\r\n<ul><li>pressure range: 0.03 ... 50 bar</li><li>max. temperature: 80 °C</li></ul>\r\n<h2>HGC 20EC/20/30</h2>\r\nFor the automated regulation of the humidity of ambient clean air, nitrogen, argon, or any other inert gas in temperature controlled measuring chambers.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 227, 0x[encoded data, 8700 characters], '', '', 0, 0, '', 0),
(1177, 206, 0, 0, 0, '', 0, 0, 0, 0, 0, 227, 1429621276, 1428314678, 1, 0, 160, 'text', 'SHD', '', '<p style="font-size: 11.9999990463257px; ">Device for the heating of gastight syringe DS xx/GT mounted in the electronic syringe module ES. For use in conjunction with a liquid circulator.</p>\r\n<h2>TFC 100</h2>\r\n<p style="font-size: 11.9999990463257px; ">Liquid temperature control unit for measurement within the range - 10 to 130 °C. Requires an external liquid circulator bath and includes the pipe work etc. to allow for a protective atmosphere (inert gas inlet) to surround the sample working area. The triple diffuser, TDI 100, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas flow towards the chamber windows.&nbsp;<br />An optional humidity sensor expands the applicability of the TFC 100 even further.</p>\r\n<h2>TPC 150</h2>\r\n<p style="font-size: 11.9999990463257px; ">Temperature control chamber with a peltier system for the software controlled temperature setting from -30 t0 160 °C (heat up and cool down rate ±1 K/s). Also includes the pipe work etc. to allow for a protective atmosphere to surround the working area. The triple diffuser, TDI 150, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas towards the chamber windows.</p>\r\n<p style="font-size: 11.9999990463257px; ">An optional humidity sensor expands the applicability of the TPC 150 even further.</p>\r\n<h2>TEC 300XL</h2>\r\n<p style="font-size: 11.9999990463257px; ">For mounting to the XL-variants of the OCA measuring devices. Temperature control chamber with electric resistance heater for manual controlled temperature setting up to 300 °C. Temperature display on controller and in SCA-software. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>\r\n<h2>TEC 400</h2>\r\n<p style="font-size: 11.9999990463257px; ">Temperature control chamber with twin electric resistance heater (above and below the sample plate) for software controlled temperature setting up to 400 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>\r\n<h2>TEC 700</h2>\r\n<p style="font-size: 11.9999990463257px; ">Temperature control chambers with twin electric resistance heater (above and below the sample plate) for software control of temperature setting up to 700 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>\r\n<h2>NHD 400 / NHD 700</h2>\r\n<p style="font-size: 11.9999990463257px; ">The NHD 400 and NHD 700 are electrical needle heating devices for generating drops at temperatures up 400 °C and 700 °C respectively (Applications areas include: polymers and hot melts, molten metals, etc.). Also extremely tenacious copolymers, with viscosities in the kPas range, e.g. with elastomer constituents, can be dosed easily at these elevated temperatures.</p>\r\n<h2>MTFQ 1200</h2>\r\n<p style="font-size: 11.9999990463257px; ">Special high temperature furnace for measurements to a maximum of 1200°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the MTFQ 1200 also.</p>\r\n<h2>HTFC 1500/1700</h2>\r\n<p style="font-size: 11.9999990463257px; ">Special high temperature furnace for measurements to a maximum of 1500/1700°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the HTFQ 1500/1700 also.</p>\r\n<h2>HTFC 1500/1800HV</h2>\r\n<p style="font-size: 11.9999990463257px; ">Electrically heated, high temperature, high vacuum tube furnace for use in combination with an OCA 15LHT-SV.&nbsp;</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>Max. temperature: 1500/1800 °C</li><li>Max. temperature for operation under vacuum: 1450°C</li><li>Max. vacuum: 10<sup>-5</sup>&nbsp;mbar</li></ul>\r\n<p style="font-size: 11.9999990463257px; "><br />vacuum tube:</p>\r\n<p style="font-size: 11.9999990463257px; ">Material: C799<br />Outer diameter: 50 mm<br />Inner diameter: 40 mm<br />Length: 990 mm<br />-&nbsp;&nbsp;&nbsp; Prepared for horizontal operation&nbsp;<br />-&nbsp;&nbsp;&nbsp; Observation windows on both sides as additional equipment for gastight flanges<br />-&nbsp;&nbsp;&nbsp; MoSi2 heating elements, mounted vertically for easy replacement<br />-&nbsp;&nbsp;&nbsp; Insulation with vacuum-formed ceramic fibre plates for an energy-saving operation and fast heating up times<br />-&nbsp;&nbsp;&nbsp; Rectangular outer casing with slots for convection cooling<br />-&nbsp;&nbsp;&nbsp; Casing made of sheets of textured stainless steel (non-rusting design)<br />-&nbsp;&nbsp;&nbsp; Type B thermocouple<br />-&nbsp;&nbsp;&nbsp; Power unit with low-voltage transformer and thyristor controller&nbsp;<br />Typically used for the measurement of contact angle and surface tension of materials like:</p>\r\n<div class="indent" style="font-size: 11.9999990463257px; "><ul><li>Compound Materials</li><li>Steel</li><li>Alloys</li><li>Molten Glass</li></ul></div>\r\n<p style="font-size: 11.9999990463257px; ">&nbsp;</p>\r\n<h2>PMC 50S</h2>\r\n<p style="font-size: 11.9999990463257px; ">Pressure cell with two viewing windows and manipulation system to allow the exchange of the sample table. Facilitates the evaluation of contact angles according to the sessile and captive drop method, the measurement of surface tensions in liquid/pressurized gas systems and the interfacial tensions in liquid/liquid systems.&nbsp;</p>\r\n<p style="font-size: 11.9999990463257px; ">Incorporates four dosing needles, for separate sessile and pendant drop measurements via four different injection liquids and a two phase pressure supply system for an external supply of the gas or liquid phases. The first pressure level is used for controlling the pressure of the primary filling liquid or gas. The higher, second pressure level is necessary for the injection dosing of the measuring liquid(s). The primary&nbsp; phase acts also a heat transfer medium.&nbsp;</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>pressure range: 0.03 ... 50 bar</li><li>max. temperature: 80 °C</li></ul>\r\n<h2>HGC 20EC/20/30</h2>\r\n<p style="font-size: 11.9999990463257px; ">For the automated regulation of the humidity of ambient clean air, nitrogen, argon, or any other inert gas in temperature controlled measuring chambers.</p>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 227, 0x[encoded data, 8700 characters], '', '', 0, 0, '', 0),
(1178, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 230, 1428314967, 1428314863, 1, 0, 3584, 'text', 'Sample positioning and securing systems', '', 'The OCA range of accessories includes several sample positioning and securing units. In the unlikely event that we cannot identify a unit that will readily address your application, we should be able to adapt a device to facilitate your special requirements.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 230, 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, '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1179, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 231, 1429621409, 1428314866, 1, 0, 3072, 'text', 'Electronic tilting base unit TBU 90E', '', '<h2>Electronic tilting base unit TBU 90E</h2>\r\nThe tilting base unit, TBU 90E, is an important instrument extension for the analysis of hydrophobic materials. It allows the measurement of the advancing and receding contact angle, the rolling angle and the contact angle hysteresis of sessile drops while maintaining a constant visual plane (optics / sample). In conjunction with the SCA software; it facilitates the fully automated execution of rolling angle experiments up to a tilt angle of 90°and at freely adjustable tilt speed of 0.03 to 2.8 °/s.\r\n<h2>Manual tilting base assemblies TBA 60M</h2>\r\nTo measure advancing and receding contact angles, as well as the contact angle hysteresis of sessile drops, with a tilting range of 0 to 70°.&nbsp;\r\n<h2>Electronic turn table with vacuum fixation ETT/VAC</h2>\r\nFor the mapping of wafers or for the automatic analysis of the surface free energy of flat samples such as paint test plates or optical lenses. The ETT/VAC can be equipped with any of a range of different top plates. The fixation of the samples on the top plates is achieved by the application of a vacuum, connected to the turn table. Below a selection out of the wide range of top plates:\r\n<ul><li><strong>WTP 6/VAC</strong>&nbsp; for 6&quot; Wafer.</li><li><strong>WTP 8/VAC</strong>&nbsp; for 8&quot; Wafer</li><li><strong>WTP 12/VAC</strong>&nbsp; for 12&quot; Wafer.</li><li><strong>STP 76-26-8</strong>&nbsp;for 8 samples with 76x26x2 mm.</li><li><strong>STP 127x85-2</strong>&nbsp;for 2 samples with 127x85 mm.</li><li><strong>STP 130x100-4</strong>&nbsp;for 4 samples with61x46x0.6 mm.</li><li><strong>STP 200-5</strong>&nbsp;for 5 convex and planar lenses.</li></ul>\r\n\r\n<h2>Manual turn table MTT</h2>\r\nFor the optional mounting of 2&quot;-, 3&quot;-, 4&quot;-, 5&quot;-, 6&quot;-, 8&quot;,-&nbsp; 12&quot;-wafer top plates (<b>WTP xxM</b>)&nbsp; or customer specific sample top plate (<b>STP xxxM</b>&nbsp;or&nbsp;<b>SQT xxxM</b>).\r\n<h2>Film and paperholder FSH 30 and FSC 80/150</h2>\r\n<ul><li><strong>FSH 30</strong>&nbsp;special holder, made of delrin, for the contact angle measurement on thin films. Consisting of&nbsp; base (outer-ø=30 mm) and three rings (inner-ø= 30.0 / 30.1 / 30.2 mm).</li><li><strong>FSC 80</strong>&nbsp;special holder for the contact angle measurement on thin films or paper;&nbsp;<br />Maximum sample size 80 x 80 x 3 mm.</li></ul>\r\n<h2>Holder for single fibres FHO 40plus</h2>\r\nHolder for Single Fibres. Unconstrained by degree of fibre thickness, adjustable to match fibres length and apply required tension.- min. fibre length 25 mm, measurable fibre length 70 mm.\r\n- adjustable X- and Z-Axis.\r\nConsisting of base plate and adjustable fibre fixation.FHM 100\r\n<h2>Film or foil sample stage FHM 100</h2>\r\nSpecial sample stage with curvature for instruments of the OCA series for a convenient fixation of films or foils by magnet bars.<br />Size: 100 x 100 mm\r\n<h2>Suction plate SP 100</h2>\r\nFor holding thin flexible samples flat against the stage surface, with adjustable suction area and fitting for vacuum pump.<br />Size: 100 x 100 mm\r\n<h2>Sample vessels SVL xx and test piecesTPL xx</h2>\r\nTest pieces for use with the lamella method:\r\n<ul><li><strong>TPL - 3.0</strong>&nbsp; sphere-ø = 3.0 mm.</li><li><strong>TPL - 3.5</strong>&nbsp; sphere-ø = 3.5 mm.</li><li><strong>TPL - 4.0</strong>&nbsp; sphere-ø = 4.0 mm.</li></ul>\r\n&nbsp;<br />Sample vessel for use with the lamella method:\r\n<ul><li><strong>SVL 12/30</strong>&nbsp;height = 12, inner-ø= 30 mm.</li><li><strong>SVL 12/50</strong>&nbsp;height = 12, inner-ø= 50 mm.</li><li><strong>SVL 12/70</strong>&nbsp;height= 12, inner-ø= 70 mm.</li><li><strong>SVL 24/30</strong>&nbsp;height= 24, inner-ø= 30 mm.</li></ul>\r\n<h2>Glass cell GC xx</h2>\r\nGlass cell (cuvette) made of special optical glass for measuring the interfacial tension between two fluids according to the Pendant Drop method or the contact angle of a fluid drop or a gas bubble in a fluid (e.g. Schultz method) ; up to 70° C.<br />Also available in ''high'' temperature version (GC xxHT) and made of Quatrz glass (GC xxQ); 70 to 350° C.\r\n<ul><li><strong>GC 10</strong>&nbsp;glass cell 20 x 20 x 20 mm.</li><li><strong>GC 20</strong>&nbsp;glass cell 20 x 40 x 20 mm.</li><li><strong>GC 40</strong>&nbsp;glass cell 40x 40 x 40 mm.</li><li><strong>GC 60</strong>&nbsp;glass cell 60x 60 x 40 mm.</li></ul>\r\n<h2>Contact lens support CL 20</h2>\r\nContact lens support for the measurement of captive bubble contact angles on soft and hard contact lenses in liquids. Used in combination with captive bubble equipment (glass cell GC 20, needle SNC 052/026).\r\n<h2>Soft contact lens support SCL 21</h2>\r\nHolder for soft contact lenses with adjustable inlet pieces with 3 different radii of curvature to attach the contact lens without deformation of the shape. Used in combination with captive bubble equipment (glass cell GC 20, needle SNC 052/026).\r\n<h2>Sample holder for captive bubble measurements SHC 20</h2>\r\nHolder for flat samples, prior to measurement by the captive bubble method, when deformation of sample might otherwise be a problem when submerged. Used in combination with captive bubble equipment (glass cell GC 20, needle SNC 052/026).', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 231, 0x[encoded data, 10802 characters], '', '', 0, 0, '', 0),
(1180, 207, 0, 0, 0, '', 0, 0, 0, 0, 0, 231, 1429621425, 1428314869, 1, 0, 2816, 'text', 'Electronic tilting base unit TBU 90E', '', '<h2>Electronic tilting base unit TBU 90E</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">The tilting base unit, TBU 90E, is an important instrument extension for the analysis of hydrophobic materials. It allows the measurement of the advancing and receding contact angle, the rolling angle and the contact angle hysteresis of sessile drops while maintaining a constant visual plane (optics / sample). In conjunction with the SCA software; it facilitates the fully automated execution of rolling angle experiments up to a tilt angle of 90°and at freely adjustable tilt speed of 0.03 to 2.8 °/s.</p></h2>\r\n<h2>Manual tilting base assemblies TBA 60M</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">To measure advancing and receding contact angles, as well as the contact angle hysteresis of sessile drops, with a tilting range of 0 to 70°.&nbsp;</p></h2>\r\n<h2>Electronic turn table with vacuum fixation ETT/VAC</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">For the mapping of wafers or for the automatic analysis of the surface free energy of flat samples such as paint test plates or optical lenses. The ETT/VAC can be equipped with any of a range of different top plates. The fixation of the samples on the top plates is achieved by the application of a vacuum, connected to the turn table. Below a selection out of the wide range of top plates:</p><ul style="font-size: 11.9999990463257px; font-weight: normal; "><li><strong>WTP 6/VAC</strong>&nbsp; for 6&quot; Wafer.</li><li><strong>WTP 8/VAC</strong>&nbsp; for 8&quot; Wafer</li><li><strong>WTP 12/VAC</strong>&nbsp; for 12&quot; Wafer.</li><li><strong>STP 76-26-8</strong>&nbsp;for 8 samples with 76x26x2 mm.</li><li><strong>STP 127x85-2</strong>&nbsp;for 2 samples with 127x85 mm.</li><li><strong>STP 130x100-4</strong>&nbsp;for 4 samples with61x46x0.6 mm.</li><li><strong>STP 200-5</strong>&nbsp;for 5 convex and planar lenses.</li></ul><p style="font-size: 11.9999990463257px; font-weight: normal; ">&nbsp;</p></h2>\r\n<h2>Manual turn table MTT</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">For the optional mounting of 2&quot;-, 3&quot;-, 4&quot;-, 5&quot;-, 6&quot;-, 8&quot;,-&nbsp; 12&quot;-wafer top plates (<strong>WTP xxM</strong>)&nbsp; or customer specific sample top plate (<strong>STP xxxM</strong>&nbsp;or&nbsp;<strong>SQT xxxM</strong>).</p></h2>\r\n<h2>Film and paperholder FSH 30 and FSC 80/150</h2>\r\n<h2><ul style="font-size: 11.9999990463257px; font-weight: normal; "><li><strong>FSH 30</strong>&nbsp;special holder, made of delrin, for the contact angle measurement on thin films. Consisting of&nbsp; base (outer-ø=30 mm) and three rings (inner-ø= 30.0 / 30.1 / 30.2 mm).</li><li><strong>FSC 80</strong>&nbsp;special holder for the contact angle measurement on thin films or paper;&nbsp;<br />Maximum sample size 80 x 80 x 3 mm.</li></ul></h2>\r\n<h2>Holder for single fibres FHO 40plus</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Holder for Single Fibres. Unconstrained by degree of fibre thickness, adjustable to match fibres length and apply required tension.- min. fibre length 25 mm, measurable fibre length 70 mm.</p><p style="font-size: 11.9999990463257px; font-weight: normal; ">- adjustable X- and Z-Axis.</p><p style="font-size: 11.9999990463257px; font-weight: normal; ">Consisting of base plate and adjustable fibre fixation.FHM 100</p></h2>\r\n<h2>Film or foil sample stage FHM 100</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Special sample stage with curvature for instruments of the OCA series for a convenient fixation of films or foils by magnet bars.<br />Size: 100 x 100 mm</p></h2>\r\n<h2>Suction plate SP 100</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">For holding thin flexible samples flat against the stage surface, with adjustable suction area and fitting for vacuum pump.<br />Size: 100 x 100 mm</p></h2>\r\n<h2>Sample vessels SVL xx and test piecesTPL xx</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Test pieces for use with the lamella method:</p><ul style="font-size: 11.9999990463257px; font-weight: normal; "><li><strong>TPL - 3.0</strong>&nbsp; sphere-ø = 3.0 mm.</li><li><strong>TPL - 3.5</strong>&nbsp; sphere-ø = 3.5 mm.</li><li><strong>TPL - 4.0</strong>&nbsp; sphere-ø = 4.0 mm.</li></ul><p style="font-size: 11.9999990463257px; font-weight: normal; ">&nbsp;<br />Sample vessel for use with the lamella method:</p><ul style="font-size: 11.9999990463257px; font-weight: normal; "><li><strong>SVL 12/30</strong>&nbsp;height = 12, inner-ø= 30 mm.</li><li><strong>SVL 12/50</strong>&nbsp;height = 12, inner-ø= 50 mm.</li><li><strong>SVL 12/70</strong>&nbsp;height= 12, inner-ø= 70 mm.</li><li><strong>SVL 24/30</strong>&nbsp;height= 24, inner-ø= 30 mm.</li></ul></h2>\r\n<h2>Glass cell GC xx</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Glass cell (cuvette) made of special optical glass for measuring the interfacial tension between two fluids according to the Pendant Drop method or the contact angle of a fluid drop or a gas bubble in a fluid (e.g. Schultz method) ; up to 70° C.<br />Also available in ''high'' temperature version (GC xxHT) and made of Quatrz glass (GC xxQ); 70 to 350° C.</p><ul style="font-size: 11.9999990463257px; font-weight: normal; "><li><strong>GC 10</strong>&nbsp;glass cell 20 x 20 x 20 mm.</li><li><strong>GC 20</strong>&nbsp;glass cell 20 x 40 x 20 mm.</li><li><strong>GC 40</strong>&nbsp;glass cell 40x 40 x 40 mm.</li><li><strong>GC 60</strong>&nbsp;glass cell 60x 60 x 40 mm.</li></ul></h2>\r\n<h2>Contact lens support CL 20</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Contact lens support for the measurement of captive bubble contact angles on soft and hard contact lenses in liquids. Used in combination with captive bubble equipment (glass cell GC 20, needle SNC 052/026).</p></h2>\r\n<h2>Soft contact lens support SCL 21</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Holder for soft contact lenses with adjustable inlet pieces with 3 different radii of curvature to attach the contact lens without deformation of the shape. Used in combination with captive bubble equipment (glass cell GC 20, needle SNC 052/026).</p></h2>\r\n<h2>Sample holder for captive bubble measurements SHC 20</h2>\r\n<h2><p style="font-size: 11.9999990463257px; font-weight: normal; ">Holder for flat samples, prior to measurement by the captive bubble method, when deformation of sample might otherwise be a problem when submerged. Used in combination with captive bubble equipment (glass cell GC 20, needle SNC 052/026).</p></h2>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 231, 0x[encoded data, 10802 characters], '', '', 0, 0, '', 0),
(1181, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 234, 1428315121, 1428314981, 1, 0, 384, 'text', 'Software Module', '', 'The SCA software assists you in the intuitive use of all OCA instruments — whether making single CA/IFT measurements with the budget-priced OCA 15EC through to the fully automated contact angle and drop shape contour analysis instrument for microstructures, the OCA 40 Micro. If your application dictates that you choose a ''dynamic'' solution; whether using a standard video­ system with 52 images/s sample rate or the ultra high-speed video system with 2200 images/s sample rate the SCA software will guide you through the steps required to maximise the impact of this invaluable ''movie'' footage and the analysed data sets generated. DataPhysics has specialised in the development of highly precise and reliable methods for evaluating drop contours in combination with statistical error analysis. At every step of hardware and software development the quality of the data produced by its combination has been a crucial consideration.\r\n\r\nThe SCA software is designed as a modular program. Under Windows 7® / Vista® it works in 32- and in 64-bit mode; under Windows XP® only in 32-bit mode. The available software modules are:\r\nSCA 20 — contact angle.\r\nSCA 21 ­— surface free energy.\r\nSCA 22 — pendant drop.\r\nSCA 23 — lamella contour.\r\nSCA 24 — drop on fibre.\r\nSCA 26 — oscillation / relaxation.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 234, 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, '', '', 0, 0, '', 0),
(1182, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 235, 1429621487, 1428314983, 1, 0, 192, 'text', 'SCA 20 — contact angle', '', '<ul><li>Control of electronic syringe modules, tilting and turn tables and all other software controlable modules within the OCA accessory range.</li><li>Video based measurement and presentation of the static and dynamic contact angle according to the “Sessile &amp; Captive Drop-method” on plan, convex, and concave surfaces.</li><li>Automated measurement of the contact angle hysteresis.</li><li>Recording of movie sequences.</li><li>Statistics and measurement error analysis.</li><li>Gas, liquid, and solids database with currently more than 170 records for all surface energy analysis methods including related citations.</li></ul>\r\nSCA 21 Analysis of the wetting properties of printing ink\r\n<h2>SCA 21 — surface free energy</h2>\r\n<ul><li>Analysis of the surface free energy of solids, as well as the contributing components (e.g. dispersive, polar and hydrogen bond parts, acid and base portions, respectively) according to nine different theories.</li><li>Representation of wetting envelopes and work of adhesion/contact angle diagrams.</li></ul>\r\n<h2>SCA 22 — pendant drop</h2>\r\n<ul><li>Analysis of surface and interfacial tension, as well as the polar and dispersive contributions, based on the analysis of the drop shape.</li></ul>\r\n<h2>SCA 23 — lamella contour</h2>\r\n<ul><li>Analysis of surface and interfacial tension based on the evaluation of the lamella contour.</li></ul>\r\n<h2>SCA 24 — drop on fibre</h2>\r\n<ul><li>Analysis of the static contact angle according to the generalized length-width-method for the drop-on-fibre or wetted fibre setup and the dynamic contact angle according to the lamella-ring-method.</li></ul>\r\n<h2>SCA 26 — oscillating/relaxing drop</h2>\r\n<ul><li>Analysis of the real and imaginary part of the interfacial dilatational modulus based on the oscillating or relaxing contour of pendant drops.</li></ul>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '2', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 235, 0x[encoded data, 4610 characters], '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1183, 208, 0, 0, 0, '', 0, 0, 0, 0, 0, 235, 1429621505, 1428314986, 1, 0, 160, 'text', 'CA 20 — contact angle', '', '<ul><li>Control of electronic syringe modules, tilting and turn tables and all other software controlable modules within the OCA accessory range.</li><li>Video based measurement and presentation of the static and dynamic contact angle according to the “Sessile &amp; Captive Drop-method” on plan, convex, and concave surfaces.</li><li>Automated measurement of the contact angle hysteresis.</li><li>Recording of movie sequences.</li><li>Statistics and measurement error analysis.</li><li>Gas, liquid, and solids database with currently more than 170 records for all surface energy analysis methods including related citations.</li><p style="font-size: 11.9999990463257px; ">SCA 21 Analysis of the wetting properties of printing ink</p><h2>SCA 21 — surface free energy</h2><ul style="font-size: 11.9999990463257px; "><li>Analysis of the surface free energy of solids, as well as the contributing components (e.g. dispersive, polar and hydrogen bond parts, acid and base portions, respectively) according to nine different theories.</li><li>Representation of wetting envelopes and work of adhesion/contact angle diagrams.</li></ul><h2>SCA 22 — pendant drop</h2><ul style="font-size: 11.9999990463257px; "><li>Analysis of surface and interfacial tension, as well as the polar and dispersive contributions, based on the analysis of the drop shape.</li></ul><h2>SCA 23 — lamella contour</h2><ul style="font-size: 11.9999990463257px; "><li>Analysis of surface and interfacial tension based on the evaluation of the lamella contour.</li></ul><h2>SCA 24 — drop on fibre</h2><ul style="font-size: 11.9999990463257px; "><li>Analysis of the static contact angle according to the generalized length-width-method for the drop-on-fibre or wetted fibre setup and the dynamic contact angle according to the lamella-ring-method.</li></ul><h2>SCA 26 — oscillating/relaxing drop</h2><ul style="font-size: 11.9999990463257px; "><li>Analysis of the real and imaginary part of the interfacial dilatational modulus based on the oscillating or relaxing contour of pendant drops.</li></ul></ul>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '2', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 235, 0x[encoded data, 4610 characters], '', '', 0, 0, '', 0),
(1184, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 238, 1428315163, 1428315141, 1, 0, 384, 'text', 'High-speed video systems', '', 'All instruments within the OCA series can be upgraded to include a high-speed video system. Your sales representative will consider your application(s), together with yourself, to determine the optimum system for your application.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '2', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 238, 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, '', '', 0, 0, '', 0),
(1185, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 239, 1428315251, 1428315165, 1, 0, 192, 'text', 'UpUSB 52H', '', 'Upgrade USB-Highspeed option for 1/2&quot;-USB-camera:\r\n<ul><li>768 x 576 Pixel, 123 images/s.</li><li>768 x 288 Pixel, 195 images/s.</li><li>640 x 240 Pixel, 215 images/s.</li></ul>\r\n<h2><br />UpUSB 87H</h2>\r\nUpgrade USB-Highspeed option for 1/3&quot;-USB-camera:\r\n<ul><li>640 x 480 Pixel, 182 images/s.</li><li>752 x 240 Pixel, 271 images/s.</li><li>640 x 240 Pixel, 311 images/s.</li></ul>\r\n<h2>UpHSV 1220</h2>\r\nHigh-speed option for contact angle measuring devices of the OCA series. For the determination of fast processes with up to 1220 images/second.<br />Consisting of:\r\n<ul><li>High-speed video system with 2/3&quot;-CCD -camera</li><li>Connection cable 1.5m USB 3.0</li></ul>\r\n<br />Resolution and image frequency:\r\n<ul><li>2048 x 1024 pixel, 75 images/s</li><li>1024 x 1024 pixel, 150 images/s</li><li>1024 x 512 pixel, 298 images/s</li><li>752 x 480 pixel, 318 images/s</li><li>640 x 240 pixel, 627 images/s</li><li>640 x 160 pixel, 928 images/s</li><li>640 x 120 pixel, 1220 images/s</li></ul>\r\nhigh-speed framegrabber and camera\r\n<h2>UpHSV 3000</h2>\r\nHigh-speed contact angle measuring device for the determination of fast processes with up to 3000 images/second<br />Consisting of:\r\n<ul><li>High-speed video system with 2/3&quot;-CCD -camera</li><li>High performance video digitizing board for the PC (PCI-e /X4)</li><li>High performance video cable</li><li>Power supply</li></ul>\r\n<br />Resolution and image frequency:\r\n<ul><li>2048 x 1024 pixel, 75 images/s</li><li>1024 x 1024 pixel, 151 images/s</li><li>1024 x 512 pixel, 300 images/s</li><li>752 x 480 pixel, 445 images/s</li><li>640 x 240 pixel, 1014 images/s</li><li>640 x 160 pixel, 1509 images/s</li><li>640 x 120 pixel, 1997 images/s</li><li>480 x 100 pixel, 3000 images/s</li></ul>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '2', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 239, 0x[encoded data, 4280 characters], '', '', 0, 0, '', 0),
(1186, 209, 0, 0, 0, '', 0, 0, 0, 0, 0, 239, 1428315254, 1428315168, 1, 0, 160, 'text', 'UpUSB 52H', '', 'Upgrade USB-Highspeed option for 1/2&quot;-USB-camera:\r\n<ul><li>768 x 576 Pixel, 123 images/s.</li><li>768 x 288 Pixel, 195 images/s.</li><li>640 x 240 Pixel, 215 images/s.</li></ul>\r\n<h2><br />UpUSB 87H</h2>\r\nUpgrade USB-Highspeed option for 1/3&quot;-USB-camera:\r\n<ul><li>640 x 480 Pixel, 182 images/s.</li><li>752 x 240 Pixel, 271 images/s.</li><li>640 x 240 Pixel, 311 images/s.</li></ul>\r\n<h2>UpHSV 1220</h2>\r\nHigh-speed option for contact angle measuring devices of the OCA series. For the determination of fast processes with up to 1220 images/second.<br />Consisting of:\r\n<ul><li>High-speed video system with 2/3&quot;-CCD -camera</li><li>Connection cable 1.5m USB 3.0</li></ul>\r\n<br />Resolution and image frequency:\r\n<ul><li>2048 x 1024 pixel, 75 images/s</li><li>1024 x 1024 pixel, 150 images/s</li><li>1024 x 512 pixel, 298 images/s</li><li>752 x 480 pixel, 318 images/s</li><li>640 x 240 pixel, 627 images/s</li><li>640 x 160 pixel, 928 images/s</li><li>640 x 120 pixel, 1220 images/s</li></ul>\r\nhigh-speed framegrabber and camera\r\n<h2>UpHSV 3000</h2>\r\nHigh-speed contact angle measuring device for the determination of fast processes with up to 3000 images/second<br />Consisting of:\r\n<ul><li>High-speed video system with 2/3&quot;-CCD -camera</li><li>High performance video digitizing board for the PC (PCI-e /X4)</li><li>High performance video cable</li><li>Power supply</li></ul>\r\n<br />Resolution and image frequency:\r\n<ul><li>2048 x 1024 pixel, 75 images/s</li><li>1024 x 1024 pixel, 151 images/s</li><li>1024 x 512 pixel, 300 images/s</li><li>752 x 480 pixel, 445 images/s</li><li>640 x 240 pixel, 1014 images/s</li><li>640 x 160 pixel, 1509 images/s</li><li>640 x 120 pixel, 1997 images/s</li><li>480 x 100 pixel, 3000 images/s</li></ul>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '2', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 239, 0x[encoded data, 4280 characters], '', '', 0, 0, '', 0),
(1187, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 242, 1428315298, 1428315266, 1, 0, 384, 'text', 'Special accessories', '', 'In close cooperation with customers, research institutes, and universities we have developed a wide range of application specific accessories to complement the OCA series of instruments. In doing so we are continuously broadening the range of surface chemistry applications within both the laboratory and process fields.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 242, 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, '', '', 0, 0, '', 0),
(1188, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 243, 1429621590, 1428315269, 1, 0, 192, 'text', 'Optical standards for verification of precise performance ', '', 'DataPhysics provide several sets of optical standards for the validation/verification of total instrument (hardware and software) performance. Each glass slide features a standard sessile, pendant drop or lamella contour for image capture and subsequent analysis. The comparison of the calculated contact angle or the surface/interfacial tension mea­sure­ment with that stated on the slide is a crucial factor in the evaluation of instrument performance and reliability. All slides are manufactured with the highest precision, as per ISO 9001, in a controlled photolithographic process to serve as reliable references for the precision of measured data.\r\n<ul><li>OCAS basic set&nbsp;three optical drop shape standards for the validation of the contact angle and the surface and interfacial tension measurement according to the height/width method, the Young-Laplace method, and the pendant drop method.</li><li>OCAS/HW&nbsp;­four optical contact angle standards according to the height/width method (15° ± 0.09°, 30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°).</li><li>OCAS/HW-LOW&nbsp; three optical contact angle standards according to the height/width method (15° ± 0.09°, 10° ± 0.09°, 5° ± 0.09°).</li><li>OCAS/YL&nbsp;three optical contact angle standards according to&nbsp; the Young/Laplace method (30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°).</li><li>OCAS/YL-HIGH&nbsp;four optical contact angle standards according to&nbsp; the Young/Laplace method (160° ± 0.04°, 165° ± 0.04°, 170° ± 0.04°, 175° ± 0.04°).</li><li>OCAS/PD&nbsp;four optical Pendant Drop standards (20, 35, 50 and 72.8 mN/m; ± 0.02 mN/m).</li><li>OCAS/LA&nbsp; two optical lamella standards (20 and 35 mN/m; ± 0.02 mN/m) and one Vertical-Rod standard (CA = 30° ± 0,09° and SFT 50 ± 0,02 mN/m).</li></ul>\r\n<h2>KeyPad for OCA 35/40 Micro KP 40</h2>\r\nFor controlling, outside of the software, the position of the motorized X-, Y-, Z-axes, the tilting angle of a tilting base unit, TBU 90E, the angle of an electronic turn table, ETT, and of the focus and observation angle (only OCA 40) of the optical system. The X,Y,Z co-ordinates of a sample table or the angle/position of a tilting unit/turn table, adjusted by the keypad, can be transferred into the SCA 20 software in order to facilitate the automated return to an optimised position during a subsequent automated measurement routine.\r\n<h2>Top view video system module for OCA TV-VS</h2>\r\nAll OCA systems can be equipped with the top view video system TV-VS, to provide an alternative view of the positioning of the dispensed drop on the surface. The USB camera, with up to 87 images per second, together with the high performance 6x parfocal zoom lens and the adjustable observation angle enables a detailed view to your sample.\r\n<h2>Oscillating drop generator ODG 20</h2>\r\nFor the measurement of surface elasticities and for relaxational studies at phase boundaries, DataPhysics has developed the oscillating drop generator, ODG 20, and the innovative oscillating drop generator ODG 20P with a high resolution pressure sensor. This extension to the pendant drop contour analysis capability of the OCA series – based on a piezoelectric transducer – excites oscillating drops with a wide range of frequencies and amplitudes. Periodic variations in the drop volume, or alternatively in the drop shape, with constant volume, can be performed. The excitation frequency ranges up to 50 Hz, depending on the density and viscosity of the fluid. The amplitude of the axis-symmetric oscillations varies from micrometers to millimeters. The oscillating drops can be analyzed by conventional drop contour analysis or by the new approach of pressure measurement with the high-performance pressure sensor (ODG 20P).\r\n<h2>Electro wetting platform EWP 100</h2>\r\nThe way in which a defined electrical field influences the wetting and adsorption behaviour of modern, engineered, materials is becoming relevant to an increasing number of applications. The EWP 100 facilitates the analysis of pendant drops and sessile drops, incident with solid surfaces, under a user definable electrical field.\r\n<h2>Liquid injection system LIA-MD</h2>\r\nThe LIA-MD facilitates the study of the effect, on a drops surface, of adding a second liquid directly into that first drop. This special accessory enables the rapid addition of a second liquid, usually a surface-active or chemically reactive phase, into a drop already visible through the OCA optics and displayed on the monitor. The relaxation (back to interfacial equilibrium with the surrounding medium) of the original drop can be observed and evaluated via the SCA software. The overall response time, together with the rates of change of surface and interfacial tensions, during the relaxation process, are indicative of the reactivity of the liquids under study. Developments of synthetic eye and lung fluids, for medical purposes, have already benefitted from studies of this type and more innovative applications are expected in the field of protein and surfactant engineering.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 243, 0x[encoded data, 9950 characters], '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1189, 210, 0, 0, 0, '', 0, 0, 0, 0, 0, 243, 1429621602, 1428315271, 1, 0, 160, 'text', 'Optical standards for verification of precise performance', '', '<p style="font-size: 11.9999990463257px; ">DataPhysics provide several sets of optical standards for the validation/verification of total instrument (hardware and software) performance. Each glass slide features a standard sessile, pendant drop or lamella contour for image capture and subsequent analysis. The comparison of the calculated contact angle or the surface/interfacial tension mea­sure­ment with that stated on the slide is a crucial factor in the evaluation of instrument performance and reliability. All slides are manufactured with the highest precision, as per ISO 9001, in a controlled photolithographic process to serve as reliable references for the precision of measured data.</p>\r\n<ul style="font-size: 11.9999990463257px; "><li>OCAS basic set&nbsp;three optical drop shape standards for the validation of the contact angle and the surface and interfacial tension measurement according to the height/width method, the Young-Laplace method, and the pendant drop method.</li><li>OCAS/HW&nbsp;­four optical contact angle standards according to the height/width method (15° ± 0.09°, 30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°).</li><li>OCAS/HW-LOW&nbsp; three optical contact angle standards according to the height/width method (15° ± 0.09°, 10° ± 0.09°, 5° ± 0.09°).</li><li>OCAS/YL&nbsp;three optical contact angle standards according to&nbsp; the Young/Laplace method (30° ± 0.09°, 60° ± 0.07°, 120° ± 0.04°).</li><li>OCAS/YL-HIGH&nbsp;four optical contact angle standards according to&nbsp; the Young/Laplace method (160° ± 0.04°, 165° ± 0.04°, 170° ± 0.04°, 175° ± 0.04°).</li><li>OCAS/PD&nbsp;four optical Pendant Drop standards (20, 35, 50 and 72.8 mN/m; ± 0.02 mN/m).</li><li>OCAS/LA&nbsp; two optical lamella standards (20 and 35 mN/m; ± 0.02 mN/m) and one Vertical-Rod standard (CA = 30° ± 0,09° and SFT 50 ± 0,02 mN/m).</li></ul>\r\n<h2>KeyPad for OCA 35/40 Micro KP 40</h2>\r\n<p style="font-size: 11.9999990463257px; ">For controlling, outside of the software, the position of the motorized X-, Y-, Z-axes, the tilting angle of a tilting base unit, TBU 90E, the angle of an electronic turn table, ETT, and of the focus and observation angle (only OCA 40) of the optical system. The X,Y,Z co-ordinates of a sample table or the angle/position of a tilting unit/turn table, adjusted by the keypad, can be transferred into the SCA 20 software in order to facilitate the automated return to an optimised position during a subsequent automated measurement routine.</p>\r\n<h2>Top view video system module for OCA TV-VS</h2>\r\n<p style="font-size: 11.9999990463257px; ">All OCA systems can be equipped with the top view video system TV-VS, to provide an alternative view of the positioning of the dispensed drop on the surface. The USB camera, with up to 87 images per second, together with the high performance 6x parfocal zoom lens and the adjustable observation angle enables a detailed view to your sample.</p>\r\n<h2>Oscillating drop generator ODG 20</h2>\r\n<p style="font-size: 11.9999990463257px; ">For the measurement of surface elasticities and for relaxational studies at phase boundaries, DataPhysics has developed the oscillating drop generator, ODG 20, and the innovative oscillating drop generator ODG 20P with a high resolution pressure sensor. This extension to the pendant drop contour analysis capability of the OCA series – based on a piezoelectric transducer – excites oscillating drops with a wide range of frequencies and amplitudes. Periodic variations in the drop volume, or alternatively in the drop shape, with constant volume, can be performed. The excitation frequency ranges up to 50 Hz, depending on the density and viscosity of the fluid. The amplitude of the axis-symmetric oscillations varies from micrometers to millimeters. The oscillating drops can be analyzed by conventional drop contour analysis or by the new approach of pressure measurement with the high-performance pressure sensor (ODG 20P).</p>\r\n<h2>Electro wetting platform EWP 100</h2>\r\n<p style="font-size: 11.9999990463257px; ">The way in which a defined electrical field influences the wetting and adsorption behaviour of modern, engineered, materials is becoming relevant to an increasing number of applications. The EWP 100 facilitates the analysis of pendant drops and sessile drops, incident with solid surfaces, under a user definable electrical field.</p>\r\n<h2>Liquid injection system LIA-MD</h2>\r\n<p style="font-size: 11.9999990463257px; ">The LIA-MD facilitates the study of the effect, on a drops surface, of adding a second liquid directly into that first drop. This special accessory enables the rapid addition of a second liquid, usually a surface-active or chemically reactive phase, into a drop already visible through the OCA optics and displayed on the monitor. The relaxation (back to interfacial equilibrium with the surrounding medium) of the original drop can be observed and evaluated via the SCA software. The overall response time, together with the rates of change of surface and interfacial tensions, during the relaxation process, are indicative of the reactivity of the liquids under study. Developments of synthetic eye and lung fluids, for medical purposes, have already benefitted from studies of this type and more innovative applications are expected in the field of protein and surfactant engineering.</p>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 4, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 243, 0x[encoded data, 9950 characters], '', '', 0, 0, '', 0),
(1190, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 246, 1428315574, 1428315528, 1, 0, 512, 'text', 'The prominence of the flat screen', '', 'The cleanliness of the glass surfaces, during each step of the manufacturing process, is of vital importance to the rate at which flat screens can be produced that pass the exacting quality standard set within his high value industry; not least to reduce the number of dead-pixels that might appear in future manufacturing steps.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 246, 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, '', '', 0, 0, '', 0),
(1191, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 246, 1428315665, 1428315530, 1, 0, 384, 'text', 'The prominence of the flat screen', '', 'The cleanliness of the glass surfaces, during each step of the manufacturing process, is of vital importance to the rate at which flat screens can be produced that pass the exacting quality standard set within his high value industry; not least to reduce the number of dead-pixels that might appear in future manufacturing steps.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '3', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 170, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 246, 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, '', '', 0, 0, '', 0),
(1192, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 806, 1428315623, 1428315615, 1, 1, 1000000000, 'shortcut', '[Translate to English:] Informationen (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 1, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 806, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1193, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 247, 1428315788, 1428315615, 1, 0, 112, 'textpic', 'PCM 100M', '', '<h3>Contact angle measuring technique almost everywhere.</h3>\r\nThe portable, video-based contact angle measuring system, PCA 100M is the ideal system for the measurement of wetting behaviour during the production process, out on the plant itself.\r\n<link 211 - - "Opens internal link in current window">read more</link>', '1', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 247, 0x613a33333a7b733a353a224354797065223b733a373a2274657874706963223b733a363a22636f6c506f73223b733a313a2232223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a363a22686561646572223b733a383a2250434d203130304d223b733a31333a226865616465725f6c61796f7574223b733a313a2230223b733a31353a226865616465725f706f736974696f6e223b733a303a22223b733a343a2264617465223b733a313a2230223b733a31313a226865616465725f6c696e6b223b733a303a22223b733a383a22626f647974657874223b733a3330373a224b6f6e74616b7477696e6b656c6d657373746563686e696b206661737420c3bc626572616c6c44617320766964656f67657374c3bc747a7465207472616762617265204b6f6e74616b74c2ad77696e6b656c6d657373c2ad676572c3a47420504341203130304d206973742064617320696465616c652053797374656d2066c3bc72204d657373756e67656e20766f6e2042656e65747ac2ad756e6773c2ad656967656ec2ad736368616674656e2077c3a46872656e64206465732050726f64756b74696f6e73c2ad70726f7a657373657320766f72204f72742e0d0a3c6c696e6b20323131202d20696e7465726e616c2d6c696e6b20224f70656e7320696e7465726e616c206c696e6b20696e2063757272656e742077696e646f77223e6d65687220657266616872656e3c2f6c696e6b3e223b733a31313a227274655f656e61626c6564223b733a313a2230223b733a353a22696d616765223b733a313a2231223b733a31303a22696d6167655f7a6f6f6d223b733a313a2230223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a333a22313630223b733a31303a22696d6167657769647468223b733a313a2230223b733a31313a22696d616765686569676874223b733a313a2230223b733a31313a22696d616765626f72646572223b733a313a2230223b733a31373a22696d6167655f636f6d7072657373696f6e223b733a313a2230223b733a31333a22696d6167655f65666665637473223b733a313a2230223b733a31313a22696d6167656f7269656e74223b733a313a2230223b733a393a22696d616765636f6c73223b733a313a2232223b733a31323a22696d6167655f6e6f526f7773223b733a313a2230223b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b733a303a22223b733a363a2268696464656e223b733a313a2230223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a393a22737461727474696d65223b4e3b733a373a22656e6474696d65223b4e3b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1194, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 248, 1428315843, 1428315615, 1, 0, 120, 'textpic', 'PCA 100R', '', '<h3>Analysis of the wetting behaviour integrated in the production process.</h3>\r\nIn conjuntion with the single or multiple axis robots of the RCC series, the contact angle measuing instrument PCA 100R is a specialised system for the analysis of wetting properties, of very large substrates, during the production process.\r\n<link 214 - - "Opens internal link in current window">read more</link>', '1', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 248, 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, '', '', 0, 0, '', 0),
(1195, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 804, 1428389548, 1428315674, 1, 0, 2336, 'shortcut', 'Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 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, '', '', 0, 0, '', 0),
(1196, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 805, 1428343625, 1428315674, 1, 0, 2688, 'shortcut', 'Informations', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1195, 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, '', '', 0, 0, '', 0),
(1197, 186, 0, 0, 0, '', 0, 0, 0, 0, 0, 1108, 1428343625, 1428315674, 1, 0, 2464, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1195, 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, '', '', 0, 0, '', 0),
(1198, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 247, 1428315809, 1428315792, 1, 0, 12, 'textpic', 'PCM 100M', '', '<h3>Contact angle measuring technique almost everywhere.</h3>\r\nThe portable, video-based contact angle measuring system, PCA 100M is the ideal system for the measurement of wetting behaviour during the production process, out on the plant itself.\r\n<link 211 - - "Opens internal link in current window">read more</link>', '1', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 247, 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, '', '', 0, 0, '', 0),
(1199, 187, 0, 0, 0, '', 0, 0, 0, 0, 0, 248, 1429530706, 1428315846, 1, 0, 14, 'textpic', 'PCA 100R', '', '<h3>Analysis of the wetting behaviour integrated in the production process.</h3>\r\nIn conjuntion with the single or multiple axis robots of the RCC series, the contact angle measuing instrument PCA 100R is a specialised system for the analysis of wetting properties, of very large substrates, during the production process.\r\n<link 214 - - "Opens internal link in current window">read more</link>', '1', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 160, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 248, 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, '', '', 0, 0, '', 0),
(1200, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 804, 1428348226, 1428315896, 1, 0, 160, 'shortcut', 'Informationen (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1201, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 805, 1428348226, 1428315896, 1, 0, 35, 'shortcut', 'Informations (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1200, 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, '', '', 0, 0, '', 0),
(1202, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 1108, 1428348226, 1428315896, 1, 0, 4, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1200, 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, '', '', 0, 0, '', 0),
(1203, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 804, 1428315915, 1428315909, 1, 0, 1536, 'shortcut', 'Informationen (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1204, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 805, 1428315915, 1428315909, 1, 0, 1280, 'shortcut', 'Informations (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1203, 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, '', '', 0, 0, '', 0),
(1205, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 1108, 1428315916, 1428315909, 1, 0, 1024, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1203, 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, '', '', 0, 0, '', 0);
INSERT INTO `tt_content` (`uid`, `pid`, `t3ver_oid`, `t3ver_id`, `t3ver_wsid`, `t3ver_label`, `t3ver_state`, `t3ver_stage`, `t3ver_count`, `t3ver_tstamp`, `t3ver_move_id`, `t3_origuid`, `tstamp`, `crdate`, `cruser_id`, `hidden`, `sorting`, `CType`, `header`, `header_position`, `bodytext`, `image`, `imagewidth`, `imageorient`, `imagecaption`, `imagecols`, `imageborder`, `media`, `layout`, `deleted`, `cols`, `records`, `pages`, `starttime`, `endtime`, `colPos`, `subheader`, `spaceBefore`, `spaceAfter`, `fe_group`, `header_link`, `imagecaption_position`, `image_link`, `image_zoom`, `image_noRows`, `image_effects`, `image_compression`, `altText`, `titleText`, `longdescURL`, `header_layout`, `menu_type`, `list_type`, `table_border`, `table_cellspacing`, `table_cellpadding`, `table_bgColor`, `select_key`, `sectionIndex`, `linkToTop`, `file_collections`, `filelink_size`, `filelink_sorting`, `target`, `section_frame`, `date`, `multimedia`, `image_frames`, `recursive`, `imageheight`, `rte_enabled`, `sys_language_uid`, `tx_impexp_origuid`, `pi_flexform`, `accessibility_title`, `accessibility_bypass`, `accessibility_bypass_text`, `l18n_parent`, `l18n_diffsource`, `selected_categories`, `category_field`, `categories`, `zzz_deleted_tx_srlanguagemenu_type`, `zzz_deleted_tx_srlanguagemenu_languages`, `tx_convert2utf8`) VALUES
(1206, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 363, 1428315971, 1428315940, 1, 0, 512, 'text', 'PCA 100M', '', '<h2>Video-based portable contact angle measuring instrument.</h2>\r\nWhether it be for large or small substrate areas, the PCA 100M is the ideal, handheld, portable instrument for controlling the qualities of coatings, examining the surface energies of pre-treated polymer films in the production process, or testing the qualities of cleaned or adhesion promoted glass surfaces, prior to the next processing stage.<br />The compact design and the “plug and play” connectivity, to any notebook or on-site available PC, without the need for a special frame grabber, enables the measurement of surface properties almost everywhere – including use in cleanrooms (US FED STD 209E cleanroom standard class 10).\r\n<b>Features of the basic configuration:</b>\r\n<ul><li>High-performance telecentric lens with integrated aperture and adjustable observation angle.</li><li>Video measuring system (max. 50 images/s sample rate). Software controlled continually adjustable back lighting, without hysteresis.</li><li>Integrated multiple dosing system, with either two dosing needles (PCA 100M/2) or four dosing needles (PCA 100M/4), for software controlled automatic needle selection and positioning.</li><li>Integrated automatic refill system with flushing and cleaning functions for dosing up to 5000 drops per filling and per liquid receptacle.</li><li>Power supply with automatic voltage adjustment.</li></ul>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 363, 0x[encoded data, 4816 characters], '', '', 0, 0, '', 0),
(1207, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 363, 1428316072, 1428315995, 1, 0, 384, 'text', 'PCA 100M', '', '<h2>Video-based portable contact angle measuring instrument.</h2>\r\nWhether it be for large or small substrate areas, the PCA 100M is the ideal, handheld, portable instrument for controlling the qualities of coatings, examining the surface energies of pre-treated polymer films in the production process, or testing the qualities of cleaned or adhesion promoted glass surfaces, prior to the next processing stage.<br />The compact design and the “plug and play” connectivity, to any notebook or on-site available PC, without the need for a special frame grabber, enables the measurement of surface properties almost everywhere – including use in cleanrooms (US FED STD 209E cleanroom standard class 10).\r\n<b>Features of the basic configuration:</b>\r\n<ul><li>High-performance telecentric lens with integrated aperture and adjustable observation angle.</li><li>Video measuring system (max. 50 images/s sample rate). Software controlled continually adjustable back lighting, without hysteresis.</li><li>Integrated multiple dosing system, with either two dosing needles (PCA 100M/2) or four dosing needles (PCA 100M/4), for software controlled automatic needle selection and positioning.</li><li>Integrated automatic refill system with flushing and cleaning functions for dosing up to 5000 drops per filling and per liquid receptacle.</li><li>Power supply with automatic voltage adjustment.</li></ul>', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 363, 0x[encoded data, 4816 characters], '', '', 0, 0, '', 0),
(1208, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 365, 1429621713, 1428316003, 1, 0, 6, 'textpic', 'Process applications and production control', '', 'For this product series, we offer a variety of components and an extensive range of accessories.\r\n<link 187 - - "Opens internal link in current window">read more</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 365, 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, '', '', 0, 0, '', 0),
(1209, 211, 0, 0, 0, '', 0, 0, 0, 0, 0, 365, 1429621722, 1428316006, 1, 0, 3, 'textpic', 'Applications du processus et contrôle de la production', '', '<p style="font-size: 11.9999990463257px; ">For this product series, we offer a variety of components and an extensive range of accessories.</p>\r\n<p style="font-size: 11.9999990463257px; "><link 187 - - "Opens internal link in current window">read more</link></p>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 365, 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, '', '', 0, 0, '', 0),
(1210, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 365, 1429621778, 1428316252, 1, 0, 768, 'textpic', 'Prozessanwendungen und Produktionskontrollen', '', 'Zu dieser Produktreihe bieten wir eine Vielzahl an Komponenten und ein umfangreiches Zubehörprogramm an.\r\n<link 187 - - "Opens internal link in current window">Übersicht</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a33323a7b733a353a224354797065223b4e3b733a363a22636f6c506f73223b4e3b733a31363a227379735f6c616e67756167655f756964223b4e3b733a363a22686561646572223b4e3b733a31333a226865616465725f6c61796f7574223b4e3b733a31353a226865616465725f706f736974696f6e223b4e3b733a343a2264617465223b4e3b733a31313a226865616465725f6c696e6b223b4e3b733a383a22626f647974657874223b4e3b733a31313a227274655f656e61626c6564223b4e3b733a353a22696d616765223b4e3b733a31303a22696d6167655f7a6f6f6d223b4e3b733a363a226c61796f7574223b4e3b733a31313a2273706163654265666f7265223b4e3b733a31303a2273706163654166746572223b4e3b733a31333a2273656374696f6e5f6672616d65223b4e3b733a31303a22696d6167657769647468223b4e3b733a31313a22696d616765686569676874223b4e3b733a31313a22696d616765626f72646572223b4e3b733a31373a22696d6167655f636f6d7072657373696f6e223b4e3b733a31333a22696d6167655f65666665637473223b4e3b733a31313a22696d6167656f7269656e74223b4e3b733a393a22696d616765636f6c73223b4e3b733a31323a22696d6167655f6e6f526f7773223b4e3b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b4e3b733a363a2268696464656e223b4e3b733a31323a2273656374696f6e496e646578223b4e3b733a393a226c696e6b546f546f70223b4e3b733a393a22737461727474696d65223b4e3b733a373a22656e6474696d65223b4e3b733a383a2266655f67726f7570223b4e3b733a31303a2263617465676f72696573223b4e3b7d, '', '', 0, 0, '', 0),
(1211, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 1208, 1429621736, 1428316252, 1, 0, 512, 'textpic', 'Process applications and production control', '', '<p style="font-size: 11.9999990463257px; ">For this product series, we offer a variety of components and an extensive range of accessories.</p>\r\n<p style="font-size: 11.9999990463257px; "><link 187 - - "Opens internal link in current window">read more</link></p>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1210, 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, '', '', 0, 0, '', 0),
(1212, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 1209, 1429621744, 1428316252, 1, 0, 256, 'textpic', 'Applications du processus et contrôle de la production', '', '<p style="font-size: 11.9999990463257px; ">For this product series, we offer a variety of components and an extensive range of accessories.</p>\r\n<p style="font-size: 11.9999990463257px; "><link 187 - - "Opens internal link in current window">read more</link></p>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1210, 0x613a33313a7b733a363a2268696464656e223b733a313a2230223b733a353a224354797065223b733a373a2274657874706963223b733a363a22636f6c506f73223b733a313a2232223b733a31363a227379735f6c616e67756167655f756964223b733a313a2230223b733a31313a226c31386e5f706172656e74223b733a313a2230223b733a363a22686561646572223b733a35333a2250726f7a657373616e77656e64756e67656e20756e642050726f64756b74696f6e736b6f6e74726f6c6c656e2028636f7079203129223b733a31333a226865616465725f6c61796f7574223b733a313a2230223b733a31353a226865616465725f706f736974696f6e223b733a303a22223b733a343a2264617465223b733a313a2230223b733a31313a226865616465725f6c696e6b223b733a303a22223b733a383a22626f647974657874223b733a3336303a224c6f72656d20697073756d20646f6c6f722073697420616d65742c20636f6e736574657475722073616469707363696e6720656c6974722c20736564206469616d206e6f6e756d79206569726d6f642074656d706f7220696e766964756e74207574206c61626f726520657420646f6c6f7265206d61676e6120616c69717579616d20657261742c20736564206469616d20766f6c75707475612e204c6f72656d20697073756d20646f6c6f722073697420616d65742c20636f6e736574657475722073616469707363696e6720656c6974722c20736564206469616d206e6f6e756d79206569726d6f642074656d706f7220696e766964756e74207574206c61626f726520657420646f6c6f72652e0d0a3c6c696e6b20313837202d20696e7465726e616c2d6c696e6b20224f70656e7320696e7465726e616c206c696e6b20696e2063757272656e742077696e646f77223ec39c626572736963687420c2bb3c2f6c696e6b3e223b733a31313a227274655f656e61626c6564223b733a313a2230223b733a353a22696d616765223b733a313a2231223b733a31303a22696d6167655f7a6f6f6d223b733a313a2230223b733a363a226c61796f7574223b733a313a2230223b733a31313a2273706163654265666f7265223b733a313a2230223b733a31303a2273706163654166746572223b733a313a2230223b733a31333a2273656374696f6e5f6672616d65223b733a333a22313130223b733a31303a22696d6167657769647468223b733a313a2230223b733a31313a22696d616765686569676874223b733a313a2230223b733a31313a22696d616765626f72646572223b733a313a2230223b733a31373a22696d6167655f636f6d7072657373696f6e223b733a313a2230223b733a31333a22696d6167655f65666665637473223b733a313a2230223b733a31313a22696d6167656f7269656e74223b733a323a223137223b733a393a22696d616765636f6c73223b733a313a2232223b733a31323a22696d6167655f6e6f526f7773223b733a313a2230223b733a32313a22696d61676563617074696f6e5f706f736974696f6e223b733a303a22223b733a31323a2273656374696f6e496e646578223b733a313a2231223b733a393a226c696e6b546f546f70223b733a313a2230223b733a383a2266655f67726f7570223b733a303a22223b733a31303a2263617465676f72696573223b733a313a2230223b7d, '', '', 0, 0, '', 0),
(1213, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 360, 1428316334, 1428316262, 1, 0, 2304, 'text', 'PCA 100R', '', '<h2>Robotically guided contact angle measuring instrument for process control.</h2>\r\nIn combination with the single and multi axis cartesian robots of the RCC series, the PCA 100R contact angle measuring instrument is a specialised system for measuring wetting properties on substrates of very large surface areas, on site, during the production process. The fully automatic instrument enables the removes operator variability and other subjective factors and of the time necessary for the contact angle measurement.<br />Some important applications for the robotically guided PCA 100R are:\r\n<ul><li>determination of the surface cleanliness of wafers and glass substrates for flat screens before the next manufacturing step.</li><li>examining the surface energies of pre-treated polymer films in the production process.</li><li>development of high-performance composites.</li><li>surface finishing and the cleaning of textiles.</li></ul>\r\n<br />The PCA 100R and the robots of the RCC series are designed for the usage in cleanrooms according the US FED STD 209E cleanroom standard class 10.\r\n\r\n<b>Features of the basic instrument configuration:</b>\r\n<ul><li>measuring head with integrated optics, illumination and multiple dosing unit.</li><li>high-performance telecentric lens with integrated aperture and adjustable observation angle.</li><li>video-based measuring system with high-resolution CCD camera and high-performance digitizing adapter.</li><li>software controlled continuous adjustable intensity without hysteresis for a homogeneous back lighting.</li><li>automatic needle selection and positioning.</li></ul>\r\n<br /><b>The consequent expansion of our accessories range allows us to adapt or to refine the robotically guided contact angle measuring instrument for your application.<br />If we aroused your interest, please give us the chance find or develop a&nbsp;</b><b>robotically guided contact angle measuring instrument ideal for your process.<br /></b>One of our strengths is the customisation of our products to your needs —please&nbsp;<link 238 - internal-link "Opens internal link in current window">call directly</link>&nbsp;or fill out the&nbsp;<link 15 - internal-link "Opens internal link in current window">contact form</link>.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 360, 0x[encoded data, 5490 characters], '', '', 0, 0, '', 0),
(1214, 214, 0, 0, 0, '', 0, 0, 0, 0, 0, 360, 1428316354, 1428316264, 1, 0, 2176, 'text', 'PCA 100R', '', '<h2>Robotically guided contact angle measuring instrument for process control.</h2>\r\nIn combination with the single and multi axis cartesian robots of the RCC series, the PCA 100R contact angle measuring instrument is a specialised system for measuring wetting properties on substrates of very large surface areas, on site, during the production process. The fully automatic instrument enables the removes operator variability and other subjective factors and of the time necessary for the contact angle measurement.<br />Some important applications for the robotically guided PCA 100R are:\r\n<ul><li>determination of the surface cleanliness of wafers and glass substrates for flat screens before the next manufacturing step.</li><li>examining the surface energies of pre-treated polymer films in the production process.</li><li>development of high-performance composites.</li><li>surface finishing and the cleaning of textiles.</li></ul>\r\n<br />The PCA 100R and the robots of the RCC series are designed for the usage in cleanrooms according the US FED STD 209E cleanroom standard class 10.\r\n\r\n<b>Features of the basic instrument configuration:</b>\r\n<ul><li>measuring head with integrated optics, illumination and multiple dosing unit.</li><li>high-performance telecentric lens with integrated aperture and adjustable observation angle.</li><li>video-based measuring system with high-resolution CCD camera and high-performance digitizing adapter.</li><li>software controlled continuous adjustable intensity without hysteresis for a homogeneous back lighting.</li><li>automatic needle selection and positioning.</li></ul>\r\n<br /><b>The consequent expansion of our accessories range allows us to adapt or to refine the robotically guided contact angle measuring instrument for your application.<br />If we aroused your interest, please give us the chance find or develop a&nbsp;</b><b>robotically guided contact angle measuring instrument ideal for your process.<br /></b>One of our strengths is the customisation of our products to your needs —please&nbsp;<link 238 - internal-link "Opens internal link in current window">call directly</link>&nbsp;or fill out the&nbsp;<link 15 - internal-link "Opens internal link in current window">contact form</link>.', '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 1, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 360, 0x[encoded data, 5490 characters], '', '', 0, 0, '', 0),
(1215, 188, 0, 0, 0, '', 0, 0, 0, 0, 0, 1203, 1428316394, 1428316386, 1, 0, 1536, 'shortcut', 'Informationen  (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '14', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 0x613a313a7b733a363a2268696464656e223b4e3b7d, '', '', 0, 0, '', 0),
(1216, 188, 0, 0, 0, '', 0, 0, 0, 0, 0, 1204, 1428316395, 1428316386, 1, 0, 1280, 'shortcut', 'Informations  (copy 1)', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '649', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 2, 0, NULL, '', 0, '', 1215, 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, '', '', 0, 0, '', 0),
(1217, 188, 0, 0, 0, '', 0, 0, 0, 0, 0, 1205, 1428316395, 1428316386, 1, 0, 1024, 'shortcut', '[Translate to Français:] Informationen', '', NULL, '0', 0, 0, NULL, 2, 0, '0', 0, 0, 0, '920', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 0, 0, '', 0, 0, 0, 0, 3, 0, NULL, '', 0, '', 1215, 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, '', '', 0, 0, '', 0),
(1218, 188, 0, 0, 0, '', 0, 0, 0, 0, 0, 1210, 1429621810, 1428316386, 1, 0, 768, 'textpic', 'Prozessanwendungen und Produktionskontrollen', '', 'Zu dieser Produktreihe bieten wir eine Vielzahl an Komponenten und ein umfangreiches Zubehörprogramm an.\r\n<link 187 - - "Opens internal link in current window">Übersicht</link>', '1', 0, 17, NULL, 2, 0, '0', 0, 0, 0, '', '', 0, 0, 2, '', 0, 0, '', '', '', NULL, 0, 0, 0, 0, NULL, NULL, NULL, '0', '0', '', 0, 0, 0, 0, '', 1, 0, '', 0, '', '', 110, 0, '', 0, 0, 0, 0, 0, 0, NULL, '', 0, '', 0, 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, '', '', 0, 0, '', 0);
