Each of the temperature control units, available to the DCAT tensiometers, are easy to install and are available for different sample vessel sizes and temperature ranges.<br /><b>TV 70</b>&nbsp;Liquid temperature control unit
<h2>TV 50 / 70 / 100</h2>
Liquid temperature control unit with a built-in Pt 100 temperature measuring sensor. For temperature range of -10 to 130 °C, controlled by an external liquid circulator.<br />Available in the following standard sizes:
<ul><li><strong>TV 50</strong>&nbsp;for sample vessels with 50 mm diameter.</li><li><strong>TV 70</strong>&nbsp;for sample vessels with 70 mm diameter.</li><li><strong>TV 100</strong>&nbsp;for sample vessels with 100 mm diameter.</li></ul>

<h2>TV 70NM</h2>
Non-magnetic (hard gold-plated brass) liquid temperature control unit, including a micro-electronic stirrer (Super flat removable magnetic stirrer without any motor), sealed and chemical resistant and a built-in Pt 100 temperature measuring sensor. For temperature range of -10 to 130 °C, controlled by an external liquid circulator.

<h2>TEC 250/DCAT</h2>
Electrical temperature control unit to use with the dynamic contact angle measuring device/tensiometer DCAT 21.<br />For measurements at temperatures up to 250 °C under protective atmosphere, e.g. for measuring of resins, hot melt adhesives and solder.<br />Manual or software controlled temperature setting over temperature controller TC 250.<br />Diameter of sample container: 50 mm