a:2:{s:9:"oldRecord";a:2:{s:8:"bodytext";s:7712:"Device for the heating of gastight syringe DS xx/GT mounted in the electronic syringe module ES. For use in conjunction with a liquid circulator.<link http://www.dataphysics.de/?action=zoom_view&id=795&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>TFC 100 with TDI 100<br />
<h2>TFC 100</h2>
Liquid temperature control unit for measurement within the range - 10 to 130 °C. Requires an external liquid circulator bath and includes the pipe work etc. to allow for a protective atmosphere (inert gas inlet) to surround the sample working area. The triple diffuser, TDI 100, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas flow towards the chamber windows.&nbsp;<br />An optional humidity sensor expands the applicability of the TFC 100 even further.<br /><link http://www.dataphysics.de/?action=zoom_view&id=796&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>TPC 150 with controller TC 150, cover plates and TPL 30 with SVL 12/30<br />
<h2>TPC 150</h2>
Temperature control chamber with a peltier system for the software controlled temperature setting from -30 t0 160 °C (heat up and cool down rate ±1 K/s). Also includes the pipe work etc. to allow for a protective atmosphere to surround the working area. The triple diffuser, TDI 150, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas towards the chamber windows.

An optional humidity sensor expands the applicability of the TPC 150 even further.TEC 300XL
<h2>TEC 300XL</h2>
For mounting to the XL-variants of the OCA measuring devices. Temperature control chamber with electric resistance heater for manual controlled temperature setting up to 300 °C. Temperature display on controller and in SCA-software. Work under a protective atmosphere is also facilitated via the inert gas inlet.<link http://www.dataphysics.de/?action=zoom_view&id=801&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>TEC 400 with controller TC 400 and TPL 30 with SVL 12/30
<h2>TEC 400</h2>
Temperature control chamber with twin electric resistance heater (above and below the sample plate) for software controlled temperature setting up to 400 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.<br /><link http://www.dataphysics.de/?action=zoom_view&id=800&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>TEC 700 with controller TC 700<br />
<h2>TEC 700</h2>
Temperature control chambers with twin electric resistance heater (above and below the sample plate) for software control of temperature setting up to 700 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.<br /><link http://www.dataphysics.de/?action=zoom_view&id=798&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>&nbsp;NHD 400<br />
<h2>NHD 400 / NHD 700</h2>
The NHD 400 and NHD 700 are electrical needle heating devices for generating drops at temperatures up 400 °C and 700 °C respectively (Applications areas include: polymers and hot melts, molten metals, etc.). Also extremely tenacious copolymers, with viscosities in the kPas range, e.g. with elastomer constituents, can be dosed easily at these elevated temperatures.<br /><link http://www.dataphysics.de/?action=zoom_view&id=844&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link><br />Molten aluminium droplet in NHD 700/TEC 700<br /><link http://www.dataphysics.de/?action=zoom_view&id=802&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>MTFQ 1200 on an OCA 15LHT with controller and second sample table for measurements at ambient temperature<br />
<h2>MTFQ 1200</h2>
Special high temperature furnace for measurements to a maximum of 1200°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the MTFQ 1200 also.<br /><link http://www.dataphysics.de/?action=zoom_view&id=846&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link><br />Molten glass drop in MTFQ 1200<br /><link http://www.dataphysics.de/?action=zoom_view&id=803&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>HTFC 1700 on an OCA 15LHT with controller<br />
<h2>HTFC 1500/1700</h2>
Special high temperature furnace for measurements to a maximum of 1500/1700°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the HTFQ 1500/1700 also.OCA 15LHT-SV with HTFC 1800HV<br />
<h2>HTFC 1500/1800HV</h2>
Electrically heated, high temperature, high vacuum tube furnace for use in combination with an OCA 15LHT-SV.&nbsp;<br />
<ul><li>Max. temperature: 1500/1800 °C</li><li>Max. temperature for operation under vacuum: 1450°C</li><li>Max. vacuum: 10<sup>-5</sup>&nbsp;mbar</li></ul>
<br />vacuum tube:<br />Material: C799<br />Outer diameter: 50 mm<br />Inner diameter: 40 mm<br />Length: 990 mm<br />-&nbsp;&nbsp;&nbsp; Prepared for horizontal operation&nbsp;<br />-&nbsp;&nbsp;&nbsp; Observation windows on both sides as additional equipment for gastight flanges<br />-&nbsp;&nbsp;&nbsp; MoSi2 heating elements, mounted vertically for easy replacement<br />-&nbsp;&nbsp;&nbsp; Insulation with vacuum-formed ceramic fibre plates for an energy-saving operation and fast heating up times<br />-&nbsp;&nbsp;&nbsp; Rectangular outer casing with slots for convection cooling<br />-&nbsp;&nbsp;&nbsp; Casing made of sheets of textured stainless steel (non-rusting design)<br />-&nbsp;&nbsp;&nbsp; Type B thermocouple<br />-&nbsp;&nbsp;&nbsp; Power unit with low-voltage transformer and thyristor controller&nbsp;<br />Typically used for the measurement of contact angle and surface tension of materials like:
<ul><li>Compound Materials</li><li>Steel</li><li>Alloys</li><li>Molten Glass</li></ul>
<link http://www.dataphysics.de/?action=zoom_view&id=684&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>PMC 50S on an OCA 20L with high pressure pump HPP 150<br />
<h2>PMC 50S</h2>
Pressure cell with two viewing windows and manipulation system to allow the exchange of the sample table. Facilitates the evaluation of contact angles according to the sessile and captive drop method, the measurement of surface tensions in liquid/pressurized gas systems and the interfacial tensions in liquid/liquid systems.&nbsp;
Incorporates four dosing needles, for separate sessile and pendant drop measurements via four different injection liquids and a two phase pressure supply system for an external supply of the gas or liquid phases. The first pressure level is used for controlling the pressure of the primary filling liquid or gas. The higher, second pressure level is necessary for the injection dosing of the measuring liquid(s). The primary&nbsp; phase acts also a heat transfer medium.&nbsp;
<ul><li>pressure range: 0.03 ... 50 bar</li><li>max. temperature: 80 °C</li></ul>
<link http://www.dataphysics.de/?action=zoom_view&id=1074&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2012+by+Dataphysics _blank></link>HGC 30<br />
<h2>HGC 20EC/20/30</h2>
For the automated regulation of the humidity of ambient clean air, nitrogen, argon, or any other inert gas in temperature controlled measuring chambers.";s:15:"l18n_diffsource";s:4363:"a:22:{s:5:"CType";s:4:"text";s:6:"colPos";s:1:"4";s:16:"sys_language_uid";s:1:"0";s:6:"header";s:0:"";s:13:"header_layout";s:1:"0";s:15:"header_position";s:0:"";s:4:"date";s:1:"0";s:11:"header_link";s:0:"";s:8:"bodytext";s:3819:"<h2>SHD</h2>
Thermostatisiereinrichtung für die Temperierung einer gasdichten Spritze DS xx/GT oder einer Einwegspritze bis 90 °C zur Montage in dem elektronischen Spritzenmodul ES in Verbindung mit einem Flüssigkeitsthermostaten.
<h2>TFC 100</h2>
Flüssigkeit-Thermostatisiereinrichtung für Temperaturen von - 10…130 °C zum Anschluss an einen Kälte-Umwälzthermostat sowie an eine Schutzgasversorgung. Der Dreifach-Diffusor TDI 100 ist ein empfohlenes Zubehörteil für Betriebstemperaturen unter Raumtemperatur, um Kondensation durch einen trockenen Gasstrom am Kammerfenster zu verhindern. Die TFC 100 ist mit einem Feuchte­sensor erweiterbar.
<h2>TPC 150</h2>
Thermostatisiereinrichtung mit Peltiersystem für eine softwaregesteuerte Kontrolle der Temperatur von -30…160 °C (Aufheiz- und Abkühlrate ±1 K/s), auch unter Schutzgasatmosphäre. Der Dreifach-Diffusor TDI 150 ist ein empfohlenes Zubehörteil für Betriebstemperaturen unter Raumtemperatur, um Kondensation durch einen trockenen Gasstrom am Kammerfenster zu verhindern. Die TPC 150 ist mit einem Feuchte­sensor erweiterbar.
<h2>TEC 300 XL</h2>
Für die Benutzung mit den XL-Varianten der OCA Messgeräte. Thermostatisiereinrichtung mit elektrischer Widerstandsdoppelheizung für eine manuelle Kontrolle der Temperatur bis zu 300 °C, auch unter Schutzgasatmosphäre. Die Kammertemperatur wird in der SCA-Software in Kombination mit Kontaktwinkelmessung protokolliert.
<h2>TEC 400</h2>
Thermostatisiereinrichtung mit elektrischer Widerstandsdoppelheizung für eine software­gesteuerte Kontrolle der Temperatur bis zu 400 °C, auch unter Schutzgasatmosphäre.
<h2>TEC 700</h2>
Thermostatisiereinrichtung mit elektrischer Widerstandsdoppelheizung für eine software­gesteuerte Kontrolle der Temperatur bis zu 700 °C, auch unter Schutzgasatmosphäre.
<h2>NHD 400 / NHD 700</h2>
Die NHD 400 und NHD 700 sind elektrische Kanülen-Heizsysteme für die Erzeugung von Tropfen bei einer Temperatur von bis zu 400 °C bzw. 700 °C (z. B. Polymere und Hotmelts, geschmolzenes Metall, etc.). Ebenso können extrem zähe Copolymere mit Viskositäten im kPas Bereich, z.B. mit Elastomer-Komponenten, leicht dosiert werden.
<h2>MTFQ 1200</h2>
Hochtemperaturrohrofen für Messungen im Hochtemperaturbereich, z.B. an Loten, Halbleitern, Legierungen und Glasschmelzen, auch für Messungen unter Schutzgas-Atmosphäre bei bis zu 1200°C.
<h2>HTFC 1500/1700</h2>
Hochtemperaturrohrofen für Messungen im Hochtemperaturbereich, z.B. an Loten, Halbleitern, Legierungen und Glasschmelzen, auch für Messungen unter Schutzgas-Atmosphäre bei bis zu 1500/1700 °C
<h2>HTFC 1500/1800 HV</h2>
Elektrisch beheizter Hochtemperaturofen mit Hochvakuum zur Verwendung mit einem OCA 15LHT-SV Max. Temperatur 1500/1800 °C und max. 1 mPa Vakuum
<h2>PMC 50S</h2>
Druckmesszelle mit Sichtfenstern aus optischem Saphirglas, für Kontaktwinkelmessungen nach der Sessile Drop- und Captive Bubble-Methode und für die Messung der Oberflächenspannung in Flüssig-/Gas-Systemen und für Grenzflächenspannungen in Flüssig-/Flüssig-Systemen.<br />Mit vier Dosierkanülen für Sessile &amp; Captive sowie Pendant Drop Messungen, mit vier verschiedenen Einspritzflüssigkeiten. Äusseres Zwei-Ebenen Druck-Zufuhr-System. Die erste Druckebene kontrolliert den Druck der Primär-Einfüllflüssigkeit. Der höhere, zweite Druck ist notwendig für die Injektion der Meßflüssigkeit/en.<br />Druckbereich: 0,03 ... 50 bar<br />Max. Temperatur: 80 °C
<h2>HGC 20EC/20/30</h2>
Für die automatiserte Regelung der Feuchte von reiner Umgebungsluft, Stickstoff, Argon oder anderer Edelgase in thermostatisierten Messzellen.
Mehr Informationen über den&nbsp;<link http://kunden.werboffice.de/dataphysics/server/feuchtegenerator-hgc.html>Feuchtegenerator HGC</link>";s:11:"rte_enabled";s:1:"0";s:6:"layout";s:1:"0";s:11:"spaceBefore";s:1:"0";s:10:"spaceAfter";s:1:"0";s:13:"section_frame";s:1:"0";s:6:"hidden";s:1:"0";s:12:"sectionIndex";s:1:"1";s:9:"linkToTop";s:1:"0";s:9:"starttime";N;s:7:"endtime";N;s:8:"fe_group";s:0:"";s:10:"categories";s:1:"0";s:11:"l18n_parent";s:1:"0";}";}s:9:"newRecord";a:2:{s:8:"bodytext";s:6585:"<p style="font-size: 11.9999990463257px; ">Device for the heating of gastight syringe DS xx/GT mounted in the electronic syringe module ES. For use in conjunction with a liquid circulator.<link http://www.dataphysics.de/?action=zoom_view&id=795&invisible=1&zoom_size=1&zoom_copyright=%26amp%3Bcopy%3B+2010+by+Dataphysics _blank></link>TFC 100 with TDI 100</p>
<h2>TFC 100</h2>
<p style="font-size: 11.9999990463257px; ">Liquid temperature control unit for measurement within the range - 10 to 130 °C. Requires an external liquid circulator bath and includes the pipe work etc. to allow for a protective atmosphere (inert gas inlet) to surround the sample working area. The triple diffuser, TDI 100, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas flow towards the chamber windows.&nbsp;<br />An optional humidity sensor expands the applicability of the TFC 100 even further.</p>
<h2>TPC 150</h2>
<p style="font-size: 11.9999990463257px; ">Temperature control chamber with a peltier system for the software controlled temperature setting from -30 t0 160 °C (heat up and cool down rate ±1 K/s). Also includes the pipe work etc. to allow for a protective atmosphere to surround the working area. The triple diffuser, TDI 150, is a recommended attachment for working temperatures below room temperature to prevent condensation by directing a dry gas towards the chamber windows.</p>
<p style="font-size: 11.9999990463257px; ">An optional humidity sensor expands the applicability of the TPC 150 even further.</p>
<h2>TEC 300XL</h2>
<p style="font-size: 11.9999990463257px; ">For mounting to the XL-variants of the OCA measuring devices. Temperature control chamber with electric resistance heater for manual controlled temperature setting up to 300 °C. Temperature display on controller and in SCA-software. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>
<h2>TEC 400</h2>
<p style="font-size: 11.9999990463257px; ">Temperature control chamber with twin electric resistance heater (above and below the sample plate) for software controlled temperature setting up to 400 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>
<h2>TEC 700</h2>
<p style="font-size: 11.9999990463257px; ">Temperature control chambers with twin electric resistance heater (above and below the sample plate) for software control of temperature setting up to 700 °C. Work under a protective atmosphere is also facilitated via the inert gas inlet.</p>
<h2>NHD 400 / NHD 700</h2>
<p style="font-size: 11.9999990463257px; ">The NHD 400 and NHD 700 are electrical needle heating devices for generating drops at temperatures up 400 °C and 700 °C respectively (Applications areas include: polymers and hot melts, molten metals, etc.). Also extremely tenacious copolymers, with viscosities in the kPas range, e.g. with elastomer constituents, can be dosed easily at these elevated temperatures.</p>
<h2>MTFQ 1200</h2>
<p style="font-size: 11.9999990463257px; ">Special high temperature furnace for measurements to a maximum of 1200°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the MTFQ 1200 also.</p>
<h2>HTFC 1500/1700</h2>
<p style="font-size: 11.9999990463257px; ">Special high temperature furnace for measurements to a maximum of 1500/1700°C. Application areas include: measurement of/on liquid solders, semi-conductors, alloys and glass melts. Protective gas atmospheres can be generated and enclosed within the HTFQ 1500/1700 also.</p>
<h2>HTFC 1500/1800HV</h2>
<p style="font-size: 11.9999990463257px; ">Electrically heated, high temperature, high vacuum tube furnace for use in combination with an OCA 15LHT-SV.&nbsp;</p>
<ul style="font-size: 11.9999990463257px; "><li>Max. temperature: 1500/1800 °C</li><li>Max. temperature for operation under vacuum: 1450°C</li><li>Max. vacuum: 10<sup>-5</sup>&nbsp;mbar</li></ul>
<p style="font-size: 11.9999990463257px; "><br />vacuum tube:</p>
<p style="font-size: 11.9999990463257px; ">Material: C799<br />Outer diameter: 50 mm<br />Inner diameter: 40 mm<br />Length: 990 mm<br />-&nbsp;&nbsp;&nbsp; Prepared for horizontal operation&nbsp;<br />-&nbsp;&nbsp;&nbsp; Observation windows on both sides as additional equipment for gastight flanges<br />-&nbsp;&nbsp;&nbsp; MoSi2 heating elements, mounted vertically for easy replacement<br />-&nbsp;&nbsp;&nbsp; Insulation with vacuum-formed ceramic fibre plates for an energy-saving operation and fast heating up times<br />-&nbsp;&nbsp;&nbsp; Rectangular outer casing with slots for convection cooling<br />-&nbsp;&nbsp;&nbsp; Casing made of sheets of textured stainless steel (non-rusting design)<br />-&nbsp;&nbsp;&nbsp; Type B thermocouple<br />-&nbsp;&nbsp;&nbsp; Power unit with low-voltage transformer and thyristor controller&nbsp;<br />Typically used for the measurement of contact angle and surface tension of materials like:</p>
<div class="indent" style="font-size: 11.9999990463257px; "><ul><li>Compound Materials</li><li>Steel</li><li>Alloys</li><li>Molten Glass</li></ul></div>
<p style="font-size: 11.9999990463257px; ">&nbsp;</p>
<h2>PMC 50S</h2>
<p style="font-size: 11.9999990463257px; ">Pressure cell with two viewing windows and manipulation system to allow the exchange of the sample table. Facilitates the evaluation of contact angles according to the sessile and captive drop method, the measurement of surface tensions in liquid/pressurized gas systems and the interfacial tensions in liquid/liquid systems.&nbsp;</p>
<p style="font-size: 11.9999990463257px; ">Incorporates four dosing needles, for separate sessile and pendant drop measurements via four different injection liquids and a two phase pressure supply system for an external supply of the gas or liquid phases. The first pressure level is used for controlling the pressure of the primary filling liquid or gas. The higher, second pressure level is necessary for the injection dosing of the measuring liquid(s). The primary&nbsp; phase acts also a heat transfer medium.&nbsp;</p>
<ul style="font-size: 11.9999990463257px; "><li>pressure range: 0.03 ... 50 bar</li><li>max. temperature: 80 °C</li></ul>
<h2>HGC 20EC/20/30</h2>
<p style="font-size: 11.9999990463257px; ">For the automated regulation of the humidity of ambient clean air, nitrogen, argon, or any other inert gas in temperature controlled measuring chambers.</p>";s:15:"l18n_diffsource";s:4350:"a:22:{s:5:"CType";s:4:"text";s:6:"colPos";s:1:"4";s:16:"sys_language_uid";s:1:"0";s:6:"header";s:0:"";s:13:"header_layout";s:1:"0";s:15:"header_position";s:0:"";s:4:"date";s:1:"0";s:11:"header_link";s:0:"";s:8:"bodytext";s:3806:"<h2>SHD</h2>
Thermostatisiereinrichtung für die Temperierung einer gasdichten Spritze DS xx/GT oder einer Einwegspritze bis 90 °C zur Montage in dem elektronischen Spritzenmodul ES in Verbindung mit einem Flüssigkeitsthermostaten.
<h2>TFC 100</h2>
Flüssigkeit-Thermostatisiereinrichtung für Temperaturen von - 10…130 °C zum Anschluss an einen Kälte-Umwälzthermostat sowie an eine Schutzgasversorgung. Der Dreifach-Diffusor TDI 100 ist ein empfohlenes Zubehörteil für Betriebstemperaturen unter Raumtemperatur, um Kondensation durch einen trockenen Gasstrom am Kammerfenster zu verhindern. Die TFC 100 ist mit einem Feuchte­sensor erweiterbar.
<h2>TPC 150</h2>
Thermostatisiereinrichtung mit Peltiersystem für eine softwaregesteuerte Kontrolle der Temperatur von -30…160 °C (Aufheiz- und Abkühlrate ±1 K/s), auch unter Schutzgasatmosphäre. Der Dreifach-Diffusor TDI 150 ist ein empfohlenes Zubehörteil für Betriebstemperaturen unter Raumtemperatur, um Kondensation durch einen trockenen Gasstrom am Kammerfenster zu verhindern. Die TPC 150 ist mit einem Feuchte­sensor erweiterbar.
<h2>TEC 300 XL</h2>
Für die Benutzung mit den XL-Varianten der OCA Messgeräte. Thermostatisiereinrichtung mit elektrischer Widerstandsdoppelheizung für eine manuelle Kontrolle der Temperatur bis zu 300 °C, auch unter Schutzgasatmosphäre. Die Kammertemperatur wird in der SCA-Software in Kombination mit Kontaktwinkelmessung protokolliert.
<h2>TEC 400</h2>
Thermostatisiereinrichtung mit elektrischer Widerstandsdoppelheizung für eine software­gesteuerte Kontrolle der Temperatur bis zu 400 °C, auch unter Schutzgasatmosphäre.
<h2>TEC 700</h2>
Thermostatisiereinrichtung mit elektrischer Widerstandsdoppelheizung für eine software­gesteuerte Kontrolle der Temperatur bis zu 700 °C, auch unter Schutzgasatmosphäre.
<h2>NHD 400 / NHD 700</h2>
Die NHD 400 und NHD 700 sind elektrische Kanülen-Heizsysteme für die Erzeugung von Tropfen bei einer Temperatur von bis zu 400 °C bzw. 700 °C (z. B. Polymere und Hotmelts, geschmolzenes Metall, etc.). Ebenso können extrem zähe Copolymere mit Viskositäten im kPas Bereich, z.B. mit Elastomer-Komponenten, leicht dosiert werden.
<h2>MTFQ 1200</h2>
Hochtemperaturrohrofen für Messungen im Hochtemperaturbereich, z.B. an Loten, Halbleitern, Legierungen und Glasschmelzen, auch für Messungen unter Schutzgas-Atmosphäre bei bis zu 1200°C.
<h2>HTFC 1500/1700</h2>
Hochtemperaturrohrofen für Messungen im Hochtemperaturbereich, z.B. an Loten, Halbleitern, Legierungen und Glasschmelzen, auch für Messungen unter Schutzgas-Atmosphäre bei bis zu 1500/1700 °C
<h2>HTFC 1500/1800 HV</h2>
Elektrisch beheizter Hochtemperaturofen mit Hochvakuum zur Verwendung mit einem OCA 15LHT-SV Max. Temperatur 1500/1800 °C und max. 1 mPa Vakuum
<h2>PMC 50S</h2>
Druckmesszelle mit Sichtfenstern aus optischem Saphirglas, für Kontaktwinkelmessungen nach der Sessile Drop- und Captive Bubble-Methode und für die Messung der Oberflächenspannung in Flüssig-/Gas-Systemen und für Grenzflächenspannungen in Flüssig-/Flüssig-Systemen.<br />Mit vier Dosierkanülen für Sessile &amp; Captive sowie Pendant Drop Messungen, mit vier verschiedenen Einspritzflüssigkeiten. Äusseres Zwei-Ebenen Druck-Zufuhr-System. Die erste Druckebene kontrolliert den Druck der Primär-Einfüllflüssigkeit. Der höhere, zweite Druck ist notwendig für die Injektion der Meßflüssigkeit/en.<br />Druckbereich: 0,03 ... 50 bar<br />Max. Temperatur: 80 °C
<h2>HGC 20EC/20/30</h2>
Für die automatiserte Regelung der Feuchte von reiner Umgebungsluft, Stickstoff, Argon oder anderer Edelgase in thermostatisierten Messzellen.
Mehr Informationen über den&nbsp;<link 183 - internal-link "Opens internal link in current window">Feuchtegenerator HGC</link>";s:11:"rte_enabled";s:1:"0";s:6:"layout";s:1:"0";s:11:"spaceBefore";s:1:"0";s:10:"spaceAfter";s:1:"0";s:13:"section_frame";s:1:"0";s:6:"hidden";s:1:"0";s:12:"sectionIndex";s:1:"1";s:9:"linkToTop";s:1:"0";s:9:"starttime";N;s:7:"endtime";N;s:8:"fe_group";s:0:"";s:10:"categories";s:1:"0";s:11:"l18n_parent";s:1:"0";}";}}